- Total ionic strength adjustment buffer
A Total Ionic Strength Adjustment Buffer ("TISAB") is a
buffer solutionwhich increases the ionic strengthof a solution to a relatively high level. This is important for potentiometric measurements, including ion selective electrodes, because they measure the activity of the analyte rather than its concentration. TISAB essentially masks most chemical interferences in the analyte solution and hence increases the accuracy of the reading.
TISAB is very commonly applied to fluoride ion analysis such as in fluoride ion selective electrodes.
There are four main constituents to TISAB, namely CDTA (cyclohexylenedinitrilotetraacetate), sodium hydroxide, sodium chloride and acetic acid (ethanoic acid), which are all dissolved in deionised water. Hence, TISAB having a specific gravity (density) ~1.0. Each constituent plays an important role to control the ionic strength and pH of the analyte solution, which may otherwise cause error and inaccuracy.
The activity of a substance in solution depends on the product of its concentration and the activity coefficient in that solution. The activity coefficient depends on the ionic strength of the solution in which the potentiometric measurements are made. This can be calculated for dilute solutions using the
Debye-Hückel equation; for more concentrated solutions other approximations must be used. In most cases, the analyst's goal is simply to make sure that the activity coefficient is constant across a set of solutions.
Example: An ion-selective electrode might be calibrated using dilute solutions of the analyte in distilled water. If this calibration is used to calculate the concentration of the analyte in sea water (high ionic strength), significant error is introduced by the difference between the activity of the analyte in the dilute solutions and the concentrated sample. This can be avoided by adding a small amount of ionic-strength buffer to the standards, so that the activity coefficients match more closely.
Adding a TISAB buffer to increase the ionic strength of the solution helps to "fix" the ionic strength at a stable level, making a linear correlation between the logarithm of the concentration of analyte and the measured voltage. By also adding the TISAB buffer to the samples from which the potentiometric equipment are calibrated, the linear correlation can be used to calculate the concentration of analyte in the solution.
:where is measured voltage, is the gas constant, the temperature measured in Kelvin, is the Faraday constant and the charge of the analyte. is the concentration of analyte.
TISAB buffers often include chelators which bind ions that could otherwise interfere with the analyte.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Ion selective electrode — An ion selective electrode (ISE), also known as a specific ion electrode (SIE), is a transducer (or sensor) that converts the activity of a specific ion dissolved in a solution into an electrical potential, which can be measured by a voltmeter or … Wikipedia
буферный раствор для корректировки общей ионной силы — — [А.С.Гольдберг. Англо русский энергетический словарь. 2006 г.] Тематики энергетика в целом EN total ionic strength adjustment buffer solution … Справочник технического переводчика
Acid dissociation constant — Acetic acid, a weak acid, donates a proton (hydrogen ion, high … Wikipedia
Chemical equilibrium — In a chemical reaction, chemical equilibrium is the state in which the concentrations of the reactants and products have not yet changed with time. It occurs only in reversible reactions, and not in irreversible reactions. Usually, this state… … Wikipedia