- Gas chromatography ion detector
Many GC detectors are
iondetectors with varying methods of ionizing the components eluting from the GC's column.
Principle of Operation
An ion detector is analogous to a
capacitoror vacuum tube. It can be envisioned as two metal grids separated by air with inverse charges placed on them. A electric potential difference (voltage) exist between the two grids. After components are ionized in the detector, they enter the region between the two grids, causing current to pass from one to the other. This current is amplified and is the signal generated by the detector. The higher the concentration of the component, the more ions are generated, and the greater the current.
Some early FIDs actually used two metal grids as their ion detectors. However, more efficient designs have been developed, so few current ion-type detectors use two metal grids. But the principle is the same, and it can be easiest to think of the detector in this manner.
Types of Ion Detectors
Flame ionization detector(FID) -- uses a flame to produce ions
Photo-ionization detector(PID) -- uses UV light to produce ions
Helium ionization detector(HID) -- uses a radioactive source to produce helium ions, which in turn ionize the components
Discharge ionization detector(DID) -- uses an electric spark source to produce helium ions, which in turn ionize the components
Pulsed discharge detector(PDD) -- similar to a Discharge ionization detector(DID), but uses a different sort of spark
Wikimedia Foundation. 2010.
Look at other dictionaries:
Gas chromatography-mass spectrometry — (GC MS) is a method that combines the features of gas liquid chromatography and mass spectrometry to identify different substances within a test sample. Applications of GC MS include drug detection, fire investigation, environmental analysis,… … Wikipedia
Gas chromatography — A gas chromatograph with a headspace sampler Acronym GLC, GC Classification chromatography Analytes organic inorganic … Wikipedia
chromatography — chromatographer, n. chromatographic /kreuh mat euh graf ik, kroh meuh teuh /, adj. chromatographically, adv. /kroh meuh tog reuh fee/, n. Chem. the separation of mixtures into their constituents by preferential adsorption by a solid, as a column… … Universalium
Gas-liquid chromatography — Infobox chemical analysis name = Gas liquid chromatography caption =A gas chromatograph with a headspace sampler acronym = GLC, GC classification =chromatography analytes = organic inorganic must be volatile manufacturers = Agilent (a spin off of … Wikipedia
Chromatography — Pictured is a sophisticated gas chromatography system. This instrument records concentrations of acrylonitrile in the air at various points throughout the chemical laboratory. Chromatography (from Greek χρῶμα chroma color and γράφειν graphein to… … Wikipedia
chromatography — The separation of chemical substances and particles (originally plant pigments and other highly colored compounds) by differential movement through a two phase system. The mixture of materials to be separated is percolated through a column or… … Medical dictionary
high-performance liquid chromatography — high per·for·mance liquid chromatography pə(r) fȯr mən(t)s n liquid chromatography in which the degree of separation is increased by forcing a solvent under pressure through a densely packed adsorbent abbr. HPLC called also high pressure liquid… … Medical dictionary
High-performance liquid chromatography — An HPLC. From left to right: A pumping device generating a gradient of two different solvents, a steel enforced column and an apparatus for measuring the absorbance. Acronym HPLC Classification Chromatography … Wikipedia
Column chromatography — A chemist in the 1950s using column chromatography. The Erlenmeyer receptacles are on the floor. Column chromatography in chemistry is a method used to purify individual chemical compounds from mixtures of compounds. It is often used for… … Wikipedia
Flame ionization detector — A flame ionization detector (FID) is a type of gas detector used in gas chromatography. The first flame ionization detector was developed in 1957 by scientists working for the CSIRO in Melbourne, Australia.… … Wikipedia