- Rockwell scale
The Rockwell scale is a
hardnessscale based on the indentation hardnessof a material. The Rockwell test determines the hardness by measuring the depth of penetration of an indenter under a large load compared to the penetration made by a preload. [E.L. Tobolski & A. Fee, "Macroindentation Hardness Testing," "ASM Handbook, Volume 8: Mechanical Testing and Evaluation", ASM International, 2000, p 203-211, ISBN 0-87170-389-0.] There are different scales, which are denoted by a single letter, that use different loads or indenters. The result, which is a dimensionless number, is noted by HRX where X is the scale letter.
The differential depth hardness measurement was conceived in 1908 by a Viennese professor named Ludwig in his book "Die Kegelprobe" (crudely, "the cone trial"). [G.L. Kehl, "The Principles of Metallographic Laboratory Practice", 3rd Ed., McGraw-Hill Book Co., 1949, p 229.] The differential-depth method subtracted out the errors associated with the mechanical imperfections of the system, such as backlash and surface imperfections. The Rockwell hardness tester, a differential-depth machine, was co-invented by Connecticut natives Hugh M. Rockwell (1890-1957) and Stanley P. Rockwell (1886-1940). A patent was applied for on July 15, 1914. [H.M. Rockwell & S.P. Rockwell, "Hardness-Tester," US Patent 1 294 171, Feb 1919.] The requirement for this tester was to quickly determine the effects of heat treatment on steel bearing races. The Brinell hardness test, invented in 1900 in Sweden, was slow, not useful on fully
hardened steel, and left too large an impression to be considered nondestructive. The application was subsequently approved on Feb. 11, 1919, and holds patent number #1,294,171. At the time of invention, both Hugh and Stanley Rockwell (not direct relations) worked for the New Departure Manufacturing Co. of Bristol, CT. New Departure was a major ball bearing manufacturer, that in 1916 became part of United Motors and shortly later, General Motors Corp. After leaving the Connecticut company, Stanley Rockwell, then in Syracuse, NY, applied for an improvement to the original invention on Sept. 11, 1919, which was approved on Nov. 18, 1924. The new tester holds patent #1,516,207. [S.P. Rockwell, "The Testing of Metals for Hardness, "Transactions of the American Society for Steel Treating", Vol. II, No. 11, Aug 1922, p 1013-1033.] [S.P. Rockwell, "Hardness-Testing Machine," US Patent 1 516 207, Nov 1924.] Rockwell moved to West Hartford, CT, and made an additional improvement in 1921. [S.P. Rockwell, "Hardness-Testing Machine," US Patent 1 516 208, Nov 1924.] Stanley collaborated with instrument manufacturer Charles H. Wilson of the Wilson-Mauelen Company in 1920 to commercialize his invention and develop standardized testing machines. [V.E. Lysaght, "Indentation Hardness Testing", Reinhold Publishing Corp., 1949, p 57-62.] Stanley started a heat-treating firm circa 1923, the Stanley P. Rockwell Company, which still exists in Hartford, CT. The later-named Wilson Mechanical Instrument Company has changed ownership over the years, and was most recently acquired by InstronCorp. in 1993.
The determination of the Rockwell hardness of a material involves the application of a minor load followed by a major load, and then noting the depth of penetration, vis a vis, hardness value directly from a dial, in which a harder material gives a higher number. The chief advantage of Rockwell hardness is its ability to display hardness values directly, thus obviating tedious calculations involved in other hardness measurement techniques. Also, the relatively simple and inexpensive set-up enables its installation in college laboratories.
It is typically used in
engineeringand metallurgy. Its commercial popularity arises from its speed, reliability, robustness, resolution and small area of indentation.
*Cleaning indenter and test-piece to be clear of dirt, grease, rust or paint
*Measuring on a perpendicular, flat surface ("round work correction factors" are invoked to adjust for test-piece curvature)
*Ensuring that the thickness of the test-piece is at least 10 times the depth of the indentation
*Maintaining an adequate spacing between multiple indentations
*Controlling the speed of the indentation.
cales and values
There are several alternative scales, the most commonly used being the "B" and "C" scales. Both express hardness as an arbitrary
The "superficial" Rockwell scales use lower loads and shallower impressions on brittle and very thin materials. The 45N scale employs a 45-kgf load on a diamond cone-shaped Brale indenter, and can be used on dense
ceramics. The 15T scale employs a 15-kgf load on a 1/16-inch diameter hardened steel ball, and can be used on sheet metal.
Readings below HRC 20 are generally considered unreliable, as are readings much above HRB 100.
* Very hard steel (e.g. a good knife blade): HRC 55 - HRC 62
* Axes, chisels, etc.: HRC 40 - 45
Several other scales, including the extensive A-scale, are used for specialized applications. There are special scales for measuring case-hardened specimens.
*International (ISO) :*ISO 6508-1 : Metallic materials -- Rockwell hardness test -- Part 1: Test method (scales A, B, C, D, E, F, G, H, K, N, T)
* US standard (
ASTM International):* ASTME18 : Standard methods for rockwell hardness and rockwell superficial hardness of metallic materials
Vickers hardness test
Brinell hardness test
Knoop hardness test
Holger F. Struer
* [http://www.westyorkssteel.com/images/rockwell_scale_conversion_chart.pdf Rockwell scale conversion chart]
* [http://www.engineersedge.com/hardness_conversion.htm Rockwell to brinell conversion chart]
Wikimedia Foundation. 2010.
Look at other dictionaries:
Rockwell — can refer to: People * Dick Rockwell, an American comic strip and comic book artist, nephew of Norman Rockwell * Francis W. Rockwell, a United States Congressman from Massachusetts * George Lincoln Rockwell, a 1960s American Neo Nazi leader *… … Wikipedia
Rockwell hardness — (RH ) a measure of the hardness of a metal introduced by Rockwell in 1922. In a Rockwell hardness test, a penetrator makes an indentation in the metal under two constant loads, a minor load (generally 10 kilograms) and then a major load. The… … Dictionary of units of measurement
Rockwell XFV-12 — Infobox Aircraft name=Rockwell XFV 12 type=VTOL fighter manufacturer=Rockwell International caption= designer= first flight= introduced= retired= produced= number built=2 status=Cancelled 1981 unit cost= primary user=United States Navy more users … Wikipedia
Rockwell X-30 — X 30 NASP Artist s Concept of the X 30 entering orbit Role SSTO aerospaceplane Manufacturer … Wikipedia
Dudley Willis Rockwell — Dudley Willis Dud Rockwell was born June 20, 1913, in New Britain, Connecticut. Dud was the son of Ruth Gowdy Rockwell and Stanley Pickett Rockwell, inventor of the Rockwell scale. Dud graduated from Yale University Sheffield Scientific School in … Wikipedia
Brinell scale — The Brinell scale characterizes the indentation hardness of materials through the scale of penetration of an indenter, loaded on a material test piece. It is one of several definitions of hardness in materials science.Proposed by Swedish engineer … Wikipedia
Four Freedoms (Norman Rockwell) — This article is about Norman Rockwell s paintings. For other uses, see Four Freedoms (disambiguation) … Wikipedia
List of scientific units named after people — This is a list of scientific units named after people. For other lists of eponyms (names derived from people) see eponym.Note that by convention, the name of the unit is properly written in all lowercase, but its abbreviation is capitalized.SI… … Wikipedia
Vickers hardness test — A Vickers hardness tester The Vickers hardness test was developed in 1924 by Smith and Sandland at Vickers Ltd as an alternative to the Brinell method to measure the hardness of materials. The Vickers test is often easier to use than other… … Wikipedia
Barcol hardness test — The Barcol hardness test characterizes the indentation hardness of materials through the depth of penetration of an indentor, loaded on a material sample and compared to the penetration in a reference material. The method is most often used for… … Wikipedia