- Electron microprobe
An electron microprobe (EMP), also known as an electron probe microanalyser (EPMA) is an analytical tool used to non-destructively determine the chemical composition of small volumes of solid materials. It works similarly to a
scanning electron microscope, in which the sample is bombarded with an electron beamand signals that come from the sample are collected. This enables the elements present within sample volumes of 10-30 cubic micrometres or less to be determined.Wittry, David B. (1958). "Electron Probe Microanalyzer", [http://www.google.co.uk/patents?id=g7tMAAAAEBAJ&d US Patent No 2916621] , Washington, DC: U.S. Patent and Trademark Office.] Elements from boronto plutoniumcan be quantitatively analysed at levels as low as 100 parts per million(ppm).
Development of the electron microprobe was preceded by that of the closely related analytical technique of
X-ray fluorescencespectrometry (XRF). This technique was first proposed by Georg von Hevesyin 1923 and applied by other workers in the following few years.
In 1944, MIT built an electron microprobe, combining an electron microscope and an
energy-loss spectrometer. Electron energy-loss spectrometry is very good for light element analysis and they obtained spectra of C-Kα, N-Kα and O-Kα radiation. In 1947, Hiller patented the idea of using an electron beam to produce analytical X-rays, but never constructed a working model. His design proposed using Bragg diffractionfrom a flat crystal to select specific X-ray wavelengths and a photographic plate as a detector.
In 1948-1950, Raymond Castaing, supervised by
André Guinier, built the first electron “microsonde électronique” (electron microprobe) at the University of Paris. This microprobe produced an electron beam diameter of 1-3 μm with a beam current of ~10 nanoamperes (nA) and used a Geiger counter to detect the X-rays produced from the sample. However, the Geiger counter could not distinguish X-rays produced from specific elements and in 1950, Castaing added a quartzcrystal between the sample and the detector to permit wavelength discrimination. He also added an optical microscope to view the point of beam impact. The resulting microprobe was described in Castaing's 1951 Ph.D. thesis, in which he laid the foundations of the theory and application of quantitative analysis by electron microprobe, establishing the theoretical framework for the matrix corrections of absorption and fluorescence effects. Castaing (1921-1999) is considered the "father" of electron microprobe analysis.
Cameca (France) produced the first commercial microprobe, the “MS85,” in 1956. It was soon followed by many microprobes from other companies; however, all companies except Cameca and
JEOL, are now out of business. In addition, many researchers build electron microprobes in their labs. Significant subsequent improvements and modifications to microprobes included scanning the electron beam to make X-ray maps (1960), the addition of solid state EDS detectors (1968) and the development of synthetic multilayer diffracting crystals for analysis of light elements (1984).
Low-energy electrons are produced from a tungsten filament cathode and accelerated by a positively biased
anodeplate to 10 to 30 thousand electron volts(keV). The anode plate has central aperture and electrons that pass through it are collimated and focused by a series of magnetic lenses and apertures. The resulting approximately 1 micrometre diameter electron beam may be rastered across the sample or used in spot mode to excite various effects from the sample. Among these are: phononexcitation (heat), cathodoluminescence(visible light fluorescence), continuum X-ray radiation ( bremsstrahlung), characteristic X-ray radiation, secondary electrons ( plasmonproduction), backscattered electron production, and Auger electronproduction.
The characteristic X-rays are used for chemical analysis. Specific X-ray wavelengths are selected and counted, either by wave-length dispersive spectrometry (WDS) or
energy dispersive X-ray spectroscopy(EDS). WDS utilizes Bragg diffraction from crystals to select X-ray wavelengths of interest and direct them to gas-flow or sealed proportional detectors. In contrast, EDS uses a solid state semiconductor detectorto accumulate X-rays of all wavelengths produced from the sample.
Chemical composition is determined by comparing the intensities of characteristic X-rays from the sample material with intensities from known composition (standards). Count from the sample must be corrected for
matrix effects ( absorptionand secondary fluorescence) to yield chemical compositions. The resulting chemical information is gathered in textural context. Variations in chemical composition within a material (zoning), such as a mineral grain or metal, can be readily determined.
* [http://earth.es.huji.ac.il/machon/e-prob/prob.html Electron Probe Laboratory, Hebrew University of Jerusalem] - web page of a lab describing their modern EPMA
Wikimedia Foundation. 2010.
Look at other dictionaries:
electron microprobe — elektronpluoštis zondas statusas T sritis chemija apibrėžtis Labai mažo skersmens fokusuotų elektronų pluoštas. atitikmenys: angl. electron microprobe; electron probe; electron beam probe rus. электронно лучевой зонд; электронный зонд … Chemijos terminų aiškinamasis žodynas
electron microprobe — elektroninis mikrozondas statusas T sritis radioelektronika atitikmenys: angl. electron microprobe vok. Elektronenmikrosonde, f rus. электронный микрозонд, m pranc. microsonde électronique, f … Radioelektronikos terminų žodynas
electron microprobe — A technique of elemental analysis in the electron microscope based on spectral analysis of the scattered X ray emission from the specimen induced by the electron beam. Using this technique it is possible to obtain quantitative data on, for… … Dictionary of molecular biology
electron microprobe analysis — elektronzondinė mikroanalizė statusas T sritis chemija apibrėžtis Labai mažo medžiagos kiekio (≤ 10⁻¹³ g) analizė, kai į bandinį nukreiptas labai siauras elektronų pluoštas sukelia tai medžiagai būdingą rentgeno spinduliuotę. atitikmenys: angl.… … Chemijos terminų aiškinamasis žodynas
Microprobe — A microprobe is an instrument that applies a stable and well focused beam of charged particles (electrons or ions) to a sample. Types When the primary beam consists of accelerated electrons, the probe is termed an electron microprobe, when the… … Wikipedia
electron microscopy — Any form of microscopy in which the interactions of electrons with the specimens are used to provide information about the fine structure of that specimen. In transmission electron microscopy (TEM) the diffraction and adsorption of electrons as… … Dictionary of molecular biology
electron probe — elektronpluoštis zondas statusas T sritis chemija apibrėžtis Labai mažo skersmens fokusuotų elektronų pluoštas. atitikmenys: angl. electron microprobe; electron probe; electron beam probe rus. электронно лучевой зонд; электронный зонд … Chemijos terminų aiškinamasis žodynas
electron-beam probe — elektronpluoštis zondas statusas T sritis chemija apibrėžtis Labai mažo skersmens fokusuotų elektronų pluoštas. atitikmenys: angl. electron microprobe; electron probe; electron beam probe rus. электронно лучевой зонд; электронный зонд … Chemijos terminų aiškinamasis žodynas
microprobe — See electron microprobe … Dictionary of molecular biology
electron probe — noun Date: 1962 a microprobe that uses an electron beam to induce X ray emissions in a sample … New Collegiate Dictionary