- Inductively coupled plasma atomic emission spectroscopy
Inductively coupled plasma atomic emission spectroscopy (ICP-AES), also referred to as inductively coupled plasma optical emission spectrometry (ICP-OES), is an analytical technique used for the detection of trace metals. It is a type of
emission spectroscopythat uses the inductively coupled plasmato produce excited atoms and ions that emit electromagnetic radiationat wavelengths characteristic of a particular element.cite journal | author = Stefánsson A, Gunnarsson I, Giroud N | title = New methods for the direct determination of dissolved inorganic, organic and total carbon in natural waters by Reagent-Free Ion Chromatography and inductively coupled plasma atomic emission spectrometry | journal = Anal. Chim. Acta | volume = 582 | issue = 1 | pages = 69–74 | year = 2007 | pmid = 17386476 | doi = 10.1016/j.aca.2006.09.001] [ cite journal|title=Is it still possible, necessary and beneficial to perform research in ICP-atomic emission spectrometry?|journal=J. Anal. At. Spectrom.|date=2005|first=J. M. |last=Mermet|coauthors=|volume=20|issue=|pages=11–16|doi= 10.1039/b416511j|url=http://www.rsc.org/publishing/journals/JA/article.asp?doi=b416511j|format=|accessdate=2007-08-31] The intensity of this emission is indicative of the concentration of the element within the sample.
The ICP-OES is composed of two parts. The ICP and the optical
spectrometer. The ICP torch consists of 3 concentric quartzglass tubes that is water cooled and a coil of the radio frequency(RF) generator which surrounds part of this torch. Argongas is typically used to create the plasma.
When the torch is turned on, an intense
magnetic fieldfrom the RF generator is turned on. The argon gas flowing through is ignited with a Teslaunit (typically a copper strip on the outside of the tube). The argon gas is ionized in this field and flows in a particular rotationally symmetrically pattern towards the magnetic field of the RF coil. A stable, high temperature plasma of about 7000K is then generated as the result of the inelastic collisions created between the neutral argon atoms and the charged particles.
peristaltic pumpdelivers an aqueous or organic sample into a nebulizerwhere it is atomized and introduced directly inside the plasma flame. The sample immediately collides with the electrons and other charged ions in the plasma and is broken down into charged ions. The various molecules break up into their respective atoms which then lose electrons and recombine repeatedly in the plasma, giving off the characteristic wavelengths of the elements involved.
A shear gas, typically
nitrogenor dry compressed air is used to 'cut' the plasma flame at a specific spot. 1 or 2 transfer lenses are then used to focus the emitted light on a diffraction gratingwhere it is separated into its component radiation in the optical spectrometer. The light intensity is then measured with a photomultipliertube at the specific wavelength for each element line involved.
The intensity of each line is then compared to previous measured intensities of known
concentrations of the element and its concentration is then computed by extrapolation along the calibration line.
Examples of the application of ICP-AES include the determination of metals in wine,cite journal | author = Aceto M, Abollino O, Bruzzoniti MC, Mentasti E, Sarzanini C, Malandrino M | title = Determination of metals in wine with atomic spectroscopy (flame-AAS, GF-AAS and ICP-AES); a review | journal = Food additives and contaminants | volume = 19 | issue = 2 | pages = 126–33 | year = 2002 | pmid = 11820494 | doi = ] arsenic in food,cite journal | author = Benramdane L, Bressolle F, Vallon JJ | title = Arsenic speciation in humans and food products: a review | journal = Journal of chromatographic science | volume = 37 | issue = 9 | pages = 330–44 | year = 1999 | pmid = 10497786 | doi = ] and trace elements bound to proteins.cite journal | author = Ma R, McLeod CW, Tomlinson K, Poole RK | title = Speciation of protein-bound trace elements by gel electrophoresis and atomic spectrometry | journal = Electrophoresis | volume = 25 | issue = 15 | pages = 2469–77 | year = 2004 | pmid = 15300764 | doi = 10.1002/elps.200405999]
In 2008 the technique was used at
Liverpool Universityto demonstrate that a "Chi-Rho" amuletfound in Shepton Malletand previously believed to be among the earliest evidence of Christianityin England,cite book|last=Leach|first=Peter|title=Shepton Mallet: Romano-Britons and Early Christians in Somerset|publisher=Birmingham University Field Archeology Unit|location=Birmingham|date=1991|isbn=0704411296] only dated to the nineteenth century. [cite news|url=http://www.telegraph.co.uk/news/newstopics/religion/2982490/Ancient-Christian-amulet-declared-a-fake.html|title='Ancient' Christian amulet declared a fake|last=Savill |first=Richard|date=2008-09-18|work=Daily Telegraph|accessdate=2008-09-18] [cite news|url=http://news.bbc.co.uk/1/hi/england/somerset/7622395.stm|title=New tests challenge age of amulet |date=2008-09-18|work=BBC News|publisher=BBC|accessdate=2008-09-18] [cite news|url=http://www.timesonline.co.uk/tol/comment/faith/article4761172.ece|title=Romano-British silver Christian cross may be fake|last=de Bruxelles |first=Simon|work=Times Online|publisher=The Times|accessdate=2008-09-18]
Atomic emission spectroscopy
Atomic absorption spectroscopy
Inductively coupled plasma mass spectrometry
* [http://www.cee.vt.edu/ewr/environmental/teach/smprimer/icp/icp.html ICP]
Wikimedia Foundation. 2010.
Look at other dictionaries:
Atomic emission spectroscopy — (AES) is a method of chemical analysis that uses the intensity of light emitted from a flame, plasma, arc, or spark at a particular wavelength to determine the quantity of an element in a sample. The wavelength of the atomic spectral line gives… … Wikipedia
Inductively coupled plasma mass spectrometry — ICP MS Instrument Acronym ICP MS Classification Mass spectrometry Analytes atomic and polyatomic species in plasma, with exceptions; usually inte … Wikipedia
Inductively Coupled Plasma Emission Spectrometry — Inductively Coupled Plasma Emission SpectrometryA test method of quantitative identification of the elements, Aluminium, Silicon, Sodium, Vanadium, Nickel, Iron, Calcium, Zinc and Phosphorus in residual fuel oils.A weighted test sample is… … Wikipedia
Atomic absorption spectroscopy — In analytical chemistry, atomic absorption spectroscopy is a technique for determining the concentration of a particular metal element in a sample.cite book |author=Sperling, Michael B.; Welz, Bernhard |title=Atomic Absorption Spectrometry… … Wikipedia
Spectroscopy — Analysis of white light by dispersing it with a prism is example of spectroscopy. Spectroscopy ( … Wikipedia
Atomic spectroscopy — is the determination of elemental composition by its electromagnetic or mass spectrum. Atomic spectroscopy is closely related to other forms of spectroscopy. It can be divided by atomization source or by the type of spectroscopy used. In the… … Wikipedia
Plasma (physics) — For other uses, see Plasma. Plasma lamp, illustrating some of the more complex phenomena of a plasma, including filamentation. The colors are a result of relaxation of electrons in excited states to lower energy states after they have recombined… … Wikipedia
Optical Emission Spectrometer — Atomspektroskopie (auch Atomspektrometrie) ist ein Sammelbegriff für spektroskopische Verfahren, die zur quantitativen und qualitativen Bestimmung von Elementen eingesetzt werden. Die Atomspektroskopie ist ein Teilbereich der Analytischen Chemie … Deutsch Wikipedia
Atom — The atom is the smallest unit of an element that retains the chemical properties of that element. An atom has an electron cloud consisting of negatively charged electrons surrounding a dense nucleus. The nucleus contains positively charged… … Wikipedia
List of materials analysis methods — List of materials analysis methods: Contents: Top · 0–9 · A B C D E F G H I J K L M N O P Q R S T U V W X Y Z μSR see Muon spin spectroscopy … Wikipedia