# Surface plasmon resonance

﻿
Surface plasmon resonance

The excitation of surface plasmons by light is denoted as a surface plasmon resonance (SPR) for planar surfaces or localized surface plasmon resonance (LSPR) for nanometer-sized metallic structures.

This phenomenon is the basis of many standard tools for measuring adsorption of material onto planar metal (typically gold and silver) surfaces or onto the surface of metal nanoparticles. It is behind many color based biosensor applications and different lab-on-a-chip sensors.

Explanation

Surface plasmons, also known as surface plasmon polaritons, are surface electromagnetic waves that propagate in a direction parallel to the metal/dielectric (or metal/vacuum) interface. Since the wave is on the boundary of the metal and the external medium (air or water for example), these oscillations are very sensitive to any change of this boundary, such as the adsorption of molecules to the metal surface.

To describe the existence and properties of surface plasmons, one can choosefrom various models (quantum theory, Drude model, etc.). The simplest way to approach the problem is to treat each material as a homogeneous continuum,described by a dielectric constant. With the terms of this description for electronic surface plasmons to exist, the real part of the dielectric constant of the metal must be negative and its magnitude must be greater than that of the dielectric. This condition is met in the IR-visible wavelength region for air/metal and water/metal interfaces (where the real dielectric constant of a metal is negative and that of air or water is positive).

Realisation

In order to excite surface plasmons in a resonant manner, one can use an electron or light beam (visible and infrared are typical). The incoming beam has to match its impulse to that of the plasmon. In the case of p-polarized light (polarization occurs parallel to the plane of incidence), this is possible by passing the light through a block of glass to increase the wavenumber (and the impulse), and achieve the resonance at a given wavelength and angle. S-polarized (polarization occurs perpendicular to the plane of incidence) light cannot excite electronic surface plasmons.Electronic and magnetic surface plasmons obey the following dispersion relation:

:$K\left(omega\right) = frac\left\{omega\right\}\left\{c\right\} sqrt\left\{frac\left\{epsilon_1 epsilon_2 mu_1 mu_2\right\}\left\{epsilon_1 mu_1 + epsilon_2 mu_2$

Typical metals that support surface plasmons are silver and gold, but metals such as copper, titanium, or chromium can also support surface plasmon generation.

Using light to excite SP waves, there are two constructions which are wellknown. In the Otto setup, the light is shone on the wall of a glass block, typically a prism, and totally reflected. A thin metal (for example gold) film is positioned close enough, that the evanescent waves can interact with the plasma waves on the surface and excite the plasmons.

In the Kretschmann configuration, the metal film is evaporated onto the glassblock. The light is again illuminating from the glass, and an evanescent wavepenetrates through the metal film. The plasmons are excited at the outer sideof the film.This configuration is used in most practical applications.

Applications

Surface plasmons have been used to enhance the surface sensitivity of several spectroscopic measurements including fluorescence, Raman scattering, and second harmonic generation. However, in their simplest form, SPR reflectivity measurements can be used to detect DNA or proteins by the changes in the local index of refraction upon adsorption of the target molecule to the metal surface. If the surface is patterned with different biopolymers, the technique is called Surface Plasmon Resonance Imaging (SPRI).

For nanoparticles, localized surface plasmon oscillations can give rise to the intense colors of solutions of plasmon resonance nanoparticles and/or very intense scattering. Nanoparticles of noble metals exhibit strong ultraviolet-Visible absorption bands that are not present in the bulk metal. Shifts in this resonance due to changes in the local index of refraction upon adsorption of biopolymers to the nanoparticles can also be used to detect biopolymers such as DNA or proteins. Related complementary techniques include plasmon waveguide resonance, QCM and Dual Polarisation Interferometry

Examples

Layer-by-layer self-assembly

One of the first common applications of surface plasmon resonance spectroscopy was the measurement of the thickness of adsorbed self-assembled nanofilms on gold substrates. An example is presented in 'figure' The resonance curves shift to higher angles as the thickness of the adsorbed film increases. This example is a 'static SPR' measurement.

When higher speed observation is desired, one can select an angle right belowthe resonance point (the angle of minimum reflectance), and measure the reflectivity changes at that point.This is the so called 'dynamic SPR' measurement. The interpretation of the data assumes, that the structure of the film does not change significantly during the measurement.

Binding constant determination

When the affinity of two ligands has to be determined, the binding constant must be determined. It is the equilibrium value for the product quotient. This value can also be found using the dynamical SPR parameters and, as in any chemical reaction, it is the association rate divided by the dissociation rate.

For this, a so-called bait ligand is coated to the gold surface of the SPR crystal. Through a microflow system, a solution with the prey ligand can flow over the bait layer and bind. Binding will make the SPR signal change to an equilibrium. After some time, a solution without the prey is applied, and a new equilibrium will be reached. From these association ('on rate', von) and dissociation speeds ('off rate', voff), the binding constant can be calculated.

The actual SPR signal can be explained by the electromagnetic 'coupling' of the incident light with the surface plasmon of the gold layer. This plasmon can be influenced by the layer just a few nanometer across the gold-solution interface i.e. the bait protein and possibly the prey protein. Binding makes the reflection angle change.;

$K_D = frac\left\{v_\left\{ ext\left\{off\right\}\left\{v_\left\{ ext\left\{on\right\}$

Magnetic Plasmon Resonance

Recently, there has been an interest in magnetic surface plasmons. These require materials with large negative magnetic permeability, a property that has only recently been made available with the construction of metamaterials.

References

*cite book | Author= Heinz Raether | title= Surface plasmons on smooth and rough surfaces and on gratings | publisher=Springer Verlag, Berlin | year=1988 | id=ISBN 978-3540173632
*
*Hutter E, Fendler J. Exploitation of Localized Surface Plasmon Resonance. Adv. Mater. 2004, 16, 19, 1685-1706.
*Aslan K, Lakowicz JR, Geddes C. Plasmon light scattering in biology and medicine: new sensing approaches, visions and perspectives. Current Opinion in Chemical Biology 2005, 9:538–544
*Smith EA, Corn RM. Surface Plasmon Resonance Imaging as a Tool to Monitor Biomolecular Interactions in an Array Based Format. Appl. Spectroscopy, 2003, 57, 320A-332A.
*J. N. Gollub, D. R. Smith, D. C. Vier, T. Perram, J. J. Mock, Phys. Rev. B 71, 195402 (2005)
*http://www.astbury.leeds.ac.uk/facil/SPR/spr_intro2004.htm (a short detailed synopsis of how surface plasmon resonance works in practice)

* [http://www.ncbi.nlm.nih.gov/books/bv.fcgi?rid=imm.figgrp.2463 The affinity and valence of an antibody can be determined by equilibrium dialysis (ncbi.nlm.nih.gov)]

Companies

[http://www.biosuplar.com] Biospular

[http://www.biacore.com] Biacore

[http://www.genoptics-spr.com] Genoptics

[http://www.Reichertspr.com] Reichert Life Sciences

[http://www.fujifilm.com/lifescience] Fujifilm Life Science

*Hydrogen microsensor
*Plasmon
*Nano-optics
*Biosensor
*Dual Polarization Interferometry

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• Surface plasmon résonance — Résonance plasmon de surface La résonance des plasmons de surface (ou surface plasmon resonance en anglais) est un phénomène physique principalement connu pour son utilisation comme méthode de mesure de la liaison d un « ligand » sur un …   Wikipédia en Français

• surface plasmon resonance — Alteration in light reflectance as a result of binding of molecules to a surface from which total internal reflection is occurring. Used in the Biacore (Pharmacia Trademark) machine that detects the binding of ligand to surface immobilised… …   Dictionary of molecular biology

• Surface plasmon — Surface plasmons, also referred to in the literature as surface plasma polaritons, are fluctuations in the electron density at the boundary of two materials. Plasmons are the collective vibrations of an electron gas (or plasma) surrounding the… …   Wikipedia

• Resonance plasmon de surface — Résonance plasmon de surface La résonance des plasmons de surface (ou surface plasmon resonance en anglais) est un phénomène physique principalement connu pour son utilisation comme méthode de mesure de la liaison d un « ligand » sur un …   Wikipédia en Français

• Plasmon — Dans un métal, un plasmon est une oscillation de plasma quantifiée, ou un quantum d oscillation de plasma. Le plasmon est une quasiparticule résultant de la quantification de Fréquence plasma, tout comme le photon et le phonon sont des… …   Wikipédia en Français

• Résonance plasmon de surface — La résonance des plasmons de surface (ou surface plasmon resonance en anglais) est un phénomène physique principalement connu pour son utilisation comme méthode de mesure de la liaison d un « ligand » sur un « récepteur »… …   Wikipédia en Français

• Plasmon — In physics, a plasmon is a quantum of a plasma oscillation. The plasmon is the quasiparticle resulting from the quantization of plasma oscillations just as photons and phonons are quantizations of light and sound waves, respectively. Thus,… …   Wikipedia

• Resonance — This article is about resonance in physics. For other uses, see Resonance (disambiguation). Resonant redirects here. For the phonological term, see Sonorant. Increase of amplitude as damping decreases and frequency approaches resonant frequency… …   Wikipedia

• Surface enhanced Raman spectroscopy — Surface Enhanced Raman Spectroscopy, or Surface Enhanced Raman Scattering, often abbreviated SERS, is a surface sensitive technique that results in the enhancement of Raman scattering by molecules adsorbed on rough metal surfaces. The enhancement …   Wikipedia

• Surface (homoymie) — Cette page d’homonymie répertorie les différents sujets et articles partageant un même nom. la surface est un espace géométrique, Voir aussi catégorie:surface (en géométrie) Lexique des surfaces Microsoft Surface : une table tactile. grande… …   Wikipédia en Français