- Laser-induced breakdown spectroscopy
Laser-induced breakdown spectroscopy (LIBS) is a type of atomic emission
spectroscopywhich utilises a highly energetic laserpulse as the excitation source. LIBS can analyse any matterregardless of its physical state, be it solid, liquid or gas. Even slurries, aerosols, gels, and more can be readily investigated. Because all elements emit light when excited to sufficiently high temperatures, LIBS can detect all elements, limited only by the power of the laser as well as the sensitivity and wavelength range of the spectrograph & detector. Basically LIBS makes use of optical emission spectrometry and is to this extent very similar to arc/spark emission spectroscopy.
LIBS operates by focusing the laser onto a small area at the surface of the specimen; when the laser is discharged it ablates a very small amount of material, in the range of nanograms to picograms, which instantaneously generates a plasma plume with temperatures of about 10,000–20,000 K. At these temperatures, the ablated material dissociates (breaks down) into excited
ionic and atomicspecies. During this time, the plasma emits a continuumof radiation which does not contain any useful information about the species present, but within a very small timeframe the plasma expands at supersonicvelocities and cools. At this point the characteristic atomic emission lines of the elements can be observed. The delay between the emission of continuum radiation and characteristic radiation is in the order of 10 µs, this is why it is necessary to temporally gate the detector.
LIBS can often be referred to as its alternative name: laser-induced plasma spectroscopy (LIPS). Unfortunately the term LIPS has alternative meanings that are outside the field of analytical spectroscopy, therefore the term LIBS is preferred.
LIBS is technically very similar to a number of other laser-based analytical techniques, sharing much of the same hardware. These techniques are the vibrational spectroscopic technique of
Raman spectroscopy, and the fluorescence spectroscopic technique of laser-induced fluorescence(LIF). In fact devices are now being manufactured which combine these techniques in a single instrument, allowing the atomic, molecularand structural characterisation of a specimen as well as giving a deeper insight into physical properties.
A typical LIBS system consists of a neodymium doped
yttrium aluminium garnet(Nd:YAG) solid-state laserand a spectrometerwith a wide spectral range and a high sensitivity, fast response rate, time gated detector. This is coupled to a computer which can rapidly process and interpret the acquired data. As such LIBS is one of the most experimentally simple spectroscopic analytical techniques, making it one of the cheapest to purchase and to operate.
The Nd:YAG laser generates energy in the near
infraredregion of the electromagnetic spectrum, with a wavelength of 1064 nm. The pulse duration is in the region of 10 ns generating a power density which can exceed 1 GW·cm-2 at the focal point. Other lasers have been used for LIBS mainly Excimer(Excited dimer) type generating energy in the visibleand ultravioletregions.
The spectrometer consists of either a
monochromator(scanning) or a polychromator(non-scanning) and a photomultiplieror CCD detector respectively. The most common monochromator is the Czerny-Turner type whilst the most common polychromator is the Echelle type, even so the Czerny-Turner type can be (and is often) used to disperse the radiation onto CCD effectively making it a polychromator. The polychromator spectrometer is the type most commonly used in LIBS as it allows simultaneous acquisition of the entire wavelength range of interest.
The spectrometer collects electromagnetic radiation over the widest wavelength range possible, maximising the number of emission lines detected for each particular element. Spectrometer response is typically from 1100 nm (
near infrared) to 170 nm ( deep ultraviolet), the approximate response range of a CCD detector. All elements have emission lines within this wavelength range. The energy resolution of the spectrometer can also affect the quality of the LIBS measurement, since high resolution systems can separate spectral emission lines in close juxtaposition, reducing interference and increasing selectivity. This feature is particularly important in specimens which have a complex matrix, containing a large number of different elements. Accompanying the spectrometer and detector is a delay generator which accurately gates the detector's response time, allowing temporal resolutionof the spectrum.
Because such a small amount of material is consumed during the LIBS process the technique is considered essentially non-destructive or minimally-destructive, and with an average power density of less than one watt radiated onto the specimen there is almost no specimen heating surrounding the ablation site.Due to the nature of this technique sample preparation is typically minimised to homogenisation or is often unnecessary where heterogeneity is to be investigated or where a specimen is known to be sufficiently , this reduces the possibility of contamination during chemical preparation steps. One of the major advantages of the LIBS technique is its ability to depth profile a specimen by repeatedly discharging the laser in the same position, effectively going deeper into the specimen with each shot. This can also be applied to the removal of surface contamination, where the laser is discharged a number of times prior to the analysing shot. LIBS is also a very rapid technique giving results within seconds, making it particularly useful for high volume analyses or on-line industrial monitoring.
LIBS is an entirely optical technique, therefore it requires only optical access to the specimen. This is of major significance as fibre optics can be employed for remote analyses. And being an optical technique it is non-invasive, non-contact and can even be used as a stand-off analytical technique when coupled to appropriate telescopic apparatus. These attributes have significance for use in areas from hazardous environments to space exploration. Additionally LIBS systems can easily be coupled to an optical microscope for micro-sampling adding a new dimension of analytical flexibility.
The use of specialised optics or a mechanically positioned specimen stage can be used raster the laser over the surface of the specimen allowing spatially resolved chemical analysis and the creation of 'elemental maps'. This is very significant as chemical imaging is becoming more important in all branches of science and technology.
Portable LIBS systems are more sensitive, faster and can detect a wider range of elements (particularly the light elements) than competing techniques such as portable
x-ray fluorescence. And LIBS does not utilise ionizing radiationto excite the sample, which is both penetrating and potentially carcinogenic.
LIBS, like all other analytical techniques is not without limitations. It is subject to variation in the laser spark and resultant plasma which often limits reproducibility. The
accuracyof LIBS measurements is typically better than 10% and precisionis often better than 5%. The detection limits for LIBS vary from one element to the next depending on the specimen type and the experimental apparatus used. Even so detection limits of 1 to 30 ppm by mass are not uncommon, but can range from >100 ppm to <1 ppm.
Recent interest in LIBS has focused on the miniaturization of the components and the development of compact, low power, portable systems. This direction has been pushed along by interest from groups such as
NASA, ESAas well as the military. The Mars Science Laboratorymission plans to bring a combined Raman/LIBS onto Mars.
Recent developments in LIBS have seen the introduction of double-pulsed laser systems. For double-pulse LIBS one distinguishes between orthogonal and perpendicular configuration. In perpendicular configuration the laser is fired twice on the same spot on the specimen with a pulse separation in the order of one to a couple of tens of microseconds. Depending on pulse separation, the second pulse is more or less absorbed by the plasma plume caused by the previous pulse, resulting in a reheating of the laser plasma leading to signal enhancement.In orthogonal configuration a laser pulse is fired parallel to the sample surface either before or after the perpendicular pulse hits the specimen. The laser plasma ignited in the surrounding medium above the surface by a first pulse causes (by its shock wave) an area of reduced pressure above the specimen into which the actual plasma from the sample can expand. This has similar positive effects on sensitivity like LIBS performed at reduced pressures.If the orthogonal laser pulse is delayed with respect to the perpendicular one, the effects are similar as in the perpendicular configuration.
Both double-pulse LIBS as well as LIBS at reduced pressures are aimed at increasing the sensitivity of LIBS and the reduction of errors caused by the differential volatility of elements (e.g. Hydrogen as an impurity in solids). It also significantly reduces the matrix effects. Double-pulsed systems are also proving useful in conducting analysis in liquids, as the initial laser pulse forms a cavity bubble in which the second pulse acts on the evaporated material.
LIBS is one of several analytical techniques that can be deployed in the field as opposed to pure laboratory techniques e.g. spark OES. Recent research on LIBS is focusing on compact and (man-)portable systems. Industrial applications of LIBS are e.g. the detection of material mix-ups, analysis of inclusions in steel, analysis of slags in secondary metallurgy and high-speed identification of scrap pieces for material specific recycling tasks.
List of surface analysis methods
* [http://www.unizar.es/janzano LIBS Research group. University of Zaragoza Spain)] Development of sorting plastics with LIBS and other applications.
* [http://www.imi.cnrc-nrc.gc.ca/Carrefour_d_informations/Factsheets/LIBS_e.html NRC/IMI LIBS Research group] Our group is devoted to on-line and in-line application of LIBS to industrial process.
* [http://libs.lanl.gov/ LIBS Planetary Science Applications Website]
* [http://www.arl.army.mil/main/main/default.cfm?Action=247&Page=247 Military Application of LIBS]
* [http://www.chem.physics.kth.se/research/research_l_projects.htm Laser Spectroscopy Projects - LIBS] Includes a Flash animation of remote LIBS
* [http://www.whoi.edu/cms/files/amichel/2007/2/060330-48_18747.pdf Evaluation of LIBS for under-water applications]
* [http://cer.ucsd.edu/REPORTS/UCSD-CER-05-03.pdf Single Particle LIBS measurements]
* [http://www.sciencedirect.com/science?_ob=ArticleURL&_udi=B7GWS-4KSD8FG-1&_user=10&_coverDate=01%2F31%2F2007&_rdoc=1&_fmt=&_orig=search&_sort=d&view=c&_acct=C000050221&_version=1&_urlVersion=0&_userid=10&md5=fd3ffba6d6ddba2ec9d38fe5b13e51f7 Also called LIPS - a Combustion Research example]
* [http://www.appliedspectra.com/LIBS.html LIBS vs other analytical techniques]
* [http://www.solartii.com/analytical_instruments/lea-s500.htm Laser Elemental Analyzer]
author= Lee W. B., Wu J. Y., Lee Y. I., Sneddon J.
title=Recent applications of laser-induced breakdown spectrometry: A review of material approaches
journal=Applied Spectroscopy Review
author= Noll R.,Bette H., Brysch A., Kraushaar M., Mönch I., Peter L., Sturm V.
title=Laser-induced breakdown spectrometry – applications for production control and quality assurance in steel industry
journal= Spectrochimica Acta Part B
*David A. Cremers & Leon J. Radziemski. "Handbook of Laser-Induced Breakdown Spectroscopy" (London: John Wiley & Sons, 2006) ISBN 0470092998
*Andrzej W. Miziolek, Vincenzo Palleschi, Israel Schechter. "Laser Induced Breakdown Spectroscopy" (New York: Cambridge University Press, 2006) ISBN 0521852749
*B. Gornushkin, K. Amponsah-Manager, B.W. Smith, N. Omenetto, and J.D. Winefordner. “Microchip Laser Induced Breakdown Spectroscopy: Preliminary Feasibility Investigation” Applied Spectroscopy 2004, 58(7), 762-769.
*K. Amponsah-Manager, N. Omenetto, B.W. Smith, I.B. Gornushkin, and J.D. Winefordner. “Microchip Laser Ablation of Metals: Investigation of the Ablation Process in View of its Application to Laser Induced Breakdown Spectroscopy” JAAS 2005, 20(6), 544-551.
*C. Lopez-Moreno, K. Amponsah-Manager, B.W. Smith I.B. Gornushkin, N. Omenetto, and J.D. Winefordner. “Quantitation of Low-alloy Steel Samples by Powerchip Laser Induced Breakdown Spectroscopy” JAAS 2005, 20(6), 552-556.
*H. Bette, R. Noll. "High-speed laser-induced breakdown spectrometry for scanning microanalysis" J. Phys. D: Appl. Phys. 2004, 37, 1281–1288
*H. Balzer, M. Höhne, R. Noll, V. Sturm. "New approach to monitoring the Al depth profile of hot-dip galvanised sheet steel online using laser-induced breakdown spectroscopy" Anal Bioanal Chem 2006, 385, 225–233
*V. Sturm, L. Peter, R. Noll. "Steel analysis with laser-induced breakdown spectrometry in the vacuum ultraviolet" Appl. Spectroscopy 2000, 54, 1275–1278
*J.M. Vadillo, J.J. Laserna. "Laser-induced plasma spectrometry: truly a surface analytical tool" Spectrochimica Acta Part B: Atomic Spectroscopy 2004, vol. 59, issue 2, p. 147-161
Wikimedia Foundation. 2010.
См. также в других словарях:
Laser ablation — is the process of removing material from a solid (or occasionally liquid) surface by irradiating it with a laser beam. At low laser flux, the material is heated by the absorbed laser energy and evaporates or sublimates. At high laser flux, the… … Wikipedia
Spectroscopy — Analysis of white light by dispersing it with a prism is example of spectroscopy. Spectroscopy ( … Wikipedia
Laser applications — There are many scientific, military, medical and commercial laser applications which have been developed since the invention of the laser in the 1958. The coherency, high monochromaticity, and ability to reach extremely high powers are all… … Wikipedia
Laser diode — Top: a packaged laser diode shown with a penny for scale. Bottom: the laser diode chip is removed from the above package and placed on the eye of a needle for scale … Wikipedia
Laser-Ablation — Als Laserablation wird das Abtragen von Material von einer Oberfläche durch Beschuss mit gepulster Laserstrahlung bezeichnet. Inhaltsverzeichnis 1 Physikalische Grundlagen 2 Anwendungen 2.1 Materialbearbeitung 2.2 Beschichtungsmethode 2.3 … Deutsch Wikipedia
spectroscopy — spectroscopist /spek tros keuh pist/, n. /spek tros keuh pee, spek treuh skoh pee/, n. the science that deals with the use of the spectroscope and with spectrum analysis. [1865 70; SPECTRO + SCOPY] * * * Branch of analysis devoted to identifying… … Universalium
Spectroscopie sur plasma induit par laser — La spectroscopie sur plasma induit par laser ou LIBS (pour Laser induced breakdown spectroscopy) est une méthode analytique physique utilisée pour analyser quantitativement (dosage) les composants de la matière. Les progrès récents des lasers… … Wikipédia en Français
Espectroscopia de plasma inducido por láser — La espectroscopia de plasma inducido por láser, conocida como LIBS, es un tipo de espectroscopia de emisión atómica que emplea como fuente de excitación láseres de alta energía. Una de las grandes ventajas de la técnica LIBS es la posibilidad de… … Wikipedia Español
Nd:YAG laser — with lid open showing frequency doubled 532 nm green light Nd:YAG laser rod … Wikipedia
Atomic emission spectroscopy — (AES) is a method of chemical analysis that uses the intensity of light emitted from a flame, plasma, arc, or spark at a particular wavelength to determine the quantity of an element in a sample. The wavelength of the atomic spectral line gives… … Wikipedia