- Laboratory centrifuge
A tabletop laboratory centrifuge
Uses Separation Related items Gas centrifuge
A laboratory centrifuge is a piece of laboratory equipment, driven by a motor, which spins liquid samples at high speed. There are various types of centrifuges, depending on the size and the sample capacity.
There are various types of centrifugation:
- Differential centrifugation, often used to separate certain organelles from whole cells for further analysis of specific parts of cells
- Isopycnic centrifugation, often used to isolate nucleic acids such as DNA
- Sucrose gradient centrifugation, often used to purify enveloped viruses and ribosomes, and also to separate cell organelles from crude cellular extracts
There are different types of laboratory centrifuges:
(devices for small tubes from 0.2 ml to 2.0 ml (micro tubes), up to 96 well-plates, compact design, small footprint; up to 30.000 g)
- Clinical centrifuges
(devices used for clinical applications like blood collection tubes, low-speed devices)
- Multipurpose benchtop centrifuges
(devices for a broad range of tube sizes, high variability, big footprint)
- Stand alone centrifuges
(heavy devices like the ultracentrifuge)
Many centrifuges are available with (refrigerated device) or without cooling function. There are different providers of laboratory centrifuges like Eppendorf, Thermo-Heraeus, Thermo-Sorvall, Hettich, Beckmann-Coulter, MSE, Sigma and Awel.
Centrifuge tubes or centrifuge tips are tapered tubes of various sizes made of glass or plastic. They may vary in capacity from tens of millilitres, to much smaller capacities used in microcentrifuges used extensively in molecular biology laboratories. The most commonly encountered tubes are of about the size and shape of a normal test tube (~ 10 cm long). Microcentrifuges typically accommodate microcentrifuge tubes with capacities from 250 μl to 2.0 ml. These are exclusively made of plastic.
Glass centrifuge tubes can be used with most solvents, but tend to be more expensive. They can be cleaned like other laboratory glassware, and can be sterilized by autoclaving. Plastic centrifuge tubes, especially microcentrifuge tubes tend to be less expensive. Water is preferred when plastic centrifuge tubes are used. They are more difficult to clean thoroughly, and are usually inexpensive enough to be considered disposable.
The load in a laboratory centrifuge must be carefully balanced. This is achieved by using a combination of samples and balance tubes which all have the same weight or by using various balancing patterns without balance tubes. Small differences in mass of the load can result in a large force imbalance when the rotor is at high speed. This force imbalance strains the spindle and may result in damage to the centrifuge or personal injury. Some centrifuges have an automatic rotor imbalance detection feature that immediately discontinues the run when an imbalance is detected.
Before starting a centrifuge, an accurate check of the rotor and lid locking mechanisms is mandatory. Centrifuge rotors should never be touched while moving, because a spinning rotor can cause serious injury. Modern centrifuges generally have features that prevent accidental contact with a moving rotor as the main lid is locked during the run.
Centrifuge rotors have tremendous kinetic energy during high speed rotation. Rotor failure, caused by mechanical stress from the high forces imparted by the motor, can occur due to manufacturing defects, routine wear and tear, or improper use and maintenance. Such a failure can be catastrophic failure, especially with larger centrifuges, and generally results in total destruction of the centrifuge. While centrifuges generally have safety shielding to contain these failures, such shielding may be inadequate, especially in older models, or the entire centrifuge unit may be propelled from its position, resulting in damage to nearby personnel and equipment. Uncontained rotor failures have shattered laboratory windows and destroyed refrigerators and cabinetry. To reduce the risk of rotor failures, centrifuge manufactures specify operating and maintenance procedures to ensure that rotors are regularly inspected and removed from service or derated (only operated at lower speeds) when they are past their expected lifetime.
Another potential hazard is the aerosolization of hazardous samples during centrifugation. To prevent contamination of the laboratory, rotor lids with special aerosol-tight gaskets are available. The rotor can be loaded with the samples within a hood and the rotor lid fixed on the rotor. Afterwards, the aerosol-tight system of rotor and lid is transferred to the centrifuge. The rotor can then be fixed within the centrifuge without opening the lid. After the run, the entire rotor assembly, including the lid, is removed from the centrifuge to the hood for further steps, maintaining the samples within a closed system.
Protocols for centrifugation typically specify the amount of acceleration to be applied to the sample, rather than specifying a rotational speed such as revolutions per minute. The acceleration is often quoted in multiples of g, the acceleration due to gravity at the Earth's surface. This distinction is important because two rotors with different diameters running at the same rotational speed will subject samples to different accelerations.
Relative centrifugal force (RCF) is the measurement of the force applied to a sample within a centrifuge. This can be calculated from the speed (RPM) and the rotational radius (cm) using the following calculation.
- g = RCF = 0.00001118 × r × N2
- g = Relative centrifuge force
- r = rotational radius (centimetre, mm)
- N = rotating speed (revolutions per minute, r/min)
To avoid having to perform a mathematical calculation every time, one can find nomograms for converting RCF to rpm for a rotor of a given radius. A ruler or other straight edge lined up with the radius on one scale, and the desired RCF on another scale, will point at the correct rpm on the third scale. Example Based on automatic rotor recognition, up to date centrifuges have a button for automatic conversion from RCF to rpm and vice versa.
- ^ Frothingham, R (February 1999). "Centrifugation without a balance tube". American Laboratory (Shelton, CT: International Scientific Communications) 31 (19): 10. ISSN 0044-7749. http://www.stanford.edu/dept/EHS/prod/researchlab/lab/centrifuge.html. Retrieved March 18, 2011.
- ^ Cornell University EH&S. "Centrifuge Accident". http://www.ehs.cornell.edu/chem_lab_safety/centrifuge_safe.cfm. Retrieved 29 December 2010.
- RCF Calculator and Nomograph
- Centrifugation Rotor Calculator
- Selection of historical centrifuges in the Virtual Laboratory of the Max Planck Institute for the History of Science
Laboratory equipment Glassware
Beaker · Boston round (bottle) · Büchner funnel · Burette · Cold finger · Condenser · Conical measure · Cuvette · Dean-Stark apparatus · Dropping funnel · Eudiometer · Evaporating dish · Gas syringe · Graduated cylinder · Pipette · Petri dish · Pycnometer · Separatory funnel · Soxhlet extractor · Watch glass
Agar plate · Aspirator · Autoclave · Biosafety cabinet · Bunsen burner · Calorimeter · Chemostat · Colony counter · Colorimeter · Laboratory centrifuge · Crucible · Eyewash · Fire blanket · Fume hood · Glove box · Homogenizer · Hot air oven · Incubator · Laminar flow cabinet · Magnetic stirrer · Meker-Fisher burner · Microscope · Microtiter plate · Picotiter plate · Plate reader · Retort stand · Safety shower · Spectrophotometer · Static mixer · Stir bar · Stirring rod · Scoopula · Teclu burner · Thermometer · Vacuum dry box · Vortex mixer · Wash bottleSee also Instruments used in medical laboratories
Wikimedia Foundation. 2010.
Look at other dictionaries:
Centrifuge — This article is about the scientific device. For the Christian camp, see Centrifuge (camps). For spin direction in quantum mechanics, see Spin (physics)#Spin direction. A laboratory tabletop centrifuge. A centrifuge is a piece of equipment,… … Wikipedia
Laboratory glassware — Three beakers, a conical flask, a graduated cylinder and a volumetric flask … Wikipedia
laboratory apparatus — ▬▬▬▬▬▬▬▬ Laboratory apparatus includes: autoclave beaker bell jar boiling tube Büchner funnel Bunsen burner burette centrifuge clamp condenser conical flask crucible cylinder desiccator distillation apparatus dropper electron microscope eva … Useful english dictionary
Centrifuge Accommodations Module — ISS Centrifuge Accommodations Module The Centrifuge Accommodations Module (CAM) is a cancelled element of the International Space Station. Although the module was planned to contain more than the a centrifuge, the 2.5 m (8.2 ft)… … Wikipedia
Medical laboratory scientist — MLS in his work environment. A Medical Laboratory Scientist (MLS) (also referred to as medical technologist or clinical laboratory technologist) is a healthcare professional who performs chemical, hematological, immunologic, microscopic, and… … Wikipedia
Condenser (laboratory) — Two different condensers are seen here: an air cooled fractionating column on left, and a Liebig condenser on the right In a laboratory a condenser is a piece of laboratory glassware used to cool hot vapors or liquids.  A condenser usually… … Wikipedia
Medical laboratory — Laboratory investigation redirects here. For the journal, see Laboratory Investigation (journal). Clinical laboratory in a Hospital setting showing several automated analysers. A medical laboratory or clinical laboratory is a laboratory where… … Wikipedia
Idaho National Laboratory — Coordinates: 43°32′00″N 112°56′41″W / 43.533429°N 112.94486°W / 43.533429; 112.94486 … Wikipedia
Materials Science Laboratory — MSRR 1 (NASA) The Materials Science Laboratory (MSL) of the European Space Agency is a payload on board the International Space Station for materials science experiments in low gravity. It is installed in NASA s first Materials Science Research… … Wikipedia
Compresseur centrifuge — Photographie de la coupe d un turboréacteur Goblin (fabriqué par de Havilland au lendemain de la Seconde Guerre mondiale) montrant un compresseur centrifuge. Le terme « compresseur centrifuge » (aussi appelé « compresseur… … Wikipédia en Français