- Oscillating U-tube
The oscillating U-tube is a technique to determine the density of liquids and gases based on an electronic measurement of the frequency of oscillation, from which the density value is calculated. This measuring principle is based on the Mass-Spring Model.
The sample is filled into a container with oscillation capacity. The eigenfrequency of this container is influenced by the sample’s mass. This container with oscillation capacity is a hollow, U-shaped glass tube (oscillating U-tube) which is electronically excited into undamped oscillation (at the lowest possible amplitude). The two branches of the U-shaped oscillator function as its spring elements.
The direction of oscillation is normal to the level of the two branches. The oscillator’s eigenfrequency is only influenced by the part of the sample that is actually involved in the oscillation. The volume involved in the oscillation is limited by the stationary oscillation knots at the bearing points of the oscillator. If the oscillator is at least filled up to its bearing points, the same precisely defined volume always participates in the oscillation, thus the measured value of the sample’s mass can be used to calculate its density.
Overfilling the oscillator beyond the bearing points is irrelevant to the measurement. For this reason the oscillator can also be employed to measure the density of sample media that flow through the tube (Continuous Measurement).
In the digital density meter, the mechanic oscillation of the U-tube is e.g. electromagnetically transformed into an alternating voltage of the same frequency. The period τ can be measured with high resolution and stands in simple relation to the density ρ of the sample in the oscillator:
A and B are the respective instrument constants of each oscillator. Their values are determined by calibrating with two substances of the precisely known densities ρ1 and ρ2. Modern instruments calculate and store the constants A and B after the two calibration measurements, which are mostly performed with air and water. They employ suitable measures to compensate various parasitic influences on the measuring result, e.g. the influence of the sample’s viscosity and the non-linearity caused by the measuring instrument’s finite mass.
In 1967 the company Anton Paar GmbH presented the first digital density meter for liquids and gases at ACHEMA. It was the first instrument to employ the oscillating U-tube principle by Dr. Hans Stabinger and Prof. Hans Leopold for density determination.
- Stabinger, Hans: "Density Measurement using modern oscillating transducers", South Yorkshire Trading Standards Unit, Sheffield 1994
- ISO 15212-1
Wikimedia Foundation. 2010.
Look at other dictionaries:
Vacuum tube — This article is about the electronic device. For experiments in an evacuated pipe, see free fall. For the transport system, see pneumatic tube. Modern vacuum tubes, mostly miniature style In electronics, a vacuum tube, electron tube (in North… … Wikipedia
electron tube — an electronic device that consists, typically, of a sealed glass bulb containing two or more electrodes: used to generate, amplify, and rectify electric oscillations and alternating currents. Also called electronic tube. Cf. gas tube, vacuum tube … Universalium
Magnetic field oscillating amplified thruster — The Magnetic Field Oscillating Amplified Thruster (MOA) (often named as Plasma Engine by the media) is a versatile electrothermodynamic system, which is able to accelerate nearly every medium to extremely high velocities, thereby generating a… … Wikipedia
Viscometer — A viscometer (also called viscosimeter) is an instrument used to measure the viscosity of a fluid. For liquids with viscosities which vary with flow conditions, an instrument called a rheometer is used. Viscometers only measure under one flow… … Wikipedia
Hydrometer — Not to be confused with Hygrometer. Hydrometer from Practical Physics Schematic drawing of a hydrometer. The lower the densi … Wikipedia
Reginald Fessenden — Infobox Celebrity name = Reginald Fessenden caption = The Father of Radio Broadcasting birth date = birth date|1866|10|6|mf=y birth place = East Bolton, Quebec, Canada death date = death date and age|1932|7|22|1866|10|6|mf=y death place = Bermuda … Wikipedia
Custody transfer — in the oil and gas industry refers to the transactions involving transporting physical substance from one operator to another. This includes the transferring of raw and refined petroleum between tanks and tankers; tankers and ships and other… … Wikipedia
Guglielmo Marconi — Born 25 April 1874(1874 04 25) Palazzo Marescalchi, Bologna … Wikipedia
Brix — For other uses, see Brix (disambiguation). Degrees Brix (symbol °Bx) is the sugar content of an aqueous solution. One degree Brix is 1 gram of sucrose in 100 grams of solution and represents the strength of the solution as percentage by weight (% … Wikipedia
Gravity (alcoholic beverage) — A floating hydrometer in use to test the specific gravity of beer Gravity, in the context of fermenting alcoholic beverages, refers to the specific gravity, or relative density compared to water, of the wort or must at various stages in the… … Wikipedia