- Thin filament pyrometry
Thin Filament Pyrometry (TFP) is an optical method used to measure temperatures. It involves the placement of a thin filament in a hot gas stream. Radiative emissions from the filament can be correlated with filament temperature. Filaments are typically
Silicon carbide( SiC) fibers with a diameter of 15 micrometres. Temperatures of about 800 - 2500 K can be measured.
TFP was first used by V. Vilimpoc and L.P. Goss (1988). A recent paper using TFP is Maun et al. (2007).
The typical TFP apparatus consists of a
flameor other hot gas stream, a filament, and a camera.
TFP has several advantages, including the ability to simultaneously measure temperatures along a line and minimal intrusiveness. Most other forms of
pyrometryare not capable of providing gas-phase temperatures.
Calibration is required. Calibration typically is performed with a
thermocouple. Both thermocouples and filaments require corrections in estimating gas temperatures from probe temperatures. Also, filaments are fragile and typically break after about an hour in a flame.
The primary application is to
* L.G. Blevins, M.W. Renfro, K.H. Lyle, N.M. Laurendeau, J.P. Gore, [http://dx.doi.org/10.1016/S0010-2180(99)00023-1 Experimental study of temperature and CH radical location in partially premixed CH4/air coflow flames ] , Combustion and Flame 118 (4) 684-696 (1999).
* J.D. Maun, [https://drum.umd.edu/dspace/bitstream/1903/3602/1/umi-umd-3450.pdf Thin-Filament Pyrometry With a Digital Still Camera] , M.S. Thesis, University of Maryland (2006).
* J.D. Maun,
Peter B. Sunderland, D.L. Urban, Applied Optics, 46:483-488 (2007).
* W.M. Pitts, Proceedings of the Combustion Institute 26:1171-1179 (1996).
* V. Vilimpoc, L.P. Goss, Proceedings of the Combustion Institute 22:1907-1914 (1988).
Wikimedia Foundation. 2010.
Look at other dictionaries:
Pyrometry — is the non contact measurement of the temperature of an object, by measuring its self emission and emissivity. An apparatus that measures the temperature by means of pyrometry is called a pyrometer or optical pyrometer.Emissivity is a measure of… … Wikipedia
Pyrometer — An optical pyrometer A sailor checking the temperature of a ventilation s … Wikipedia
Silicon carbide — Chembox new Name = Silicon carbide ImageFile = Silicon carbide 3D balls.png ImageSize = 140px ImageName = Ball and stick model of part of a crystal of SiC ImageFile1 = silicon carbide detail.jpg ImageSize1 = 140px OtherNames = Section1 = Chembox… … Wikipedia
Карбид кремния — Карбид кремния … Википедия
Combustion — This article is about the process of burning. For combustion without external ignition, see spontaneous combustion. For the engine used in mobile propulsion, see internal combustion engine. For the visible part of a fire, see flame. Burning… … Wikipedia
TFP — may stand for:* The Food Project * thin filament pyrometry * time for print * American Society for the Defense of Tradition, Family and Property * total functional programming * total factor productivity * Transference focused psychotherapy *… … Wikipedia
Peter B. Sunderland — is Assistant Professor of Fire Protection Engineering at the University of Maryland, College Park. He earned a Bachelor s Degree in Mechanical Engineering at Cornell University, a Master s Degree in Mechanical Engineering from the University of… … Wikipedia