- Scanning capacitance microscopy
Scanning capacitance microscopy (SCM) is a variety of
scanning probe microscopyin which a narrow probe electrode is held just above the surface of a sample and scanned across the sample. SCM characterizes the surface of the sample using information obtained from the change in electrostatic capacitance between the surface and the probe.
More precisely SCM uses an ultra-sharp conducting probe (often Pt/Ir or Co/Cr metal covering an etched silicon probe) to form a metal-insulator-semiconductor (MIS/MOS) capacitor with a semiconductor sample if a native oxide is present. When no oxide is present, a shottky capacitor is formed. When the probe and surface are in contact, an AC bias is applied, generating capacitance variations in the sample which can be detected using a GHz resonant capacitance sensor. The tip is then scanned across the semiconductor's surface in 2D while the tip's height is controlled by conventional contact force feedback .
By applying an alternating bias to the metal-coated probe, carriers alternately accumulate and deplete within the semiconductor’s surface layers, changing the tip-sample capacitance. The magnitude of this change in capacitance with the applied voltage gives information about the concentration of carriers (SCM amplitude data), whereas the difference in phase between the capacitance change and the applied, alternating bias carries information about the sign of the charge carriers (SCM phase data). Due to the fact that SCM functions even through an insulating layer, a finite conductivity is not required to measure the electrical properties.
Limit of resolution of SCM
On the conducting surfaces, the limit of resolution at the scale of 2nm is estimatedcite journal
title=The resolution limit of scanning capacitance microscopes
doi=10.1088/0022-3727/36/5/326] .For the high resolution, the quick analysis of capacitance of a capacitor with rough electrode is requiredcite journal
title=The lateral resolution limit for imaging periodic conducting surfaces in capacitive microscopy
doi=10.1088/0022-3727/33/22/305] cite journal
title=Rough-surface capacitor: approximations of the capacitance with elementary functions
doi=10.1088/0022-3727/32/20/317] .The limit of resolution of SCM seems to be an order of magnitude better than that estimated for the
atomic nanoscope;however, as other kinds of the probe microscopy, SCM requires careful preparation of the analyzed surface,which is supposed to be almost flat.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Scanning probe microscopy — Part of a series of articles on Nanotechnology … Wikipedia
Scanning voltage microscopy — (SVM) sometimes also called nanopotentiometry is a scientific experimental technique based on atomic force microscopy. A conductive probe, usually only a few nanometers wide at the tip, is placed in full contact with an operational electronic or… … Wikipedia
Scanning tunneling spectroscopy — (STS) is a powerful experimental technique in scanning tunneling microscopy (STM) that uses a scanning tunneling microscope (STM) to probe the local density of electronic states (LDOS) and band gap of surfaces and materials on surfaces at the… … Wikipedia
Scanning tunneling microscope — Image of reconstruction on a clean Gold(100) surface … Wikipedia
Photoconductive atomic force microscopy — (pc AFM) is a scientific technique.. Multi layer photovoltaic cells have gained popularity since mid 1980s. At the time, research was primarily focused on single layer photovoltaic (PV) devices between two electrodes, in which PV properties… … Wikipedia
Near-field scanning optical microscope — Near field scanning optical microscopy (NSOM/SNOM) is a microscopic technique for nanostructure investigation that breaks the far field resolution limit by exploiting the properties of evanescent waves. This is done by placing the detector very… … Wikipedia
Magnetic resonance force microscopy — (MRFM) is an imaging technique that acquires magnetic resonance images (MRI) at nanometer scales, and possibly at atomic scales in the future. MRFM is potentially able to observe protein structures which cannot be seen using X ray crystallography … Wikipedia
Feature-oriented scanning — (FOS) is a method intended for high precision measurement of nanotopography as well as other surface properties and characteristics on a scanning probe microscope (SPM) using features (objects) of the surface as reference points of the… … Wikipedia
Counter-scanning — (CS) is a method for measuring surface topography with a scanning probe microscope enabling correction of raster distortions resulted from drift of the microscope probe relative to the surface being measured. Two surface scans, viz. direct… … Wikipedia
SCM — • scanning capacitance microscopy; • Schwann cell membrane; • sensation, circulation, and motion; • single chip microcomputer; • Society of Computer Medicine; • soluble cytotoxic medium; • spleen cell conditioned medium; • split cord… … Dictionary of medical acronyms & abbreviations