- Low-energy ion scattering
: "LEIS redirects here; for the Hawaiian garland see
Low-energy ion scattering spectroscopy (LEIS), sometimes referred to simply as ion scattering spectroscopy (ISS), is a surface-sensitive analytical technique used to characterize the chemical and structural makeup of materials. LEIS involves directing a stream of charged particles known as
ionsat a surfaceand making observations of the positions, velocities, and energies of particles originating from that surface. Data that is thus collected can be used to deduce information about the material such as the relative positions of atomsin a surface lattice and the elemental identity of those atoms. LEIS is closely related to both medium-energy ion scattering(MEIS) and high-energy ion scattering (HEIS, known in practice as Rutherford backscatteringspectroscopy, or RBS), differing primarily in the energy range of the ion beam used to probe the surface. While much of the information collected using LEIS can be obtained using other surface science techniques, LEIS is unique in its sensitivity to both structure and composition of surfaces. Additionally, LEIS is one of a very few surface-sensitive techniques capable of directly observing hydrogenatoms, an aspect that may make it an increasingly more important technique as the hydrogen economyis being explored.
LEIS systems consist of the following:
Ion source, used to direct a beam of ionsat a target sample. Electron impact ionization is typically used to ionize noble gasatoms such as He, Ne or Ar, while heating of wafers containing alkali atoms is used to create an alkali ion beam. Ions thus created hold a positive charge, typically +1, due to ejection of electrons from atoms. The range of energies used most often in LEIS is 500 eV to 20 keV. In order to attain good experimental resolution it is important to have a narrow energy spread (ΔE/E < 1%) in the outgoing ion beam.
# Ion beam manipulator, including
electrostatic lenses for focusing and beam-chopping. Lenses consist of a series of either plate or cylindergeometries and serve to collimate the beam as well as to selectively filter the beam based on massand velocity. Beam chopping is performed using a pulsed-wave generator when time-of-flight(TOF) experiments are performed. Ions only pass through the chopper when there is no applied voltage.
# Sample manipulator, allows an operator to change the position and/or
angleof the target in order to perform experiments with varying geometries. Using directional controls, azimuthal (rotational) and incident angle adjustments may be made.
# Drift tube/drift region, used in TOF setup. TOF measurements are used when analysis of particle velocity is required. By pulsing ions towards the sample with a regular
frequency, and observing the time to travel a certain distance after surface impactto a detector, it is possible to calculate the velocity of ions and neutrals coming from the surface. An accelerator may also be used in this setup, prior to the drift tube, in order to achieve separation of ions from neutrals when desired.
electrostatic analyzer, used to detect the velocities and/or energies of scattered particles including ions and, in some cases, neutral species. As opposed to TOF analyzers, electrostatic analyzers achieve ion energy resolution using electrostatic deflectors to direct only ions of a particular energy range into a collector, while all other ions are redirected. This type of analyzer can give good energy resolution (and thus, selectivity) but typically suffers from poor sensitivity due to the fact that it only detects ions of a certain energy range and ignores neutral species altogether. Two types of detectors are used: channel electron multiplier (CEM) and microchannel plate (MCP) detectors. CEMs operate in a similar manner to photomultipliers, displaying a cascade of secondary electron emission processes initiated by ion or fast neutral (energy > 1 keV) impact to give a gainin signal current. In this way it is possible to efficiently detect even small ion or neutral particle fluxes. MCP detectors are essentially 2-dimensional arrays of CEMs, and they allow additional information about particle position to be obtained at the cost of sensitivity at any given position.
Vacuum pumps; Studies are performed in ultra-high vacuum (UHV) conditions (< 10-10 torr) in order to prevent unwanted interference with the ion beam and/or sample. Common UHV pumps include turbomolecular and ion pumps, with roughing pumping typically performed using a rotary vane pump. Due to the extreme surface (i.e. first-layer) sensitivity of LEIS, samples also need to be rigorously cleaned prior to analysis. Some common processes used to clean samples include sputteringand annealing. Appropriate equipment for cleaning must be contained within the vacuum chamber.
# Other analysis tools; in many cases it is desirable to perform multiple types of analysis on a sample within the same UHV system, or even at the same time. Some additional tools may include
Auger electron spectroscopy(AES), low-energy electron diffraction(LEED), and x-ray photoelectron spectroscopy(XPS). Use of these tools typically requires the presence of additional detectors as well as electron and/or x-raysources where applicable.
Physics of ion-surface interactions
Several different types of events may take place as a result of the ion beam impinging on a target surface. Some of these events include electron or photon emission, electron transfer (both ion-surface and surface-ion),
scattering, adsorption, and sputtering(i.e. ejection of atoms from the surface). For each system and each interaction there exists an interaction cross-section, and the study of these cross-sections is a field in its own right. As the name suggests, LEIS is primarily concerned with scattering phenomena.
Elemental composition and two-body collision model
Due to the energy range typically used in ion scattering experiments (> 500 eV), effects of thermal vibrations,
phononoscillations, and interatomic binding are ignored since they are far below this range (~a few eV), and the interaction of particle and surface may be thought of as a classical two-body elastic collisionproblem. Measuring the energy of ions scattered in this type of interaction can be used to determine the elemental composition of a surface, as is shown in the following: Two-body elastic collisions are governed by the concepts of energy and momentum conservation. Consider a particle with mass mx, velocity v0, and energy given as impacting another particle at rest with mass my. The energies of the particles after collision are and where and thus . Additionally, we know. Using trigonometrywe are able to determine
Where and are the atomic numbers of the primary ion and surface atom, respectively, is the
elementary charge, is the interatomic distance, and is the screening function. accounts for the interference of the electrons orbiting each nucleus. In the case of MEIS and RBS, this potential can be used to calculate the Rutherford scattering cross section :
As shown at right, represents a finite region for an incoming particle, while represents the solid scattering angle after the scattering event. However, for LEIS is typically unknown which prevents such a clean analysis. Additionally, when using noble gas ion beams there is a high probability of neutralization on impact (which has strong angular dependence) due to the strong desire of these ions to be in a neutral, closed shell state. This results in poor secondary ion flux. See AISS and TOF-SARS below for approaches to avoiding this problem.
Shadowing and blocking
Shadowing and blocking are important concepts in almost all types of ion-surface interactions and result from the repulsive nature of the ion-nucleus interaction. As shown at right, when a flux of ions flows in parallel towards a scattering center (nucleus), they are each scattered according to the force of the Coulomb repulsion. This effect is known as shadowing. In a simple Coulomb repulsion model, the resulting region of “forbidden” space behind the scattering center takes the form of a
paraboloidwith radiusat a distance L from the scattering center. The flux density is increased near the edge of the paraboloid.
Blocking is closely related to shadowing, and involves the interaction between scattered ions and a neighboring scattering center (as such it inherently requires the presence of at least two scattering centers). As shown, ions scattered from the first nucleus are now on diverging paths as they undergo interaction with the second nucleus. This interaction results in another “shadowing cone” now called a blocking cone where ions scattered from the first nucleus are blocked from exiting at angles below . Focusing effects again result in an increased flux density near .
In both shadowing and blocking, the "forbidden" regions are actually accessible to trajectories when the mass of incoming ions is greater than that of the surface atoms (e.g. Ar+ impacting Si or Al). In this case the region will have a finite but depleted flux density.
For higher energy ions such as those used in MEIS and RBS the concepts of shadowing and blocking are relatively straightforward since ion-nucleus interactions dominate and electron screening effects are insignificant. However, in the case of LEIS these screening effects do interfere with ion-nucleus interactions and the repulsive potential becomes more complicated. Also, multiple scattering events are very likely which complicates analysis. Importantly, due to the lower energy ions used LEIS is typically characterized by large interaction cross-sections and shadow cone radii. For this reason penetration depth is low and the method has much higher first-layer sensitivity than MEIS or RBS. Overall, these concepts are essential for data analysis in impact collision LEIS experiments (see below).
Diffraction does not play a major role
de Broglie wavelengthof ions used in LEIS experiments is given as . Using a worst-case value of 500 eV for an 4He+ ion, we see λ is still only 0.006 Å, still well below the typical interatomic spacing of 2-3 Å. Because of this, the effects of diffractionare not significant in a normal LEIS experiment.
Variations of technique
Depending on the particular experimental setup, LEIS may be used to obtain a variety of information about a sample. The following includes several of these methods.
* Alkali ion scattering spectroscopy (AISS) uses alkali ions in place of noble gas ions to give a distinctly different type of interaction. The primary difference between AISS and normal ISS is the increase in ion survival
probabilitywhen using alkali ions. This is due to the relative stability of alkali (+1) ions as opposed to noble gas ions which have a much stronger energetic incentive for abstracting electrons from the sample. Increasing the ion survival probability results in an increase in ion fluxand an improvement in sensitivity, which in turn allows for a reduction in primary ion flux to a point where the method is almost non-destructive. A disadvantage of using alkali ions in place of noble gas ions is the increased likelihood of adsorptionor deposition to the sample surface.
* Impact-collision ion scattering spectroscopy (ICISS) takes advantage of shadowing and blocking in order to make precise determinations about interatomic spacing of the first 1-2 layers in a surface. The specific scattering geometry (180 degrees) ensures detection of only those particles which have undergone head-on collisions with surface atoms (thereby avoiding the complications of multiple scattering events). Starting sampling at a relatively high
angle of incidenceand scanning over varying incidence angles, the intensityof one particular energy peak is monitored. Scattered ions form shadow cones (see above) behind each atom, which prevents any backscattering at low incidence angles. A peak in scattering intensity is observed when the cones line up such that each passes over the adjacent atom. Performing such an analysis on a sample with known interatomic spacing enables determination of the shape of the shadow cone, where as shown at right, and . If the shape of the shadow cone is known, the interatomic spacing between surface atoms as well as the spacing and directionality between surface and subsurface atoms can then be calculated from the resulting peak-and-valley structure in a graph of intensity versus scattering angle. In the graph at right showing scattering intensity from a subsurface (second layer) atom, corresponds to the middle of the "valley" where the atom is being blocked by a surface atom. and correspond to the peaks due to intersection of the shadow cone with the subsurface atom. Interatomic spacing can be directly calculated from these values if the shape of the shadow cone is known.
* Reactive ion scattering (RIS) utilizes a stream of very low-energy (1-100 eV) Cs+ ions to probe molecules adsorbed at the surface of a sample. Upon impact the ions may interact with and chemically bind to species present at the surface. These interactions take place on a rapid (
picosecond) timescaleand can be used to analyze for the presence of different molecules or molecular fragments by observing spectra of Cs-X+ coming from the surface.
Time-of-flightscattering and recoiling spectroscopy (TOF-SARS) uses the TOF analysis setup. Elemental analysis may be performed via observation of in-plane scattering, while structural information may be obtained by following certain spectral peaks while shifting either sample incident or azimuthal angle.
* Scattering and recoiling imaging spectroscopy (SARIS) takes advantage of blocking cone geometries to focus ions in a manner similar to conventional
optics. This gives very large magnifications (~109) when projected onto a 2-d detector and may be used to give element-specific images of the sample surface. The use of a wide 2-d MCP detector greatly reduces sample analysis time as opposed to the TOF geometry with an inherently narrow-angle detector (see drift tube above). J. Wayne Rabalais at the University of Houstonis one of the pioneers of this method, and a fine image of the output of a SARIS experiment can be found [http://www.chem.uh.edu/Faculty/rabalais/ here] .
Comparison to other analytical techniques
Medium energy ion scattering(MEIS) and Rutherford backscattering (RBS) spectroscopies involve a similar setup to LEIS but use ions in the energy range of ~100 keV (MEIS) and ~1-2 MeV (RBS) to probe surfaces. Surface sensitivity is lost as a result of the use of higher energy particles, so while MEIS and RBS can still provide information about a sample they are incapable of providing true first-layer sensitivity.
Secondary ion mass spectrometry(SIMS) involves the detection of ionic species ejected from a surface as a result of energetic particle impact. While SIMS is capable of giving depth profiles of the elemental composition of a sample, it is an inherently destructive method and is generally does not give structural information.
X-ray photoelectron spectroscopy(XPS) is capable of surface elemental analysis, but samples a much more broad region of a sample than LEIS and so is not able to distinguish the first layer from subsurface layers. Since XPS relies on ejection of core-level electrons from atoms it is unable to detect hydrogenor heliumatoms in a sample.
Low-energy electron diffraction(LEED) is often used in combination with LEIS in order to facilitate proper sample alignment. LEED can give detailed structural information about a sample including surface superstructures and alignment of adsorbates. LEED is not element-specific and so cannot be used to determine surface elemental composition.
Auger electron spectroscopy(AES) involves the detection of electrons emitted as a result of core hole excitation and relaxation processes. Since the process involves core levels it is insensitive to hydrogen and helium atoms. AES results may typically be used to infer information on the chemical environment of particular atoms in a surface.
* [http://www.chem.uh.edu/Faculty/rabalais/ Rabalais] , J. Wayne. A professor of chemistry at the University of Houston.
* [http://www.calipso.nl/ Calipso] , provider of analysis using LEIS. Contains some nice application notes.
* [http://www.iontof.com/ ION-TOF] , provider of instrumentation for high-sensitivity LEIS and TOF-SIMS.
* [http://www.kratosanalytical.net/ Kratos] , provider of a variety of instruments for surface analysis including AES, ISS, and XPS. Includes a discussion of a variety of applications in surface analysis.
* [http://www.omicron.de/ Omicron NanoTechnology] , the world’s leading supplier of solutions for analytic requirements under UHV-conditions in the fields of Surface Science and Nanotechnology. Includes interesting Applications, Publications and Techniques.
List of materials analysis methods
Wikimedia Foundation. 2010.
Look at other dictionaries:
Low-energy electron diffraction — (LEED) is a technique used to characterize the structures of surfaces.History =Davisson and Germer s discovery of electron diffraction= The development of electron diffraction was closely linked to the progress of quantum mechanics and atomic… … Wikipedia
Helium atom scattering — (HAS) is a surface analysis technique used in materials science. HAS provides information about the surface structure and lattice dynamics of a material by measuring the diffracted atoms from a monochromatic helium beam incident on the sample.… … Wikipedia
Carrier scattering — Defect types include atom vacancies, adatoms, steps, and kinks which occur most frequently at surfaces due to finite material size causing crystal discontinuity. What all types of defects have in common, whether they be surface or bulk, is that… … Wikipedia
Reflection high energy electron diffraction — (RHEED) is a technique used to characterize the surface of crystalline materials. RHEED systems gather information only from the surface layer of the sample, which distinguishes RHEED from other materials characterization methods that rely also… … Wikipedia
Static secondary ion mass spectrometry — Static secondary ion mass spectrometry, or static SIMS is a technique for chemical analysis including elemental composition and chemical structure of the uppermost atomic or molecular layer of a solid which may be a metal, semiconductor or… … Wikipedia
Threshold displacement energy — The threshold displacement energy Td is the minimum kinetic energy that an atom in a solid needs to be permanently displaced from its lattice site to a defect position. It is also known as displacement threshold energy or just displacement energy … Wikipedia
Raman scattering — or the Raman effect (pronounced: IPA| [rə.mən] ) is the inelastic scattering of a photon. Discovered By Dr. C.V. Raman in liquids and by Grigory Landsberg and Leonid Mandelstam in crystals.When light is scattered from an atom or molecule, most… … Wikipedia
Self-energy — In theoretical physics and quantum field theory a particle s self energy Σ represents the contribution to the particle s energy, or effective mass, due to interactions between the particle and the system it is part of. For example, in… … Wikipedia
Ultra-high-energy cosmic ray — Unsolved problems in physics Why is it that some cosmic rays appear to possess energies that are theoretically too high? In high energy physics, an ultra high energy cosmic ray (UHECR) or extreme energy cosmic ray (EECR) is a cosmic ray with an… … Wikipedia
LEIS — low energy ion scattering … Medical dictionary