- Rotary variable differential transformer
A rotary variable differential transformer (RVDT) is a type of electrical
transformerused for measuring angular displacement.
More precisely, a Rotary Variable Differential Transformer (RVDT) is an electromechanical transducer that provides a variable alternating current (AC) output voltage that is linearly proportional to the angular displacement of its input shaft. When energized with a fixed AC source, the output signal is linear within a specified range over the angular displacement.
RVDT’s utilize brushless, non-contacting technology to ensure long-life and reliable, repeatable position sensing with infinite resolution. Such reliable and repeatable performance assures accurate position sensing under the most extreme operating conditions.
Most RVDT are composed of a wound,
laminated statorand a salienttwo-pole rotor. The stator, containing four slots, contains both the primary windingand the two secondary windings. Some secondary windings may also be connected together.
Operation of RVDT's
The two induced
voltages of the secondary windings, and , varies lineary to the mechanical angle of the rotor, θ:
where is the
gainor sensitivity. The second voltage can be reverse determined by:
The difference gives a proportional voltage:
and the sum of the voltages is a constant:
This constant gives the RVDT great stability of the angular information, independence of the input voltage or
frequency, or temperature, and enables it to also detect a malfunction.
Putting the above mathematical equations in some theorotical form, the working of RVDT can be explanied as below :-
Basic RVDT construction and operation is provided by rotating an iron-core bearing supported within a housed stator assembly. The housing is passivated stainless steel. The stator consists of a primary excitation coil and a pair of secondary output coils.A fixed alternating current excitation is applied to the primary stator coil that is electromagnetically coupled to the secondary coils. This coupling is proportional to the angle of the input shaft. The output pair is structured so that one coil is in-phase with the excitation coil, and the second is 180 degrees out-of-phase with the excitation coil.When the rotor is in a position that directs the available flux equally in both the in-phase and out-of-phase coils, the output voltages cancel and result in a zero value signal. This is referred to as the electrical zero position or E.Z. When the rotor shaft is displaced from E.Z., the resulting output signals have a magnitude and phase relationship proportional to the direction of rotation.Because RVDT’s perform essentially like a transformer, excitation voltages changes will cause directly proportional changes to the output (transformation ratio). However, the voltage out to excitation voltage ratio will remain constant. Since most RVDT signal conditioning systems measure signal as a function of the transformation ratio (TR), excitation voltage drift beyond 7.5% typically has no effect on sensor accuracy and strict voltage regulation is not typically necessary. Excitation frequency should be controlled within +/- 1% to maintain accuracy
Although the RVDT can theoretically operate between ±45°,
accuracydecreases quickly after ±35°. Thus, it operational limits lies mostly within ±30°, but some up to ±40°. Certain types can operate up to ±60°.
The advantages of the RVDT are :
* low sensitivity to temperature, primary voltage & frequency variations
* low cost
* simple control electronics
* small size
An RVDT can also be designed with two laminations, one containing the primary and the other, the secondaries. These types can operate on larger rotations.
A similar transformer is called the Rotary Variable Transformer and contains only one secondary winding giving only one voltage:
LVDT, the RVDT function in the longitudinal position.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Rotary Variable Differential Transformer — Un RVDT (de l anglais Rotary Variable Differential Transformer) est un capteur électrique actif (inductif) de déplacements de rotation. Le débattement est en général limité jusqu à +/ 40°. Principe Un RVDT est composé le plus souvent d un paquet… … Wikipédia en Français
Rotary encoder — A rotary encoder, also called a shaft encoder, is an electro mechanical device used to convert the angular position of a shaft or axle to an analog or digital code, making it an angle transducer. These devices are used in industrial controls,… … Wikipedia
Variable capacitor — Rotary variable capacitor A variable capacitor (also known as a variable air condenser ) is a capacitor whose capacitance may be intentionally and repeatedly changed mechanically or electronically. Variable capacitors are often used in L/C… … Wikipedia
RVDT — Rotary Variable Differential Transformer Un RVDT (de l anglais Rotary Variable Differential Transformer) est un capteur électrique actif (inductif) de déplacements de rotation. Le débattement est en général limité jusqu à +/ 40°. Principe Un RVDT … Wikipédia en Français
RVDT — Rotary Variable Differential Transformer Contributor: GSFC … NASA Acronyms
Electronic component — Various components An electronic component is a basic electronic element and may be available in a discrete form having two or more electrical terminals (or leads). These are intended to be connected together, usually by soldering to a printed… … Wikipedia
Abkürzungen/Luftfahrt/L–R — Dies ist der vierte Teil der Liste Abkürzungen/Luftfahrt. Liste der Abkürzungen Teil 1 A A Teil 2 B–D B; C; D Teil 3 E–K … Deutsch Wikipedia
List of sensors — * Accelerometer * Touch sensor * Active pixel sensor * Air flow meter * Alarm sensor * Bedwetting alarm * Bhangmeter * Biochip * Biosensor * Breathalyzer * Capacitance probe * Carbon paste electrode * Carbon monoxide detector * Catadioptric… … Wikipedia
Position sensor — A position sensor is any device that enables position measurement. It can either be an absolute position sensor or a relative one (displacement sensor). Position sensors can be either linear or angular.Some position sensors available today: *… … Wikipedia
List of Tesla patents — Below is a list of Tesla patents. Dr. Nikola Tesla was an inventor who obtained around 300 patents [Snezana Sarbo, [http://www.tesla symp06.org/papers/Tesla Symp06 Sarboh.pdfNikola Tesla s Patents] , Sixth International Symposium Nikola Tesla,… … Wikipedia