- Infrared thermometer
Infrared thermometers infer temperature using a portion of the thermal radiation sometimes called blackbody radiation emitted by the object of measurement. They are sometimes called laser thermometers if a laser is used to help aim the thermometer, or non-contact thermometers to describe the device’s ability to measure temperature from a distance. By knowing the amount of infrared energy emitted by the object and its emissivity, the object's temperature can be determined most of the time. They are a subset of devices more precisely called "Thermal Radiation Thermometers" or Radiation Thermometer for short.
Sometimes, especially near ambient temperatures, false readings will be obtained indicating incorrect temperature. This is most often due to other thermal radiation reflected from the object being measured, but having its source elsewhere, like a hotter wall or other object nearby - even the person holding the thermometer can be an error source in some cases. It can also be due to an incorrect emissivity on the emissivity control or a combination of the two possibilities.
The most basic design consists of a lens to focus the infrared thermal radiation on to a detector, which converts the radiant power to an electrical signal that can be displayed in units of temperature after being compensated for ambient temperature. This configuration facilitates temperature measurement from a distance without contact with the object to be measured. As such, the infrared thermometer is useful for measuring temperature under circumstances where thermocouples or other probe type sensors cannot be used or do not produce accurate data for a variety of reasons.
Some typical circumstances are where the object to be measured is moving; where the object is surrounded by an electromagnetic field, as in induction heating; where the object is contained in a vacuum or other controlled atmosphere; or in applications where a fast response is required, an accurate surface temperature is desired or the object temperature is above the recommended use point of a contact sensors, or contact with a sensor would mar the object or the sensor, or introduce a significant temperature gradient on the object's surface.
Infrared thermometers can be used to serve a wide variety of temperature monitoring functions. A few examples provided to this article include:
- Detecting clouds for remote telescope operation
- Checking mechanical equipment or electrical circuit breaker boxes or outlets for hot spots
- Checking heater or oven temperature, for calibration and control purposes
- Detecting hot spots / performing diagnostics in electrical circuit board manufacturing
- Checking for hot spots in fire fighting situations
- Monitoring materials in process of heating and cooling, for research and development or manufacturing quality control situations
There are many varieties of infrared temperature sensing devices available today, including configurations designed for flexible and portable handheld use, as well many designed for mounting in a fixed position to serve a dedicated purpose for long periods.
Specifications of portable handheld sensors available to the home user will include ratings of temperature accuracy (usually with measurement uncertainty of ±2 °C/±4 °F) and other parameters.
The distance-to-spot ratio (D:S) is the ratio of the distance to the object and the diameter of the temperature measurement area. For instance if the D:S ratio is 12:1, measurement of an object 12 inches (30 cm) away will average the temperature over a 1-inch-diameter (25 mm) area. The sensor may have an adjustable emissivity setting, which can be set to measure the temperature of reflective (shiny) and non-reflective surfaces.
A non-adjustable thermometer sometimes can be used to measure the temperature of a shiny surface by applying a non-shiny paint or tape to the surface, if the allowed measurement error is acceptable.
The most common infrared thermometers are the:
- Spot Infrared Thermometer or Infrared Pyrometer, which measures the temperature at a spot on a surface (actually a relatively small area determined by the D:S ratio).
Related equipment, although not strictly thermometers, includes:
- Infrared Scanning Systems scan a larger area, typically by using what is essentially a spot thermometer pointed at a rotating mirror. These devices are widely used in manufacturing involving conveyors or "web" processes, such as large sheets of glass or metal exiting an oven, fabric and paper, or continuous piles of material along a conveyor belt.
- Infrared Thermal Imaging Cameras or Infrared Cameras are essentially infrared radiation thermometers that measure the temperature at many points over a relatively large area to generate a two-dimensional image, called a thermogram, with each pixel representing a temperature. This technology is more processor- and software-intensive than spot or scanning thermometers, and is used for monitoring large areas. Typical applications include perimeter monitoring used by military or security personnel, inspection / process quality monitoring of manufacturing processes, and equipment or enclosed space hot or cold spot monitoring for safety and efficiency maintenance purposes.
- IRINFO.ORG - The Definitive Online Resource for Infrared Thermography
- UK Thermography Association
- Introduction to Infrared Thermometers - Omega
- Applications of Infrared Thermometers and Thermal Imagers: a free online compilation of applications from various websites and publications
- 2010 updated Radiation Thermometer pages on the About Temperature Sensors website (www.Temperatures.com) since the mid-1990s
- News of infrared and thermal imaging products & services with dedicated RSS newsfeeds
- Infrared camera
- ASTM Subcommittee E20.02 on Radiation Thermometry
- Sakuma–Hattori equation
Wikimedia Foundation. 2010.
Look at other dictionaries:
Infrared — For other uses, see Infrared (disambiguation). An image of two people in mid infrared ( thermal ) light (false color) … Wikipedia
Thermometer — A clinical mercury in glass thermometer … Wikipedia
Thermometer — A device used to measure the temperature of gaseous, liquid or solid matter or of a chemical reaction such as fire. Temperature measurement is important to a wide range of activities, including industry, scientific research, and health care. In… … Medical dictionary
infrared tympanic thermometer — a clinical thermometer inserted into the external acoustic meatus to measure body temperature by infrared tympanic thermography … Medical dictionary
tympanic thermometer — infrared tympanic t … Medical dictionary
Medical thermometer — A medical/clinical thermometer showing the temperature of 38.7 °C Medical thermometers are used for measuring human body temperature, with the tip of the thermometer being inserted either into the mouth under the tongue (oral or sub lingual… … Wikipedia
Passive infrared sensor — A Passive InfraRed sensor (PIR sensor) is an electronic device that measures infrared (IR) light radiating from objects in its field of view. PIR sensors are often used in the construction of PIR based motion detectors (see below). Apparent… … Wikipedia
Thermography — Infrared Thermography, thermal imaging, or thermal video, is a type of infrared imaging science. Thermographic cameras detect radiation in the infrared range of the electromagnetic spectrum (roughly 900 ndash;14,000 nanometers or 0.9 ndash;14 µm) … Wikipedia
Exergen Corporation — Infobox Company company name = Exergen Corporation company company type = Private company foundation = 1980 founder = Francesco Pompeicite web |url = http://www.exergen.com/html/about.htm |title = About Us |author = Exergen Corporation |… … Wikipedia
Measuring instrument — Captain Nemo and Professor Aronnax contemplating measuring instruments in Twenty Thousand Leagues Under the Sea … Wikipedia