- Scanning ion-conductance microscopy
The scanning ion-conductance microscope (SICM) consists of an electrically charged glass micro- or nanopipette probe filled with electrolyte lowered toward the surface of the sample (which is non-conducting for ions) in an oppositely charged bath of electrolyte. As the tip of the micropipette approaches the sample, the ion conductance and therefore current decreases since the gap through which ions can flow, is reduced in size. Variations in the ion current are measured by an amplifier, and are used as a feedback signal by a scanner control unit to keep the distance between pipette tip and sample constant by applying corresponding voltages to the Z-piezo drive during the scanning procedure. Therefore, the path of the tip follows the contours of the surface.
The SICM can also sample and map the local ion currents above the surface. This is useful in imaging ion currents through membrane channels.
This scanning technique can be used on living tissues, whereas traditional SEM imaging necessitated killing the sample. Biological processes can now be observed whilst in action.
SICM is part of the larger family of
scanning probe microscopy(SPM), and was specially designed for the sub-micrometer resolution scanning of soft non-conductive materials that are bathed in electrolyte solution.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Scanning probe microscopy — Part of a series of articles on Nanotechnology … Wikipedia
Scanning voltage microscopy — (SVM) sometimes also called nanopotentiometry is a scientific experimental technique based on atomic force microscopy. A conductive probe, usually only a few nanometers wide at the tip, is placed in full contact with an operational electronic or… … Wikipedia
Scanning tunneling microscope — Image of reconstruction on a clean Gold(100) surface … Wikipedia
Near-field scanning optical microscope — Near field scanning optical microscopy (NSOM/SNOM) is a microscopic technique for nanostructure investigation that breaks the far field resolution limit by exploiting the properties of evanescent waves. This is done by placing the detector very… … Wikipedia
Magnetic resonance force microscopy — (MRFM) is an imaging technique that acquires magnetic resonance images (MRI) at nanometer scales, and possibly at atomic scales in the future. MRFM is potentially able to observe protein structures which cannot be seen using X ray crystallography … Wikipedia
Feature-oriented scanning — (FOS) is a method intended for high precision measurement of nanotopography as well as other surface properties and characteristics on a scanning probe microscope (SPM) using features (objects) of the surface as reference points of the… … Wikipedia
Counter-scanning — (CS) is a method for measuring surface topography with a scanning probe microscope enabling correction of raster distortions resulted from drift of the microscope probe relative to the surface being measured. Two surface scans, viz. direct… … Wikipedia
List of materials analysis methods — List of materials analysis methods: Contents: Top · 0–9 · A B C D E F G H I J K L M N O P Q R S T U V W X Y Z μSR see Muon spin spectroscopy … Wikipedia
Microscope — This article is about microscopes in general. For light microscopes, see optical microscope. Microscope Us … Wikipedia
Optical microscope — Microscope Uses Small sample observation Notable experiments Discovery of cells Inventor Hans Lippershey Zacharias Jans … Wikipedia