- Single photon emission computed tomography
Single photon emission computed tomography (SPECT, or less commonly, SPET) is a
nuclear medicinetomographic imaging technique using gamma rays. It is very similar to conventional nuclear medicine planar imaging using a gamma camera. However, it is able to provide true 3D information. This information is typically presented as cross-sectional slices through the patient, but can be freely reformatted or manipulated as required.
In the same way that a plain X-ray is a 2-dimensional (2-D) view of a 3-dimensional structure, the image obtained by a
gamma camerais a 2-D view of 3-D distribution of a radionuclide.
SPECT imaging is performed by using a gamma camera to acquire multiple 2-D images (also called
projections), from multiple angles. A computer is then used to apply a tomographic reconstructionalgorithm to the multiple projections, yielding a 3-D dataset. This dataset may then be manipulated to show thin slices along any chosen axis of the body, similar to those obtained from other tomographic techniques, such as MRI, CT, and PET.
SPECT is similar to PET in its use of radioactive tracer material and detection of gamma rays. In contrast with PET, however, the tracer used in SPECT emits gamma radiation that is measured directly, whereas PET tracer emits positrons which annihilate with electrons up to a few millimeters away, causing two gamma photons to be emitted in opposite directions. A PET scanner detects these emissions "coincident" in time, which provides more radiation event localization information and thus higher resolution images than SPECT (which has about 1 cm). [http://www.dr-bob.org/babble/20050308/msgs/468618.html] SPECT scans, however, are significantly less expensive. [http://www.google.com/url?sa=t&source=web&ct=res&cd=3&url=http%3A%2F%2Fwww.nd.edu%2F~aapraham%2FMedicalPhys%2FSPECT.ppt&ei=tjngSIbAMIjegQKfm4zzAw&usg=AFQjCNH1FxWGKdBzanpFQYKUQm2QSLElRA&sig2=ThCD1bwKgTWRlUm9imJpkw]
Because SPECT acquisition is very similar to planar gamma camera imaging, the same
radiopharmaceuticalsmay be used. If a patient is examined in another type of nuclear medicine scan but the images are non-diagnostic, it may be possible to proceed straight to SPECT by moving the patient to a SPECT instrument, or even by simply reconfiguring the camera for SPECT image acquisition while the patient remains on the table.
To acquire SPECT images, the gamma camera is rotated around the patient. Projections are acquired at defined points during the rotation, typically every 3-6 degrees. In most cases, a full 360 degree rotation is used to obtain an optimal reconstruction. The time taken to obtain each projection is also variable, but 15 – 20 seconds is typical. This gives a total scan time of 15-20 minutes.
Multi-headed gamma cameras can provide accelerated acquisition. For example, a dual headed camera can be used with heads spaced 180 degrees apart, allowing 2 projections to be acquired simultaneously, with each head requiring 180 degrees of rotation. Triple-head cameras with 120 degree spacing are also used.
Cardiac gated acquisitions are possible with SPECT, just as with planar imaging techniques such as MUGA. Triggered by Electrocardiogram (
EKG) to obtain differential information about the heart in various parts of its cycle, gated myocardial SPECT can be used to obtain quantitative information about myocardial perfusion, thickness, and contractility of the myocardium during various parts of the cardiac cycle; and also to allow calculation of left ventricular ejection fraction, stroke volume, and cardiac output.
SPECT can be used to complement any gamma imaging study, where a true 3D representation can be helpful. E.g. tumor imaging, infection (
leukocyte) imaging, thyroid imaging or bone imaging.
Because SPECT permits accurate localisation in 3D space, it can be used to provide information about localised function in internal organs. E.g. functional cardiac or brain imaging.
Myocardial perfusion imaging
Myocardial perfusion imaging (MPI) is a form of functional cardiac imaging, used for the diagnosis of
ischemic heart disease. The underlying principle is that under conditions of stress, diseased myocardiumreceives less blood flow than normal myocardium. MPI is one of several types of cardiac stress test.
A cardiac specific radiopharmaceutical is administered. E.g. 99mTc-tetrofosmin (Myoview, GE healthcare), 99mTc-sestamibi (Cardiolite, Bristol-Myers Squibb). Following this, the heart rate is raised to induce myocardial stress, either by exercise or pharmacologically with
adenosine, dobutamineor dipyridamole( aminophyllinecan be used to reverse the effects of dipyridamole).
SPECT imaging performed after stress reveals the distribution of the radiopharmaceutical, and therefore the relative blood flow to the different regions of the myocardium. Diagnosis is made by comparing stress images to a further set of images obtained at rest. As the radionuclide redistributes slowly, it is not usually possible to perform both sets of images on the same day, hence a second attendance is required 1-7 days later (although, with a Tl-201 myocardial perfusion study with dipyridamole, rest images can be acquired as little as two-hours post stress). However, if stress imaging is normal, it is unnecessary to perform rest imaging, as it too will be normal – thus stress imaging is normally performed first.
MPI has been demonstrated to have an overall accuracy of about 83% (sensitivity: 85%; specificity: 72%) [http://jnm.snmjournals.org/cgi/content/abstract/43/12/1634] , and is comparable (or better) than other non-invasive tests for ischemic heart disease, including stress
Functional brain imaging
Usually the gamma-emitting tracer used in functional brain imaging is 99mTc-HMPAO (hexamethylpropylene amine oxime). 99mTc is a metastable
nuclear isomerwhich emits gamma rays which can be detected by a gamma camera. When it is attached to HMPAO, this allows 99mTc to be taken up by brain tissue in a manner proportial to brain blood flow, in turn allowing brain blood flow to be assessed with the nuclear gamma camera.
Because blood flow in the brain is tightly coupled to local brain metabolism and energy use, the 99mTc-HMPAO tracer (as well as the similar 99mTc-EC tracer) is used to assess brain metabolism regionally, in an attempt to diagnose and differentiate the different causal pathologies of
dementia. Meta analysis of many reported studies suggests that SPECT with this tracer is about 74% sensitive at diagnosing Alzheimer's disease, vs. 81% sensitivity for clinical exam (mental testing, etc.). More recent studies have show accuracy of SPECT in Alzheimer diagnosis as high as 88% PMID 16785801. In meta analysis, SPECT was superior to clinical exam and clinical criteria (91% vs. 70%) in being able to differentiate Alzheimer's disease from vascular dementias. PMID 15545324 This latter ability relates to SPECT's imaging of local metabolism of the brain, in which the patchy loss of cortical metabolism seen in multiple strokes differs clearly from the more even or "smooth" loss of non-occipital cortical brain function typical of Alzheimer's disease.
99mTc-HMPAO SPECT scanning competes with
FDGPET scanning of the brain, which works to assess regional brain glucose metabolism, to provide very similar information about local brain damage from many processes. SPECT is more widely available, however, for the basic reason that the radioisotope generation technology is longer-lasting and far less expensive in SPECT, and the gamma scanning equipment is less expensive as well. The reason for this is that 99mTc ( technetium-99m) is extracted from relatively simple technetium-99m generators which are delivered to hospitals and scanning centers weekly, to supply fresh radioisotope, whereas FDGPET relies on FDGwhich must be made in an expensive medical cyclotronand "hot-lab" (automated chemistry lab for radiopharmaceutical manufacture), then must be delivered directly to scanning sites, with delivery-fraction for each trip handicapped by its natural short 110 minute half-life.
Reconstructed images typically have resolutions of 64x64 or 128x128 pixels, with the pixel sizes ranging from 3-6 mm. The number of projections acquired is chosen to be approximately equal to the width of the resulting images. In general, the resulting reconstructed images will be of lower resolution, have increased noise than planar images, and be susceptible to artifacts.
Scanning is time consuming, and it is essential that there is no patient movement during the scan time. Movement can cause significant degradation of the reconstructed images, although movement compensation reconstruction techniques can help with this. A highly uneven distribution of radiopharmaceutical also has the potential to cause artifacts. A very intense area of activity (e.g. the bladder) can cause extensive streaking of the images and obscure neighboring areas of activity. (This is a limitation of the
filtered back projectionreconstruction algorithm. Iterative reconstructionis an alternative algorithm which is growing in importance, as it is less sensitive to artifacts and can also correct for attenuation and depth dependent blurring).
Attenuation of the gamma rays within the patient can lead to significant underestimation of activity in deep tissues, compared to superficial tissues. Approximate correction is possible, based on relative position of the activity. However, optimal correction is obtained with measured attenuation values. Modern SPECT equipment is available with an integrated x-ray CT scanner. As X-ray CT images are an attenuation map of the tissues, this data can be incorporated into the SPECT reconstruction to correct for attenuation. It also provides a precisely registered CT image which can provide additional anatomical information.
* [http://jnm.snmjournals.org/cgi/content/abstract/43/12/1634 Elhendy "et al.", Dobutamine Stress Myocardial Perfusion Imaging in Coronary Artery Disease, J Nucl Med 2002 43: 1634-1646]
* For anyone interested in the brain-imaging applications of SPECT then this is a great review, although the full-text article is not available online without a subscription to the journal. The following link provides a link to the abstract only, and you might be able to access the full article through a membership with a medical library. [http://www.sciencedirect.com/science?_ob=ArticleURL&_udi=B7CV0-4HHTT0N-C&_user=121723&_handle=V-WA-A-W-AZ-MsSAYZA-UUW-U-AACBZCVEVU-AACAWBCDVU-ECZYDCBEZ-AZ-U&_fmt=summary&_coverDate=12%2F31%2F2005&_rdoc=11&_orig=browse&_srch=%23toc%2318064%232005%23999329999%23610507!&_cdi=18064&view=c&_acct=C000009999&_version=1&_urlVersion=0&_userid=121723&md5=87d339fef505792b47975d83c1561171 W. Gordon Frankle, Mark Slifstein, Peter S. Talbot, and Marc Laruelle (2005). "Neuroreceptor Imaging in Psychiatry: Theory and Applications". International Review of Neurobiology, 67: 385-440.]
Typical SPECT acquisition protocols
Magnetic resonance imaging
Positron emission tomography
ISAS (Ictal-Interictal SPECT Analysis by SPM)
* [http://www.nucmedinfo.com/ Nuclear Medicine Information]
* [http://www.mediso.com Mediso] [http://www.gehealthcare.com/usen/fun_img/nmedicine/hawkeye/index.html , GE HealthCare] [http://www.medical.philips.com/main/products/nuclearmedicine/products/precedence/features/ , Philips] [http://www.medical.siemens.com/webapp/wcs/stores/servlet/ProductDisplay?productId=138597&storeId=10001&langId=-1&catalogId=-1&catTree=100001,12788,12756*4237961997&level=0 , Siemens] - Major manufacturers of SPECT scanners
* [http://spect.yale.edu ISAS (Ictal-Interictal SPECT Analysis by SPM)] - Yale University
* [http://nuccast.com The nuclear medicine and molecular medicine podcast] - Podcast
Wikimedia Foundation. 2010.
Look at other dictionaries:
single photon emission computed tomography — n a medical imaging technique that is used esp. for mapping brain function and that is similar to positron emission tomography in using the photons emitted by the agency of a radioactive tracer to create an image but that differs in being able to … Medical dictionary
single-photon emission computed tomography — vienfotonė emisinė kompiuterinė tomografija statusas T sritis Standartizacija ir metrologija apibrėžtis Tomografija, kai įvairiomis kryptimis registruojama radionuklidų sukelta gama spinduliuotė, kuri vizualizuojama kompiuterinės tomografijos… … Penkiakalbis aiškinamasis metrologijos terminų žodynas
Single Photon Emission Computed Tomography — Die Single Photon Emission Computed Tomography (SPECT) (dt. Einzelphotonen Emissions Tomografie) ist ein diagnostisches Verfahren zur Herstellung von Schnittbildern lebender Organismen. SPECT Bilder zeigen die Verteilung eines… … Deutsch Wikipedia
single photon emission computed tomography — noun : a medical imaging technique that is used especially for mapping brain function and that is similar to positron emission tomography in using the photons emitted by the agency of a radioactive tracer to create an image but that differs in… … Useful english dictionary
single-photon emission computed tomography — SPECT. A special type of computed tomography (CT) scan in which a small amount of a radioactive drug is injected into a vein and a scanner is used to make detailed images of areas inside the body where the radioactive material is taken up by the… … English dictionary of cancer terms
single photon emission computed tomography — ▪ imaging technique imaging technique used in biomedical research and in diagnosis. SPECT is similar to positron emission tomography (PET), in which a compound labeled with a positron emitting radionuclide is injected into the body; however, its … Universalium
single-photon emission computed tomography — (SPECT) a type of tomography in which gamma photonâ€“emitting radionuclides are administered to patients and then detected by one or more gamma cameras rotated around the patient. From the series of two dimensional images produced, a three… … Medical dictionary
single photon emission computed tomography — see SPECT scanning … Medical dictionary
single photon emission computed tomography — see SPECT scanning … The new mediacal dictionary
Томография Одиночных Фотонов Эмиссион Ная Компьютерная (Single Photon Emission Computed Tomography (Spect)) — рентгенологическое исследование, позволяющее на ранней стадии выявить имеющееся у человека поражение головного мозга. Поврежденные в результате нарушения кровоснабжения вследствие травмы или инсульта клетки головного мозга выделяют глютамат,… … Медицинские термины