- Tunable diode laser absorption spectroscopy
Tunable diode laser absorption spectroscopy (TDLAS) is a technique for measuring the concentration of certain species such as
methane, water vaporand many more, in a gaseous mixture using tunable diode lasers and laser absorption spectrometry. The advantage of TDLAS over other techniques for concentration measurement is its ability to achieve very low detection limits (of the order of ppbv). Apart from concentration, it is also possible to determine the temperature, pressure, velocity and mass flux of the gas under observation. [D. T. Cassidy and J. Reid, "Atmospheric pressure monitoring of trace gases using tunable diode lasers," Appl. Opt. 21(7), 1185 - 1190 (1982)] [P. Werle, F. Slemr, K. Maurer, R. Kormann, R. Mucke, and B. Janker, "Near- and mid-infrared laser-optical sensors for gas analysis," Opt. Las. Eng. 37(2-3), 101-114 (2002)] TDLAS is by far the most common laser based absorption technique for quantitative assessments of species in gas phase.
A basic TDLAS setup consists of tunable diode laser light source, transmitting (i.e. beam shaping) optics, optically accessible absorbing medium, receiving optics and detector/s. The emission wavelength of the tunable diode laser, viz. VCSEL, DFB, etc., is tuned over the characteristic absorption lines of a species in the gas in the path of the laser beam. This causes a reduction of the measured signal intensity, which can be detected by a
photodiode, and then used to determine the gas concentration and other properties as described later.
Different diode lasers are used based on the application and the range over which tuning is to be performed. Typical examples are InGaAsP/InP (tunable over 900 nm to 1.6 µm), InGaAsP/InAsP (tunable over 1.6 µm to 2.2 µm), etc. These lasers can be tuned by either adjusting their temperature or by changing injection current density into the gain medium. While temperature changes allow tuning over 100 cm−1, it is limited by slow tuning rates (a few hertz), due to the thermal inertia of the system. On the other hand, adjusting the injection current can provide tuning at rates as high as ~10 GHz, but it is restricted to a smaller range (about 1 to 2 cm−1) over which the tuning can be performed. The typical laser linewidth is of the order of 10−3 cm−1 or smaller.
The basic principle behind the TDLAS technique is simple. The focus here is on a single absorption line in the absorption spectrum of the a particular species of interest. To start with the wavelength of a diode laser is tuned over a particular absorption line of interest and the intensity of the transmitted radiation is measured. The transmitted intensity can be related to the concentration of the species present by the
Beer-Lambert law, which states that when a radiation of wavenumber passes through an absorbing medium, the intensity variation along the path of the beam is given by [See Bernath, Peter F. (2005), C7§6 p.272-4.] ,
:where, : is the transmitted intensity of the radiation after it has traversed a distance through the medium,: is the initial intensity of the radiation,: is the absorbance of the medium, : is the absorption cross-section of the absorbing species,: is the
number densityof the absorbing species,: is the line strength (i.e. the total absorption per molecule) of the absorbing species at temperature ,: is the lineshape functionfor the particular absorption line. Sometimes also represented by ,: is the center frequency of the spectrum.
The above relation requires that the temperature of the absorbing species is known. However, it is possible to overcome this difficulty and measure the temperature simultaneously. There are number of ways to measure the temperature, a widely applied method, which can measure the temperature simultaneously, uses the fact that the line strength is a function of temperature alone. Here two different absorption lines for the same species are probed while sweeping the laser across the absorption spectrum, the ratio of the integrated absorbance, is then a function of temperature alone.
:where, : is some reference temperature at which the line strengths are known,: is the difference in the lower
energy levelsinvolved in the transitions for the lines begin probed.
Another way to measure the temperature is by relating the
FWHMof the probed absorption line to the Doppler line width of the species at that temperature. This is given by,
:where,: is the weight of one molecule of the species, and: is the molecular weight of the species.Note: In the last expression, is in kelvins and is in g/mol.However, this method can be used, only when the gas pressure is low (of the order of few
mbar). At higher pressures (tens of millibars or more), pressure or collisional broadening becomes important and the lineshape is no longer a function of temperature alone.
The effect of a mean flow of the gas in the path of the laser beam can been seen as a shift in the absorption spectrum, also known as Doppler shift. The shift in the frequency spectrum is related to the mean flow velocity by,
:where, : is the angle between the flow direction and the laser beam direction.Note : is not same as the one mentioned before where it refers to the width of the spectrum. The shift is usually very small (3×10−5 cm−1 ms−1 for near-IR diode laser) and the shift-to-width ratio is of the order of 10−4.
Limitations and means of improvements
The main disadvantage of
absorption spectrometry(AS) as well as laser absorption spectrometry(LAS) in general is that it relies on a measurement of a small change of a signal on top of a large background. Any noise introduced by the light source or the optical system will deteriorate the detectability of the technique. The sensitivity of direct absorption techniques is therefore often limited to an absorbance of ~10-3, far away from the shot noise level, which for single pass direct AS (DAS) is in the 10-7 - 10-8 range. Since this is insufficient for many types of applications, AS is seldom used in its simplest mode of operation.
There are basically two ways to improve on the situation; one is to reduce the noise in the signal, the other is to increase the absorption. The former can be achieved by the use of a modulation technique, whereas the latter can be obtained by placing the gas inside a cavity in which the light passes through the sample several times, thus increasing the interaction length. If the technique is applied to trace species detection, it is also possible to enhance the signal by performing detection at wavelengths where the transitions have larger line strengths, e.g. using fundamental vibrational bands or electronic transitions.
Modulation techniques make use of the fact that technical noise usually decreases with increasing frequency (which is why it is often referred to as 1/f noise) and improve the signal to noise ratio by encoding and detecting the absorption signal at a high frequency, where the noise level is low. The most common modulation techniques are wavelength modulation spectroscopy (WMS) and frequency modulation spectroscopy (FMS).
In WMS the wavelength of the light is continuously scanned across the absorption profile, and the signal is detected at a harmonic of the modulation frequency.
In FMS, the light is modulated at a much higher frequency but with a lower modulation index. As a result, a pair of sidebands separated from the carrier by the modulation frequency appears, giving rise to a so called FM-triplet. The signal at the modulation frequency is a sum of the beat signals of the carrier with each of the two sidebands. Since these two sidebands are fully out of phase with each other, the two beat signals cancel in the absence of absorbers. However, an alteration of any of the sidebands, either by absorption or dispersion, or a phase shift of the carrier, will give rise to an unbalance between the two beat signals, and therefore a net-signal.
Although in theory baseline-free, both modulation techniques are usually limited by residual amplitude modulation (RAM), either from the laser or from multiple reflections in the optical system (etalon effects). If these noise contributions are held low, the sensitivity can be brought into the 10-5 - 10-6 range.
Cavity Enhanced Absorption Spectrometry (CEAS)
The second way of improving the detectability of TDLAS technique is to extend the interaction length. This can be obtained by placing the species inside a cavity in which the light bounces back and forth many times, whereby the interaction length can be increased considerably. This has led to a group of techniques denoted as cavity enhanced AS (CEAS). The cavity can either be placed inside the laser, giving rise to intracavity AS, or outside, when it is referred to as an external cavity. Although the former technique can provide a high sensitivity, its practical applicability is limited because of all the non-linear processes involved.
External cavities can either be of multi-pass type, i.e. Herriott or White cells, of non- resonant type (off-axis cavity alignment), or of resonant type, most often working as a
Fabry-Perot(FP) etalon. Multi-pass cells, which typically can provide an enhanced interaction length of up to ~2 orders of magnitude, are nowaday common together with TDLAS.
Resonant cavities can provide a much larger path length enhancement, in the order of the finesse of the cavity, "F", which for a balanced cavity with high reflecting mirrors with reflectivities of ~99.99-99.999% can be ~ 104 to 105. It should be clear that if all this increase in interaction length can be utilized efficiently, this vouches for a significant increase in detectability! A problem with resonant cavities is though that a high finesse cavity has very narrow cavity modes, often in the low kHz range (the width of the cavity modes is given by FSR/F, where FSR is the free-spectral range of the cavity, which is given by "c"/2"L", where "c" is the speed of light and "L" is the cavity length). Since cw lasers often have free-running linewidths in the MHz range, and pulsed even larger, it is non-trivial to couple laser light effectively into a high finesse cavity.
The most important resonant CEAS techniques are
cavity ring-down spectrometry(CRDS), Integrated cavity output spectroscopy(ICOS) or cavity enhanced absorption spectroscopy (CEAS), phase-shift cavity ring-down spectroscopy (PS-CRDS) and Continuous wave Cavity Enhanced Absorption Spectrometry (cw-CEAS), either with optical locking, referred to as (OF-CEAS) [D. Romanini, A. A. Kachanav, J. Morville, and M. Chenevier, Proc. SPIE EUROPTO (Ser. Environmental Sensing) 3821(8), 94 (1999)] , as has been demonstrated Romanini et al. [J. Morville, S. Kassi, M. Chenevier, and D. Romanini, "Fast, low-noise, mode-by-mode, cavity-enhanced absorption spectroscopy by diode-laser self-locking," Applied Physics B-Lasers and Optics 80(8), 1027-1038 (2005)] or by electronic locking. [J. Morville, S. Kassi, M. Chenevier, and D. Romanini, "Fast, low-noise, mode-by-mode, cavity-enhanced absorp-tion spectroscopy by diode-laser self-locking," Applied Physics B-Lasers and Optics 80(8), 1027-1038 (2005)] , as for example is done in the Noise-Immune Cavity-Enhanced Optical-Heterodyne Molecular Spectroscopy(NICE-OHMS) technique. [L. S. Ma, J. Ye, P. Dube, and J. L. Hall, "Ultrasensitive frequency-modulation spectroscopy enhanced by a high-finesse optical cavity: theory and application to overtone transitions of C2H2 and C2HD," Journal of the Optical Society of America B-Optical Physics 16(12), 2255-2268 (1999)] [M. S. Taubman, T. L. Myers, B. D. Cannon, and R. M. Williams, "Stabilization, injection and control of quantum cascade lasers, and their application to chemical sensing in the infrared," Spectrochimica Acta Part a-Molecular and Biomolecular Spectroscopy 60(14), 3457-3468 (2004)] [F. M. Schmidt, A. Foltynowicz, W. Ma, T. Lock, and O. Axner, "Doppler-broadened fiber-laser-based NICE-OHMS - Improved detectability," Optics Express 15(17), 10822-10831 (2007)]
The most important non-resonant CEAS techniques are off-axis ICOS (OA-ICOS) or off-axis CEAS (OA-CEAS), wavelength modulation off-axis CEAS (WM-OA-CEAS), off-axis phase-shift cavity enhanced absorption spectroscopy (off-axis PS-CEAS).
These resonant and non-resonant cavity enhanced absorption technqies have so far not been used that frequently with TDLAS. However, since the field is developing fast, they will presumably be more used with TDLAS in the future.
Sometimes TDLAS may refer to "tunable diode laser atomic absorption spectroscopy", a technique used in plasma diagnostic. Here the TDLAS technique is used for measurement of metastable atom densities or neutral atom temperature and for identification of small amounts of impurities in plasmas. It is, also, possible to correlate the metastable atom densities with the partial pressure of a particularly gas in various gases mixture plasma. The main benefit of this method compared to optical emission methods is the exclusion of spectral lines enlargement due to optical instruments used, e.g. spectrometers, since the width of the laser light is significantly smaller than the Doppler width. Hence, the processes investigated can be studied with no further broadening than the Doppler width (which is about 1 GHz).
* Cavity Ring Down Spectroscopy (CRDS)
* Diode Lasers
* Laser Absorption Spectrometry
* Noise-Immune Cavity-Enhanced Optical-Heterodyne Molecular Spectroscopy (NICE-OHMS)
* [http://www.huxley.ic.ac.uk/research/AIRPOLL/TDL/adins/ADINS51.htm Review of TDLAS for measurement of atmospheric species] .
* [http://clu-in.org/programs/21m2/openpath/tdl/ Application of TDLAS in open path spectroscopy] .
Wikimedia Foundation. 2010.
Look at other dictionaries:
Tunable Diode Laser Absorption Spectroscopy — (TDLAS, englisch, zu deutsch etwa „Absorptionsspektroskopie mittels durchstimmbarer Laserdioden“) ist ein Verfahren, mit dem aus einer gemessenen Absorption die Konzentration oder Dichte des zu untersuchenden Gases bzw. Gasbestandteils… … Deutsch Wikipedia
Laser absorption spectrometry — (LAS) refers to techniques that utilize lasers to assess the concentration or amount of a species in gas phase by absorption spectrometry (AS).It is well known that optical spectroscopic techniques in general and laser based techniques in… … Wikipedia
Absorption spectroscopy — An overview of electromagnetic radiation absorption. This example discusses the general principle using visible light as a specific example. A white beam source – emitting light of multiple wavelengths – is focused on a sample (the complementary… … Wikipedia
spectroscopy — spectroscopist /spek tros keuh pist/, n. /spek tros keuh pee, spek treuh skoh pee/, n. the science that deals with the use of the spectroscope and with spectrum analysis. [1865 70; SPECTRO + SCOPY] * * * Branch of analysis devoted to identifying… … Universalium
Laser — For other uses, see Laser (disambiguation). United States Air Force laser experiment … Wikipedia
Laser diode — Top: a packaged laser diode shown with a penny for scale. Bottom: the laser diode chip is removed from the above package and placed on the eye of a needle for scale … Wikipedia
laser — /lay zeuhr/, n. Physics. a device that produces a nearly parallel, nearly monochromatic, and coherent beam of light by exciting atoms to a higher energy level and causing them to radiate their energy in phase. Also called optical maser. [1955 60; … Universalium
Noise-Immune Cavity-Enhanced Optical-Heterodyne Molecular Spectroscopy — (NICE OHMS) is an ultra sensitive laser based absorption technique that utilizes laser light to assess the concentration or the amount of a species in gas phase by absorption spectrometry (AS). Principles The NICE OHMS technique combines in a… … Wikipedia
Noise-immune cavity-enhanced optical heterodyne molecular spectroscopy — (NICE OHMS) is an ultra sensitive laser based absorption technique that utilizes laser light to assess the concentration or the amount of a species in gas phase by absorption spectrometry (AS). Contents 1 Principles 1.1 Types of signals 1.2… … Wikipedia
Cavity ring-down spectroscopy — (CRDS) is a spectroscopic technique for measuring the transmission or more accurately, the absorbance of light through a material. CRDS can provide extremely sensitive measurements, allowing scientists to measure very small differences in the… … Wikipedia