- Coordinate-measuring machine
A coordinate measuring machine (CMM) is a device for measuring the physical geometrical characteristics of an object. This machine may be manually controlled by an operator or it may be computer controlled. Measurements are defined by a probe attached to the third moving axis of this machine. Probes may be mechanical, optical, laser, or white light, amongst others.
The typical "bridge" CMM is composed of three axes, an X, Y and Z. These axes are orthogonal to each other in a typical three dimensional coordinate system. Each axis has a scale system that indicates the location of that axis. The machine will read the input from the touch probe, as directed by the operator or programmer. The machine then uses the X,Y,Z coordinates of each of these points to determine size and position with micrometre precision typically.
A coordinate measuring machine (CMM) is also a device used in manufacturing and assembly processes to test a part or assembly against the design intent. By precisely recording the X, Y, and Z coordinates of the target, points are generated which can then be analyzed via regression algorithms for the construction of features. These points are collected by using a probe that is positioned manually by an operator or automatically via Direct Computer Control (DCC). DCC CMMs can be programmed to repeatedly measure identical parts, thus a CMM is a specialized form of industrial robot.
Coordinate-measuring machines include three main components:
- The main structure which include three axes of motion
- Probing system
- Data collection and reduction system - typically includes a machine controller, desktop computer and application software.
They are often used for:
- Dimensional measurement
- Profile measurement
- Angularity or orientation measurement
- Depth mapping
- Digitizing or imaging
- Shaft measurement
They are offered with features like:
- Crash protection
- Offline programming
- Reverse engineering
- Shop floor suitability
- SPC software and temperature compensation.
- CAD Model import capability
- Compliance with the DMIS standard
- I++ controller compatibility
The machines are available in a wide range of sizes and designs with a variety of different probe technologies. They can be operated manually or automatically through Direct Computer Control (DCC). They are offered in various configurations such as benchtop, free-standing, handheld and portable.
The first CMM was developed by the Ferranti Company of Scotland in the 1950s as the result of a direct need to measure precision components in their military products, although this machine only had 2 axes. The first 3-axis models began appearing in the 1960s (DEA of Italy) and computer control debuted in the early 1970s (Sheffield of the USA). Leitz Germany subsequently produced a fixed machine structure with moving table.
In modern machines, the gantry type superstructure has two legs and is often called a bridge. This moves freely along the granite table with one leg (often referred to as the inside leg) following a guide rail attached to one side of the granite table. The opposite leg (often outside leg) simply rests on the granite table following the vertical surface contour. Air bearings are the chosen method for ensuring friction free travel. In these, compressed air is forced through a series of very small holes in a flat bearing surface to provide a smooth but controlled air cushion on which the CMM can move in a frictionless manner. The movement of the bridge or gantry along the granite table forms one axis of the XY plane. The bridge of the gantry contains a carriage which traverses between the inside and outside legs and forms the other X or Y horizontal axis. The third axis of movement (Z axis) is provided by the addition of a vertical quill or spindle which moves up and down through the center of the carriage. The touch probe forms the sensing device on the end of the quill. The movement of the X, Y and Z axes fully describes the measuring envelope. Optional rotary tables can be used to enhance the approachability of the measuring probe to complicated workpieces. The rotary table as a fourth drive axis does not enhance the measuring dimensions, which remain 3D, but it does provide a degree of flexibility. Some touch probes are themselves powered rotary devices with the probe tip able to swivel vertically through 90 degrees and through a full 360 degree rotation.
As well as the traditional three axis machines (as pictured above), CMMs are now also available in a variety of other forms. These include CMM arms that use angular measurements taken at the joints of the arm to calculate the position of the stylus tip. Such arm CMMs are often used where their portability is an advantage over traditional fixed bed CMMs. Because CMM arms imitate the flexibility of a human arm they are also often able to reach the insides of complex parts that could not be probed using a standard three axis machine.
In the early days of coordinate measurement mechanical probes were fitted into a special holder on the end of the quill. A very common probe was made by soldering a hard ball to the end of a shaft. This was ideal for measuring a whole range of flat, cylindrical or spherical surfaces. Other probes were ground to specific shapes, for example a quadrant, to enable measurement of special features. These probes were physically held against the workpiece with the position in space being read from a 3-Axis digital readout (DRO) or, in more advanced systems, being logged into a computer by means of a footswitch or similar device. Measurements taken by this contact method were often unreliable as machines were moved by hand and each machine operator applied different amounts of pressure on the probe or adopted differing techniques for the measurement.
A further development was the addition of motors for driving each axis. Operators no longer had to physically touch the machine but could drive each axis using a handbox with joysticks in much the same way as with modern remote controlled cars. Measurement accuracy and precision improved dramatically with the invention of the electronic touch trigger probe. The pioneer of this new probe device was David McMurtry who subsequently formed what is now Renishaw plc. Although still a contact device, the probe had a spring-loaded steel ball (later ruby ball) stylus. As the probe touched the surface of the component the stylus deflected and simultaneously sent the X.Y,Z coordinate information to the computer. Measurement errors caused by individual operators became fewer and the stage was set for the introduction of CNC operations and the coming of age of CMMs.
Optical probes are lens-CCD-systems, which are moved like the mechanical ones, and are aimed at the point of interest, instead of touching the material. The captured image of the surface will be enclosed in the borders of a measuring window, until the residue is adequate to contrast between black and white zones. The dividing curve can be calculated to a point, which is the wanted measuring point in space. The horizontal information on the CCD is 2D (XY) and the vertical position is the position of the complete probing system on the stand Z-drive (or other device component). This allows entire 3D-probing.
New Probing Systems
There are newer models that have probes that drag along the surface of the part taking points at specified intervals, known as scanning probes. This method of CMM inspection is often more accurate than the conventional touch-probe method and most times faster as well.
The next generation of scanning, known as non-contact scanning includes high speed laser single point triangulation, laser line scanning, and white light scanning, is advancing very quickly. This method uses either laser beams or white light that are projected against the surface of the part. Many thousands of points can then be taken and used to not only check size and position, but to create a 3D image of the part as well. This "point-cloud data" can then be transferred to CAD software to create a working 3D model of the part. These optical scanners often used on soft or delicate parts or to facilitate reverse engineering.
Probing systems for microscale metrology applications are another emerging area . There are several commercially available coordinate measuring machines (CMM) that have a microprobe integrated into the system, several specialty systems at government laboratories, and any number of university built metrology platforms for microscale metrology. Although these machines are good and in many cases excellent metrology platforms with nanometric scales their primary limitation is a reliable, robust, capable micro/nano probe. Challenges for microscale probing technologies include the need for a high aspect ratio probe giving the ability to access deep, narrow features with low contact forces so as to not damage the surface and high precision (nanometer level). Additionally microscale probes are susceptible to environmental conditions such as humidity and surface interactions such as stiction (caused by adhesion, meniscus, and/or Van der Waals forces among others).
Technologies to achieve microscale probing include scaled down version of classical CMM probes, optical probes, and a standing wave probe  among others. However, current optical technologies cannot be scaled small enough to measure deep, narrow feature, and optical resolution is limited by the wavelength of light. X-ray imaging provides a picture of the feature but no traceable metrology information.
Optical probes and/or laser probes can be used (if possible in combination), which change CMMs to measuring microscopes or multi-sensor measuring machines. Fringe projection systems, theodolite triangulation systems or laser distant and triangulation systems are not called measuring machines, but the measuring result is the same: a space point. Laser probes are used to detect the distance between the surface and the reference point on the end of the kinematic chain (i.e.: end of the Z-drive component). This can use an interferometrical function, focus variation, light deflection or a half beam shadowing principle.
Portable Coordinate Measuring Machines
Portable CMMs are different from "traditional CMMs" in that they most commonly take the form of an articulated arm. These arms have six or seven rotary axes with rotary encoders, instead of linear axes. Portable arms are lightweight (typically less than 20 pounds) and can be carried and used nearly anywhere. The inherent trade-offs of a portable CMM are manual operation (always requires a human to use it), and overall accuracy is somewhat to much less accurate than a bridge type CMM. Certain non-repetitive applications such as reverse engineering, rapid prototyping, and large-scale inspection of low-volume parts are ideally suited for portable CMMs.
Multi-Sensor Measuring Machines
Traditional CMM technology using touch probes is today often combined with other measurement technology. This includes laser, video or white light sensors to provide what is known as multi-sensor measurement.
- ^ "WIZprobe Kit". nextec-wiz.com. http://www.nextec-wiz.com/fr_wizblade.html. Retrieved 2010-06-26.
- ^ "Laser Line Probe". BrownAndSharpe.com. http://brownandsharpe.com/laser-metrology-solutions. Retrieved 2009-08-26.
- ^ "CMM-V Vision Probe". BrownAndSharpe.com. http://brownandsharpe.com/vision-metrology-solutions. Retrieved 2009-08-26.
- ^ Hansen H.N., Carneiro K., Haitjema H., De Chiffre L., (2006). Dimensional Micro and Nano Metrology. CIRP Annals, 55-2, 721-743
- ^ Weckenmann A., Peggs G., Hoffmann J., (2006). Probing systems for dimensional micro- and nano-metrology. Meas. Sci. Technol. 17, 504–509,
- ^ M.B. Bauza, R.J Hocken, S.T Smith, S.C Woody, (2005). The development of a virtual probe tip with application to high aspect ratio microscale features. Rev. Sci Instrum, 76 (9) 095112
- ^ "OGP Multi-Sensor Technology". ogpuk.com. http://www.ogpuk.com. Retrieved 2011-08-02.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Universal measuring machine — Universal measuring machines are measurement devices used for objects in which geometric relationships are the most critical element, with dimensions specified from geometric locations rather than absolute coordinates. The very first uses for… … Wikipedia
Milling machine — For the machine used in road construction, see Asphalt milling machine. Not to be confused with mill (grinding). Example of a bridge type CNC vertical milling center … Wikipedia
Nanometrology — NIST Next Generation Nanometrology research. Nanometrology is a subfield of metrology, concerned with the science of measurement at the nanoscale level. Nanometrology has a crucial role in order to produce nanomaterials and devices w … Wikipedia
3D scanner — A 3D scanner is a device that analyzes a real world object or environment to collect data on its shape and possibly its appearance (i.e. color). The collected data can then be used to construct digital, three dimensional models useful for a wide… … Wikipedia
Renishaw plc — is an engineering company based in Wotton under Edge in Gloucestershire, England. Renishaw is listed on the London Stock Exchange and is a constituent of the FTSE 250 Index.Renishaw specialises in industrial metrology and spectroscopy and sells… … Wikipedia
Numerical control — CNC redirects here. For other uses, see CNC (disambiguation). A CNC Turning Center … Wikipedia
CNC — The abbreviation CNC stands for computer numerical control, and refers specifically to a computer controller that reads G code instructions and drives a machine tool, a powered mechanical device typically used to fabricate components by the… … Wikipedia
Toda Racing — Infobox Company company name = Toda Racing company company type = Public foundation = 1971 location = Okayama, Japan key people = num employees = industry = Engine tuning and auto parts products = revenue = homepage = [http://www.toda… … Wikipedia
Surface plate — A surface plate is a solid, flat plate used as the main horizontal reference plane for precision inspection, marking out (layout), and tooling setup. The surface plate is often used as the baseline for all measurements to the workpiece, therefore … Wikipedia
Manufacturing process management — (MPM) is a collection of technologies and methods used to define how products are to be manufactured. MPM differs from ERP/MRP which is used to plan the ordering of materials and other resources, set manufacturing schedules, and compile cost data … Wikipedia