- Infrared spectroscopy
Infrared spectroscopy (IR spectroscopy) is the subset of
spectroscopythat deals with the infraredregion of the electromagnetic spectrum. It covers a range of techniques, the most common being a form of absorption spectroscopy. As with all spectroscopic techniques, it can be used to identify compounds or investigate sample composition. Infrared spectroscopy correlation tables are tabulated in the literature.
Background and Theory
The infrared portion of the electromagnetic spectrum is divided into three regions; the near-, mid- and far- infrared, named for their relation to the visible spectrum. The far-infrared, approximately 400-10 cm-1 (1000–30 μm), lying adjacent to the
microwaveregion, has low energy and may be used for rotational spectroscopy. The mid-infrared, approximately 4000-400 cm-1 (30–1.4 μm) may be used to study the fundamental vibrations and associated rotational-vibrational structure. The higher energy near-IR, approximately 14000-4000 cm-1 (1.4–0.8 μm) can excite overtone or harmonic vibrations. The names and classifications of these subregions are merely conventions. They are neither strict divisions nor based on exact molecular or electromagnetic properties.
Infrared spectroscopy exploits the fact that molecules have specific frequencies at which they rotate or vibrate corresponding to discrete
energy levels (vibrational modes). These resonant frequencies are determined by the shape of the molecular potential energy surfaces, the masses of the atoms and, by the associated vibronic coupling. In order for a vibrational mode in a molecule to be IR active, it must be associated with changes in the permanent dipole. In particular, in the Born-Oppenheimer and harmonic approximations, i.e. when the molecular Hamiltoniancorresponding to the electronic ground statecan be approximated by a harmonic oscillator in the neighborhood of the equilibrium molecular geometry, the resonant frequencies are determined by the normal modescorresponding to the molecular electronic ground state potential energy surface. Nevertheless, the resonant frequencies can be in a first approach related to the strength of the bond, and the mass of the atoms at either end of it. Thus, the frequency of the vibrations can be associated with a particular bond type.
Simple diatomic molecules have only one bond, which may stretch. More complex molecules have many bonds, and vibrations can be conjugated, leading to infrared absorptions at characteristic frequencies that may be related to chemical groups. For example, the atoms in a CH2 group, commonly found in
organic compounds can vibrate in six different ways: symmetrical and antisymmetrical stretching, scissoring, rocking, wagging and twisting:
The infrared spectrum of a sample is collected by passing a beam of infrared light through the sample. Examination of the transmitted light reveals how much energy was absorbed at each wavelength. This can be done with a
monochromaticbeam, which changes in wavelength over time, or by using a Fourier transforminstrument to measure all wavelengths at once. From this, a transmittanceor absorbancespectrum can be produced, showing at which IR wavelengths the sample absorbs. Analysis of these absorption characteristics reveals details about the molecular structure of the sample.
This technique works almost exclusively on samples with
covalent bonds. Simple spectra are obtained from samples with few IR active bonds and high levels of purity. More complex molecular structures lead to more absorption bands and more complex spectra. The technique has been used for the characterization of very complex mixtures.
Gaseous samples require little preparation beyond purification, but a sample cell with a long
pathlength(typically 5-10 cm) is normally needed, as gases show relatively weak absorbances.
Liquid samples can be sandwiched between two plates of a high purity salt (commonly
sodium chloride, or common salt, although a number of other salts such as potassium bromideor calcium fluorideare also used). The plates are transparent to the infrared light and will not introduce any lines onto the spectra. Some salt plates are highly soluble in water, so the sample and washing reagents must be anhydrous(without water).
Solid samples can be prepared in four major ways. The first is to crush the sample with a
mulling agent(usually nujol) in a marbleor agatemortar, with a pestle. A thin film of the mull is applied onto salt plates and measured.
The second method is to grind a quantity of the sample with a specially purified salt (usually
potassium bromide) finely (to remove scattering effects from large crystals). This powder mixture is then crushed in a mechanical die pressto form a translucent pellet through which the beam of the spectrometer can pass.
The third technique is the Cast Film technique, which is used mainly for polymeric materials. The sample is first dissolved in a suitable, non hygroscopic solvent. A drop of this solution is deposited on surface of KBr or NaCl cell. The solution is then evaporated to dryness and the film formed on the cell is analysed directly. Care is important to ensure that the film is not too thick otherwise light cannot pass through. This technique is suitable for qualitative analysis.
The final method is to use
microtomyto cut a thin (20-100 micrometre) film from a solid sample. This is one of the most important ways of analysing failed plastic products for example because the integrity of the solid is preserved.
It is important to note that spectra obtained from different sample preparation methods will look slightly different from each other due to differences in the samples' physical states.
A beam of infrared light is produced and split into two separate beams. One is passed through the sample, the other passed through a reference which is often the substance the sample is dissolved in. The beams are both reflected back towards a detector, however first they pass through a splitter which quickly alternates which of the two beams enters the detector. The two signals are then compared and a printout is obtained.
A reference is used for two reasons:
*This prevents fluctuations in the output of the source affecting the data
*This allows the effects of the solvent to be cancelled out (the reference is usually a pure form of the solvent the sample is in)
Summary of absorptions of bonds in organic molecules
Wavenumbers listed in cm-1.
Uses and applications
Infrared spectroscopy is widely used in both research and industry as a simple and reliable technique for measurement, quality control and dynamic measurement. It is of especial use in
forensic analysisin both criminal and civil cases, enabling identification of polymer degradationfor example. It is perhaps the most widely used method of applied spectroscopy.Fact|date=May 2008
The instruments are now small, and can be transported, even for use in field trials. With increasing technology in computer filtering and manipulation of the results, samples in solution can now be measured accurately (water produces a broad absorbance across the range of interest, and thus renders the spectra unreadable without this computer treatment). Some machines will also automatically tell you what substance is being measured from a store of thousands of reference spectra held in storage.
By measuring at a specific frequency over time, changes in the character or quantity of a particular bond can be measured. This is especially useful in measuring the degree of polymerization in
polymermanufacture. Modern research machines can take infrared measurements across the whole range of interest as frequently as 32 times a second. This can be done whilst simultaneous measurements are made using other techniques. This makes the observations of chemical reactions and processes quicker and more accurate.
Techniques have been developed to assess the quality of tea-leaves using infrared spectroscopy. This will mean that highly trained experts (also called 'noses') can be used more sparingly, at a significant cost saving. [citation|last1=Luypaert|first1=J.|last2=Zhang|first2=M.H.|last3=Massart|first3=D.L.
contribution= Feasibility study for the use of near infrared spectroscopy in the qualitative and quantitative analysis of green tea, Camellia sinensis (L.)
title=Analytica Chimica Acta|publisher=Elsevier
Infrared spectroscopy has been highly successful for applications in both organic and inorganic chemistry. Infrared spectroscopy has also been successfully utilized in the field of semiconductor microelectronics [cite book |last= Lau|first= W.S. |coauthors= |title= Infrared characterization for microelectronics |publisher= World Scientific |year= 1999 |month= |isbn= ] : for example, infrared spectroscopy can be applied to semiconductors like
silicon, gallium arsenide, gallium nitride, zinc selenide, amorphous silicon, silicon nitride, etc.
The different isotopes in a particular species may give fine detail in infrared spectroscopy. For example, the O-O stretching frequency of oxy
hemocyaninis experimentally determined to be 832 and 788 cm-1 for ν(16O-16O) and ν(18O-18O) respectively.
By considering the O-O as a spring, the wavelength of absorbance, ν can be calculated:
where "k" is the spring constant for the bond, and "μ" is the
reduced massof the A-B system:
( is the mass of atom ).
The reduced masses for 16O-16O and 18O-18O can be approximated as 8 and 9 respectively. Thus
Fourier transform infrared spectroscopy
Fourier transform infrared (FTIR) spectroscopy is a measurement technique for collecting infrared spectra. Instead of recording the amount of energy absorbed when the frequency of the infra-red light is varied (monochromator), the IR light is guided through an
interferometer. After passing the sample the measured signal is the interferogram. Performing a mathematical Fourier transformon this signal results in a spectrum identical to that from conventional (dispersive) infrared spectroscopy.
FTIR spectrometers are cheaper than conventional spectrometers because building of interferometers is easier than the fabrication of a monochromator. In addition, measurement of a single spectrum is faster for the FTIR technique because the information at all frequencies is collected simultaneously. This allows multiple samples to be collected and averaged together resulting in an improvement in sensitivity. Because of its various advantages, virtually all modern infrared spectrometers are FTIR instruments.
Two-dimensional infrared spectroscopy
Two-dimensional infrared correlation spectroscopy analysis is the application of
2D correlation analysison infrared spectra. By extending the spectral information of a perturbed sample, spectral analysis is simplified and resolution is enhanced. The 2D synchronous and 2D asynchronous spectra represent a graphical overview of the spectral changes due to a perturbation (such as a changing concentration or changing temperature) as well as the relationship between the spectral changes at two different wavenumbers.
Nonlinear two-dimensional infrared spectroscopy [cite journal | author = P. Hamm, M. H. Lim, R. M. Hochstrasser | title = Structure of the amide I band of peptides measured by femtosecond nonlinear-infrared spectroscopy
journal = J. Phys. Chem. B | volume = 102 | pages = 6123 | year = 1998 | doi = 10.1021/jp9813286] [cite journal | author = S. Mukamel | title = Multidimensional Fentosecond Correlation Spectroscopies of Electronic and Vibrational Excitations | journal = Annual Review of Physics and Chemistry | volume =51 | pages = 691 | year = 2000 | doi = 10.1146/annurev.physchem.51.1.691 ] is the infrared version of
correlation spectroscopy. Nonlinear two-dimensional infrared spectroscopy is a technique that has become available with the development of femtosecondinfrared laser pulses. In this experiment first a set of pump pulses are applied to the sample. This is followed by a waiting time, where the system is allowed to relax. The waiting time typically lasts from zero to several picoseconds and the duration can be controlled with a resolution of tens of femtoseconds. A probe pulse is then applied resulting in the emission of a signal from the sample. The nonlinear two-dimensional infrared spectrum is a two-dimensional correlation plot of the frequency that was excited by the initial pump pulses and the frequency excited by the probe pulse after the waiting time. This allows the observation of coupling between different vibrational modes. Because of its extremely high time resolution it can be used to monitor molecular dynamics on a picosecond timescale. It is still a largely unexplored technique and is becoming increasingly popular for fundamental research.
Like in two-dimensional nuclear magnetic resonance (2DNMR) spectroscopy this technique spreads the spectrum in two dimensions and allow for the observation of cross peaks that contain information on the coupling between different modes. In contrast to 2DNMR nonlinear two-dimensional infrared spectroscopy also involve the excitation to overtones. These excitations result in excited state absorption peaks located below the diagonal and cross peaks. In 2DNMR two distinct techniques, COSY and NOESY, are frequently used. The cross peaks in the first are related to the scalar coupling, while in the later they are related to the spin transfer between different nuclei. In nonlinear two-dimensional infrared spectroscopy analogs have been drawn to these 2DNMR techniques. Nonlinear two-dimensional infrared spectroscopy with zero waiting time corresponds to COSY and nonlinear two-dimensional infrared spectroscopy with finite waiting time allowing vibrational population transfer corresponds to NOESY. The COSY variant of nonlinear two-dimensional infrared spectroscopy has been used for determination of the secondary structure content proteins. [cite journal | author = N. Demirdöven, C. M. Cheatum, H. S. Chung, M. Khalil, J. Knoester, A. Tokmakoff | title= Two-dimensional infrared spectroscopy of antiparallel beta-sheet secondary structure | journal =
Journal of the American Chemical Society| volume = 126 | pages = 7981 | year = 2004 | doi = 10.1021/ja049811j]
Infrared spectroscopy correlation table
Fourier transform spectroscopy
Near infrared spectroscopy
Far infrared astronomy
Forensic polymer engineering
* A useful gif animation of different vibrational modes: [http://www.shu.ac.uk/schools/sci/chem/tutorials/molspec/irspec1.htm here]
* [http://www.organicworldwide.net/infrared Infrared spectroscopy for organic chemists]
* [http://riodb01.ibase.aist.go.jp/sdbs/cgi-bin/cre_index.cgi?lang=eng Organic compounds spectrum database]
Wikimedia Foundation. 2010.
См. также в других словарях:
infrared spectroscopy — infraraudonoji spektroskopija statusas T sritis Standartizacija ir metrologija apibrėžtis Optinė spektroskopija, aprėpianti nuo 750 nm iki 2,5 mm ilgio elektromagnetinių bangų spektrus. atitikmenys: angl. infrared spectroscopy; IR spectroscopy… … Penkiakalbis aiškinamasis metrologijos terminų žodynas
infrared spectroscopy — infraraudonoji spektroskopija statusas T sritis Standartizacija ir metrologija apibrėžtis Optinės spektroskopijos dalis, aprėpianti spinduliavimo, sugerties ir atspindžio spektrus infraraudonojoje spektro srityje. atitikmenys: angl. infrared… … Penkiakalbis aiškinamasis metrologijos terminų žodynas
infrared spectroscopy — infraraudonoji spektroskopija statusas T sritis fizika atitikmenys: angl. infrared spectroscopy; IR spectroscopy vok. Infrarotspektroskopie, f; IR Spektroskopie, f rus. ИК спектроскопия, f; инфракрасная спектроскопия, f pranc. spectroscopie… … Fizikos terminų žodynas
infrared spectroscopy — infraraudonoji spektroskopija statusas T sritis chemija apibrėžtis Spektroskopija, tirianti 750 nm–1,5 mm ilgio elektromagnetinių bangų spektrus. atitikmenys: angl. infrared spectroscopy rus. инфракрасная спектроскопия … Chemijos terminų aiškinamasis žodynas
infrared spectroscopy — examination by means of an infrared spectrometer, to study systems or molecules by examining their interactions with infrared radiation … Medical dictionary
infrared spectroscopy — noun The spectroscopic study of the interaction of matter with infrared radiation; used as an analytical tool to identify (mostly organic) compounds … Wiktionary
infrared spectroscopy — The study of the properties of material systems by means of their interaction with infrared radiation; ordinarily the radiation is dispersed into a spectrum after passing through the material … Forensic science glossary
Infrared spectroscopy correlation table — In physical and analytical chemistry, infrared spectroscopy ( IR spectroscopy ) is a technique used to identify chemical compounds based on how infrared radiation is absorbed by the compounds chemical bonds. This article is an IR spectroscopy… … Wikipedia
Near-infrared spectroscopy — Near IR absorption spectrum of dichloromethane showing complicated overlapping overtones of mid IR absorption features. Near infrared spectroscopy (NIRS) is a spectroscopic method that uses the near infrared region of the electromagnetic spectrum … Wikipedia
Near infrared spectroscopy — (NIRS) is a spectroscopic method utilising the near infrared region of the electromagnetic spectrum (from about 800 nm to 2500 nm). Typical applications include pharmaceutical, medical diagnostics (including blood sugar and oximetry), food and… … Wikipedia