A voltmeter is an instrument used for measuring electrical potential difference between two points in an electric circuit. Analog voltmeters move a pointer across a scale in proportion to the voltage of the circuit; digital voltmeters give a numerical display of voltage by use of an analog to digital converter.
Voltmeters are made in a wide range of styles. Instruments permanently mounted in a panel are used to monitor generators or other fixed apparatus. Portable instruments, usually equipped to also measure current and resistance in the form of a multimeter, are standard test instruments used in electrical and electronics work. Any measurement that can be converted to a voltage can be displayed on a meter that is suitably calibrated; for example, pressure, temperature, flow or level in a chemical process plant.
General purpose analog voltmeters may have an accuracy of a few percent of full scale, and are used with voltages from a fraction of a volt to several thousand volts. Digital meters can be made with high accuracy, typically better than 1%. Specially calibrated test instruments have higher accuracies, with laboratory instruments capable of measuring to accuracies of a few parts per million. Meters using amplifiers can measure tiny voltages of microvolts or less.
Part of the problem of making an accurate voltmeter is that of calibration to check its accuracy. In laboratories, the Weston Cell is used as a standard voltage for precision work. Precision voltage references are available based on electronic circuits.
A moving coil galvanometer can be used as a voltmeter by inserting a resistor in series with the instrument. It employs a small coil of fine wire suspended in a strong magnetic field. When an electric current is applied, the galvanometer's indicator rotates and compresses a small spring. The angular rotation is proportional to the current through the coil. For use as a voltmeter, a series resistance is added so that the angular rotation becomes proportional to the applied voltage.
One of the design objectives of the instrument is to disturb the circuit as little as possible and so the instrument should draw a minimum of current to operate. This is achieved by using a sensitive ammeter or microammeter in series with a high resistance.
The sensitivity of such a meter can be expressed as "ohms per volt", the number of ohms resistance in the meter circuit divided by the full scale measured value. For example a meter with a sensitivity of 1000 ohms per volt would draw 1 milliampere at full scale voltage; if the full scale was 200 volts, the resistance at the instrument's terminals would be 200,000 ohms and at full scale the meter would draw 1 milliampere from the circuit under test. For multi-range instruments, the input resistance varies as the instrument is switched to different ranges.
Moving-coil instruments with a permanent-magnet field respond only to direct current. Measurement of AC voltage requires a rectifier in the circuit so that the coil deflects in only one direction. Moving-coil instruments are also made with the zero position in the middle of the scale instead of at one end; these are useful if the voltage reverses its polarity.
Voltmeters operating on the electrostatic principle use the mutual repulsion between two charged plates to deflect a pointer attached to a spring. Meters of this type draw negligible current but are sensitive to voltages over about 100 volts and work with either alternating or direct current.
VTVMs and FET-VMs
The sensitivity and input resistance of a voltmeter can be increased if the current required to deflect the meter pointer is supplied by an amplifier and power supply instead of by the circuit under test. The electronic amplifier between input and meter gives two benefits; a rugged moving coil instrument can be used, since its sensitivity need not be high, and the input resistance can be made high, reducing the current drawn from the circuit under test. Amplified voltmeters often have an input resistance of 1, 10, or 20 megohms which is independent of the range selected. A once-popular form of this instrument used a vacuum tube in the amplifer circuit and so was called the vacuum tube voltmeter, or VTVM. These were almost always powered by the local AC line current and so were not particularly portable. Today these circuits use a solid-state amplifier using field-effect transistors, hence FET-VM, and appear in handheld digital multimeters as well as in bench and laboratory instruments. These are now so ubiquitous that they have largely replaced non-amplified multimeters except in the least expensive price ranges.
Most VTVMs and FET-VMs handle DC voltage, AC voltage, and resistance measurements; modern FET-VMs add current measurements and often other functions as well. A specialized form of the VTVM or FET-VM is the AC voltmeter. These instruments are optimized for measuring AC voltage. They have much wider bandwidth and better sensitivity than a typical multifunction device.
Digital voltmeters (DVMs) are usually designed around a special type of analog-to-digital converter called an integrating converter. Voltmeter accuracy is affected by many factors, including temperature and supply voltage variations. To ensure that a digital voltmeter's reading is within the manufacturer's specified tolerances, they should be periodically calibrated against a voltage standard such as the Weston cell.
Digital voltmeters necessarily have input amplifiers, and, like vacuum tube voltmeters, generally have a constant input resistance of 10 megohms regardless of set measurement range.
- Electrical measurements
- Electronic test equipment
- Meter (electronics)
- Potentiometer (measuring instrument)
- Solenoid voltmeter
- Voltage divider
- Class of accuracy in electrical measurements
Electrical and electronic measuring equipment
Ammeter · Capacitance meter · Distortionmeter · Electric energy meter · LCR meter · Microwave power meter · Multimeter · Network analyzer · Ohmmeter · Oscilloscope · Psophometer · Q meter · Signal analyzer · Signal generator · Spectrum analyzer · Transistor tester · Tube tester · Wattmeter · Vectorscope · Video signal generator · Voltmeter · VU meter
Wikimedia Foundation. 2010.
См. также в других словарях:
Voltmeter — Volt me ter, n. [2d volt + meter.] (elec.) An instrument for measuring in volts the differences of potential between different points of an electrical circuit. [1913 Webster] … The Collaborative International Dictionary of English
Voltmēter — (Spannungsmesser), s. Elektrotechnische Meßinstrumente, S. 693, und Elektrometer … Meyers Großes Konversations-Lexikon
Voltmeter — Voltmeter, s. Meßinstrumente, elektrotechnische. Bd. 6, S. 376 u. 380 … Lexikon der gesamten Technik
Voltmeter — Voltmēter, ein Galvanometer, das unmittelbar die elektromotorische Kraft einer Batterie in Volt angibt. (S. auch Voltameter.) … Kleines Konversations-Lexikon
voltmeter — (n.) 1882, from VOLT (Cf. volt) + METER (Cf. meter) (n.3) … Etymology dictionary
voltmeter — ► NOUN ▪ an instrument for measuring electric potential in volts … English terms dictionary
voltmeter — [vōlt′mēt΄ər] n. [ VOLT2 + METER] Elec. an instrument for measuring voltage … English World dictionary
voltmeter — /vohlt mee teuhr/, n. Elect. a calibrated instrument for measuring the potential difference between two points. [1880 85; VOLT1 + METER] * * * ▪ measurement instrument that measures voltages of either direct or alternating electric current… … Universalium
Voltmeter — Ein Spannungsmessgerät (auch als Spannungsmesser oder überholt als Voltmeter bezeichnet) dient zur Messung elektrischer Spannungen. Der Begriff Voltmeter ist seit Langem in der Normung ersetzt worden und sollte nicht mehr verwendet werden, da mit … Deutsch Wikipedia
voltmeter — UK [ˈvəʊltˌmiːtə(r)] / US [ˈvoʊltˌmɪtər] noun [countable] Word forms voltmeter : singular voltmeter plural voltmeters physics a piece of equipment used for measuring voltage … English dictionary