- Particle counter
A particle counter is an instrument that detects and counts particles. Applications of particle counters are separated into two primary categories:
*Aerosol particle counters
*Liquid particle counters
Aerosol particle counters
Aerosol Particle Counters are used to determine the air quality by counting and sizing the number of particles in the air. This information is useful in determining the amount of pollutants inside a building or in the ambient air. It also is useful in understanding the cleanliness level in a controlled environment. A common controlled environment aerosol particle counters are used is a
cleanroom. Cleanrooms are used extensively in semiconductor manufacturing, biotechnology, pharmaceutical, disk drive, aerospaceand other fields that are very sensitive to environmental contamination. Cleanrooms have defined particle count limits. Aerosol particle counters are used to test and classify a cleanroom to insure its performance is up to a specific cleanroom classification standard. Several standards exist for cleanroom classification. The most frequently referred to classification is from the United States. Though originating in the United States, the standard Federal Standard 209E was the first and most commonly referred to. This standard was replaced in 1999 by an international standard, however Federal Standard 209E remains today the most widely referenced standard in the world.
= US FED STD 209E cleanroom standards=
Liquid particle counters
Liquid Particle Counters are used to determine the quality of the liquid passing through them. The size and number of particles can determine if the liquid is clean enough to be used for the designed application. Liquid particle counters can be used to test the quality of
drinking wateror cleaning solutions, or the cleanliness of power generation equipment, manufacturing parts, or injectable drugs.
Liquid particle counters are also used to determine the cleanliness level of hydraulic fluids, the reason being that 75-80% of hydraulic breakdowns can be attributed to contamination. There are various types, installed on the equipment, or portable units that can be transported to site ( e.g. a construction site) and then used on the machine ( e.g. a bulldozer) to determine fluid cleanliness. By determining and monitoring these levels, and following a pro-active or predictive maintenance program, the user can reduce hydraulic failures, increase uptime and machine availability, and to reduce oil consumption. They can also be used to assure that hydraulic fluids have been cleaned using filtration, to acceptable or target cleanliness levels. There are various standards in use in the hydraulic industry, of which ISO 4406:1999, NAS1638 and SAE AS 4059 are probably the most common.
Three methods are commonly used for detecting and measuring particle size or size distribution (though many exist); Light Blocking, Light Scattering and the
The Light Blocking optical particle counter method is typical useful for detecting and sizing particles greater than 1
micrometrein size and is based upon the amount of light a particle blocks when passing through the detection area of the particle counter. This type of technique allows high resolution and reliable measurement.
ScatteringMethod is capable of detecting smaller sizing particles. This technique is based upon the amount of light that is redirected by a particle passing through the detection area of the particle counter. This redirection is referred to as light scattering. Typical detection sensitivity of the Light Scattering method is 0.05 micrometre or larger. However, employment of the condensation nuclei counter (CNC) technique would allow a higher detection sensitivity in particle sizes down to nanometrerange.
The light blocking method is typically used in the hydraulic industry. Particle counters are used here to measure contamination of hydraulic oil, and therefore allow the user to maintain their hydraulic system, reduce breakdowns, schedule maintenance during no or slow work periods, monitor filter performance etc. Particle counters used for this purpose typically use ISO Standard 4406:1999 as their reporting standard, and ISO Standard 11171 as their calibrations standard. Others currently in use are NAS 1888 and SAE 4059D.
Other types of particle counters:
Remote Particle Counters
Small particle counters that are used to monitor a fixed location typically inside a cleanroom or Minienvironment to continuously monitor particle levels 24 hours / day seven days a week. These smaller counters typically do not have a local display and are connected to a network of other particle counters and other types of sensors to monitoring the overall cleanroom performance. This network of sensors is typically connected to a facility monitoring system (FMS),
data acquisitionsystem or programmable logic controller.
This computer based system can integrate into a database, alarming and may have e-mail capability to notify facility or process personnel when conditions inside the cleanroom have exceeded predetermined environmental limits. Remote particle counters are available in several different configurations, from single channel to models that detect up to 8 channels simultaneously. Remote particle counters can have a particle size detection range from 0.1 to 100 micrometres and may feature one of a variety of output options including 4-20 mA,
RS-485 Modbus, ethernet and pulse output.
Manifold particle counters
Modified aerosol portable particle counter that has been attached to a sequencing sampling system. The sequencing sampling system allows for one particle counter to sample multiple locations, via a series of tubes drawing air from up to 32 locations inside a cleanroom. Typically less expensive then utilizing remote particle counters, however each tube is monitored in sequence.
Handheld particle counters
Small self contained particle counter that is easily transported and used. Though lower flow rates of 0.1 ft³/min (one 10th of a cubic foot) (0.2 m³/h) than larger portables with 1 ft³/m (2 m³/h), handhelds are useful for most of the same applications. However longer sample times may be required when doing
cleanroomcertification and testing. Most handheld particle counters have direct mount isokinetic sampling probes. You may use a barbed probe on a short piece of sample tubing, but it is recommended that the length of the tubing not exceed 6 ft, due to loss of larger particles in the sample tubing.
* [http://www.cemag.us Controlled Environments Magazine]
* [http://www.cleanrooms.com Cleanrooms Magazine]
* [http://www.iest.org IEST]
* [http://www.semiconductor.net/article/CA495610.html Semiconductor International]
Wikimedia Foundation. 2010.
Look at other dictionaries:
Condensation Particle Counter — Der Condensation Particle Counter, kurz CPC, ist ein optisches Messgerät, mit dem kleine Partikel mit Größen von wenigen Nanometern bis Mikrometern, zum Beispiel als Aerosol in der Atmosphäre erfasst werden können. Das physikalische Prinzip,… … Deutsch Wikipedia
counter — Synonyms and related words: Geiger counter, Wilson chamber, X ray spectrograph, X ray spectrometer, abjure, adversary, adversative, adverse, adversive, advocate, against the grain, against the tide, against the wind, agree to disagree, alien,… … Moby Thesaurus
Particle (ecology) — In marine and freshwater ecology, a particle is a small object. Particles can remain in suspension in the ocean or freshwater, however they eventually settle (rate determined by Stokes law) and accumulate as sediment. Some can enter the… … Wikipedia
Particle detector — In experimental and applied particle physics and nuclear engineering, a particle detector, also known as a radiation detector, is a device used to detect, track, and/or identify high energy particles, such as those produced by nuclear decay,… … Wikipedia
particle accelerator — accelerator (def. 7). [1945 50] * * * Device that accelerates a beam of fast moving, electrically charged atoms (ions) or subatomic particles. Accelerators are used to study the structure of atomic nuclei (see atom) and the nature of subatomic… … Universalium
Particle size distribution — The particle size distribution [Jillavenkatesa A, Dapkunas S J, Lin Sien Lum, Particle Size Characterization , NIST Special Publication 960 1, 2001] (PSD) of a powder, or granular material, or particles dispersed in fluid, is a list of values or… … Wikipedia
Particle-size analysis — This article describes the determination of the size of particulate solids, where the meaning of the particle excludes those of subatomic particles, which belong to the field of particle physics.Particle size analysis, particle size measurement,… … Wikipedia
counter — A device that counts, usually scintillations. automated differential leukocyte c. an instrument using digital imaging or cytochemical techniques to differentiate leukocytes. electronic cell c. an automatic blood cell c. in which cells passing… … Medical dictionary
Particle experiments at Kolar Gold Fields — The Kolar Gold Fields (KGF), located in the Kolar district of the state of Karnataka, India, are a set of defunct gold mines known for the neutrino particle experiments and observations that took place here starting in 1960 cite… … Wikipedia
particle detector — noun a chamber in which particles can be made visible • Hypernyms: ↑chamber • Hyponyms: ↑bubble chamber, ↑cloud chamber, ↑Wilson cloud chamber, ↑spark chamber, ↑spark counter … Useful english dictionary