- Proportional counter
A proportional counter is a measurement device to count particles of
ionizing radiationand measure their energy.
A proportional counter is a type of
Gaseous ionization detector- it works on the same principle as the Geiger-Müller counter, but uses a lower operating voltage. An incoming ionizing particle, if it has sufficient energy, liberates electrons from the atomic orbitals of the gas atoms (see ionization potential), leaving an electron and positively charged atom, commonly known as an ion pair. As the charged particle travels through the chamber it leaves a trail of ion pairs along its trajectory. The electrons created in this process drift toward a readout electrode, known as the anode, under the influence of an applied electric field. At the same time, the positive ions drift towards the cathode, at much lower speed; in practical devices, the drift times are measured in microseconds and milliseconds, respectively.
A proportional counter differs from an
ionization chamberin that the operating voltage is sufficiently high that the drifting electrons gain enough energy over a mean free pathto create further ion pairs when they collide with other neutral atoms of the gas. The electrons created in these new events also drift toward the readout electrode and can create further ion pairs themselves. In this manner, a cascade of ion pairs can be created, this is known as a Townsend avalanche. If the operating voltage is chosen carefully, each avalanche process occurs independently of other avalanches which derive from the same initial ionizing event. Therefore, even though the total number of electrons liberated can increase exponentially with distance, the total amount of charge created remains proportional to the amount of charge liberated in the original event.
By measuring the total charge (time
integralof the electric current) between the electrodes, we can find out the particle's kinetic energy, because the number of ion pairs created by the incident ionizing charged particle is proportional to its energy.
The geometry of the electrodes and the voltages on them are chosen such that in most of the volume of the counter the electric field strength is not enough to produce a Townsend avalanche. The electrons just drift until they get close to the anode, where a strong field allows avalanche multiplication to occur. In this way each electron is multiplied by approximately the same factor (up to about a million) independent of the distance it has covered in the low-field 'drift region'. If the field strength everywhere is below a critical value, Townsend avalanches do not occur at all, and the detector operates as an ionization chamber. If the voltage (and therefore the field strength) is too high, the degree of charge amplification tends to a maximum value, and all pulses from the chamber have the same amplitude, so the detector operates as a
This process of charge amplification can improve the
signal-to-noise ratioof the detector and also reduce the amount of amplification required from external electronics. The proportionality between the energy of the charged particle travelling through the chamber and the total charge created makes proportional counters useful for charged particle spectroscopy. The energy resolution of a proportional counter, however, is limited because both the initial ionization event and the subsequent 'multiplication' event are subject to statistical fluctuations.
Proportional counters are also useful for detection of high energy
photons, such as X-rays or gamma-rays, provided these can penetrate the entrance window.
* US patent|3092747, S. Fine, "Proportional counter"
* US patent|2499830, E. W. Molloy, "Air proportional counter"
author=G.Charpak and F.Sauli
title=High-resolution Electronic Particle Detectors
journal=Annual review of Nuclear Science
publisher=Annual Reviews Inc.
* E. Mathieson, Induced charge distributions in proportional detectors, http://www.inst.bnl.gov/programs/gasnobledet/publications/Mathieson's_Book.pdf
Gaseous ionization detectors
Wikimedia Foundation. 2010.
Look at other dictionaries:
proportional counter — noun counter tube whose output pulse is proportional to number of ions produced • Syn: ↑proportional counter tube • Hypernyms: ↑counter tube • Hyponyms: ↑boron counter tube … Useful english dictionary
Proportional counter — A device for measuring radioactivity. More technically, a proportional counter counts the emission of radioactive particles which permits the discrimination between particles or rays of different energies. * * * a gas filled radiation detection… … Medical dictionary
proportional counter — proporcingasis skaitiklis statusas T sritis automatika atitikmenys: angl. proportional counter vok. Proportionalzähler, m rus. пропорциональный счетчик, m pranc. compteur proportionnel, m … Automatikos terminų žodynas
proportional counter — proporcingasis skaitiklis statusas T sritis fizika atitikmenys: angl. proportional counter vok. Proportionalzähler, m; Proportionalzählrohr, n rus. пропорциональный счётчик, m pranc. compteur proportionnel, m … Fizikos terminų žodynas
proportional counter — Physics. a radiation counter in which the strength of each electric pulse generated per count is proportional to the energy of the particle or photon producing the pulse, alpha particles producing a different electric pulse from beta rays. [1935… … Universalium
proportional counter tube — noun counter tube whose output pulse is proportional to number of ions produced • Syn: ↑proportional counter • Hypernyms: ↑counter tube • Hyponyms: ↑boron counter tube … Useful english dictionary
multiwire proportional counter — daugialaidis proporcingasis skaitiklis statusas T sritis radioelektronika atitikmenys: angl. multiwire proportional counter vok. Mehrdrahtproportionalzähler, m rus. пропорциональный многопроволочный счётчик, m pranc. compteur proportionnel aux… … Radioelektronikos terminų žodynas
counter tube — noun a measuring instrument for counting individual ionizing events • Hypernyms: ↑measuring instrument, ↑measuring system, ↑measuring device • Hyponyms: ↑crystal counter, ↑Geiger counter, ↑Geiger Muller counter, ↑ … Useful english dictionary
counter — A device that counts, usually scintillations. automated differential leukocyte c. an instrument using digital imaging or cytochemical techniques to differentiate leukocytes. electronic cell c. an automatic blood cell c. in which cells passing… … Medical dictionary
boron counter tube — noun a proportional counter tube for counting neutrons • Hypernyms: ↑proportional counter tube, ↑proportional counter • Part Meronyms: ↑boron chamber … Useful english dictionary