Photomultiplier tubes (photomultipliers or PMTs for short), members of the class of
vacuum tubes, and more specifically phototubes, are extremely sensitive detectors of light in the ultraviolet, visible, and near-infraredranges of the electromagnetic spectrum. These detectors multiply the signal produced by incident light by as much as 100 million times (i.e., 160 dB), enabling (for example) single photonsto be detected individually when the incident fluxof light is very low.
The combination of high
gain, low noise, high frequency response, and large area of collection has earned photomultipliers an essential place in nuclear and particle physics, astronomy, medical diagnosticsincluding blood tests, medical imaging, motion picture film scanning ( telecine), and high-end image scanners known as drum scanners. Semiconductor devices, particularly avalanche photodiodes, are alternatives to photomultipliers; however, photomultipliers are uniquely well-suited for applications requiring low-noise, high-sensitivity detection of light that is imperfectly collimated. While photomultipliers are extraordinarily sensitive and moderately efficient, research is still underway to create a photon-counting light detection device that is much more than 99% efficient. Such a detector is of interest for applications related to quantum informationand quantum cryptography. Elements of photomultiplier technology, when integrated differently, are the basis of night vision devices.
The photomultiplier was invented in 1936, and is rooted in the science of the
photoelectric effectand secondary emission(i.e., the ability of electronsin a vacuum tube to cause the emission of additional electrons by striking an electrode).
The photoelectric effect is perhaps most associated with
Albert Einstein, who educed the fundamental principle of quantization(i.e., the basis of quantum mechanics) for which he received the 1921 Nobel Prize. However, the phenomenon had been known earlier without understanding the quantum mechanical proportionality between optical frequency and photon energy.
The phenomenon of
secondary emissionwas first limited to purely electronic inventions (i.e., those lacking photosensitivity). In 1902, Austin and Starke reported that the metal surfaces impacted by electron beams emitted a larger number of electrons than were incident. [H. Bruining, Physics and applications of secondary electron emission, (McGraw-Hill Book Co., Inc.; 1954).] But the use of secondary emission as a means for signal amplification was not proposed until after World War I, by Joseph Slepianin a 1919 Westinghouse patent. [J. Slepian, Westinghouse Electric, "Hot Cathode Tube," U.S. Patent 1, 450, 265, April 3, 1923 (Filed 1919)]
Sixteen years later, the phenomenon of
photoemission(i.e., the photoelectric effect) was combined with secondary emissionto create the photomultiplier. In 1935 H.E. Iams and B. Salzberg of RCAreported on a single-stage photomultiplier. [H. E. Iams and B. Salzberg, "The secondary emission phototube," Proc. IRE, Vol. 23, pp. 55-64 (1935).] The device consisted of a semi-cylindrical photocathode, a secondary emitter mounted on the axis, and a collector grid surrounding the secondary emitter. The tube had a gain of about eight.
In 1936, Vladimir Zworykin, G.A. Morton, and L. Malter of RCA first reported a tube that amplified the current of photoemitted electrons in multiple stages—a device that was later called a "photomultiplier". [V.K. Zworykin, G.A. Morton, and L.Malter, "The secondary-emission multiplier-a new electronic device," Proc. IRE, Vol. 24, pp. 351-375 (1936).] The first experimental photomultipliers used a Ag-O-Cs (silver-oxide-cesium) photocathode having a typical peak quantum efficiency of 0.4% at 800 nm.
Also in 1936, a much improved photocathode, Cs3Sb (
cesium- antimony), was reported by P. Gorlich. [P. Gorlich, "Uber zusammengesetzte, durchsichtige Photokathoden," 2. Physik, Vol. 101, p. 335 (1936).] The cesium-antimony photocathode had a dramatically improved quantum efficiency of 12% at 400 nm, and was used in the first commercially successful photomultipliers manufactured by RCA (i.e., the 931-type) both as a photocathode and as a secondary-emitting material for the dynodes. Different photocathodes provided differing spectral responses.
pectral response of photocathodes
In the early 1940s the
JEDEC(Joint Electron Devices Engineering Council), an industry committee on photosensitive devices, developed a system of designating spectral responses. ["Relative spectral response data forphotosensitive devices ("S" curves)," JEDEC Publication No. 50, Electronic Industries Association, Engineering Department, 2001 I Street, N.W., Washington, D.C. 20006 (1964)] The philosophy included the idea that the product's user need only be concerned about the response of the device rather than how the device may be fabricated. The various types of photocathode devices were assigned "S-numbers" (spectral numbers) ranging from S-1 through S-40, which are still in use today. For example, S-11 uses the cesium-antimony photocathode, and S-25 uses a so-called "multialkali" photocathode (Na-K-Sb-Cs, or sodium- potassium- antimony- cesium) that provides extended response in the red portion of the visible light spectrum. No suitable photoemissive surfaces have yet been reported to detect wavelengths longer than approximately one micrometer.
Role of RCA
For decades, RCA was responsible for performing the most important work in developing and refining photomultipliers. RCA was also largely responsible for the commercialization of photomultiplers. The company compiled and published an authoritative and very-widely used "Photomultiplier Handbook". RCA made printed copies available for free upon request. The handbook, which continues to be made available online at no cost by the successors to RCA, is considered to be an essential reference.
Following a corporate break-up in the late 1980s involving the acquisition of RCA by
General Electricand disposition of the divisions of RCA to numerous third-parties, RCA's photomultiplier business became an independent company.
Lancaster, Pennsylvania facility
Lancaster, Pennsylvaniafacility was opened by the U.S. Navyin 1942 and operated by RCA for the manufacture of radio and microwave tubes. Following the Allied victory in World War II, the naval facility was acquired by RCA. "RCA Lancaster", as it became known, was the base for development and production of commercial televisionproducts. In subsequent years other products were added, such as cathode ray tubes, photomultiplier tubes, motion-sensing light control switches, and closed-circuit televisionsystems.
Transition to Burle Industries
Burle Industries, as the successor to the RCA Corporation, has carried the RCA photomultiplier business forward since 1986, and is based in the Lancaster, Pennsylvania facility. The 1986 acquisition of RCA by
General Electricresulted in the divestitureof the RCA Lancaster New Products Division. Hence, 45 years after being founded by the U.S. Navy, its management team, led by Erich Burlefinger, purchased the division and in 1987 founded Burle Industries.
The RCA "Photomultipler Handbook", along with another famous RCA reference work, is available on the Burle Industries website. [ [http://www.burle.com/cgi-bin/byteserver.pl/pdf/Photo.pdf PhotoMultiplier] Hand-Book]
In 2005, after eighteen years as an independent enterprise, Burle Industries and a key subsidiary were acquired by Photonis, a
European holding company[http://www.photonis.com Photonis Group] . Following the acquisition, Photonis was comprised of Photonis Netherlands, Photonis France, Photonis USA, and Burle Industries. Photonis USA operates the former Galileo Corporation Scientific Detector Products Group ( Sturbridge, Massachusetts), which had been purchased by Burle Industries in 1999. The Group is known for microchannel plate detector(MCP) electron multipliers—an integrated micro-vacuum tube version of photomultipliers. MCPs are used for imaging and scientific applications, including night vision devices.
Japan-based company Hamamatsu Photonics(also known as Hamamatsu) has emerged since the 1960s as a leader in the photomultiplier industry. Hamamatsu, in the tradition of RCA, has published its own handbook, which is available without cost on the company's website.
tructure and operating principles
Photomultipliers are constructed from a glass
vacuum tube, which houses a photocathode, several dynodes, and an anode. Incident photonsstrike the photocathodematerial, which is present as a thin deposit on the entry window of the device, with electrons being produced as a consequence of the photoelectric effect. These electrons are directed by the focusing electrodetoward the electron multiplier, where electrons are multiplied by the process of secondary emission. The electron multiplier consists of a number of electrodes, called "dynodes". Each dynode is held at a more positive voltage than the previous one. The electrons leave the photocathode, having the energy of the incoming photon (minus the work functionof the photocathode). As the electrons move toward the first dynode, they are accelerated by the electric field and arrive with much greater energy. Upon striking the first dynode, more low energy electrons are emitted, and these electrons in turn are accelerated toward the second dynode. The geometry of the dynode chain is such that a cascade occurs with an ever-increasing number of electrons being produced at each stage. Finally, the electrons reach the anode, where the accumulation of charge results in a sharp current pulse indicating the arrival of a photon at the photocathode.
Photomultiplier tubes typically utilize 1000 to 2000
voltsto accelerate electrons within the chain of dynodes. The most negative voltage is connected to the cathode, and the most positive voltage is connected to the anode. Negative high-voltage supplies (with the positive terminal grounded) are preferred, because this configuration enables the photocurrentto be measured at the low voltage side of the circuit for amplification by subsequent electronic circuits operating at low voltage. Voltages are distributed to the dynodes by a resistive voltage divider, although variations such as active designs (with transistorsor diodes) are possible. The divider design, which influences frequency response or rise time, can be selected to suit varying applications.
While powered (energized), photomultipliers must be shielded from
ambient lightto prevent their destruction through overexcitation. If used in a location with strong magnetic fields, which can curve electron paths, photomultipliers are usually shielded by a layer of mu-metal.
* Photomultipliers were the first
electric eyedevices, being used to measure interruptions in beams of light.
* Photomultipliers are used in conjunction with
scintillators to detect nuclear and particle radiationin physics experiments.
* Photomultipliers are used in research laboratories to measure the intensity and spectrum of light-emitting materials such as
compound semiconductors and quantum dots.
* Photomultipliers are used in numerous medical equipment designs. For example,
blood analysisdevices used by clinical medical laboratories utilize photomultipliers to determine the relative concentration of various components in vials of blood drawn in doctors' offices, in combination with optical filters and incandescent lamps.
High sensitivity applications
After fifty years, during which solid-state electronic components have largely displaced the vacuum tube, the photomultiplier remains a unique and important optoelectronic component. Perhaps its most useful quality is that it acts, electronically, as a nearly perfect current source owing to the high voltage utilized in extracting the tiny currents associated with weak light signals. There is no
Johnson noiseassociated with photomultiplier signal currents even though they are greatly amplified, e.g., by 100 thousand times (i.e., 100 dB) or more.
Photomultiplier-amplified photocurrents can be electronically amplified by a high-input-impedance electronic amplifier (in the signal path, subsequent to the photomultiplier), thus producing appreciable voltages even for nearly infinitesimally small photon fluxes. Photomultipliers offer the best possible opportunity to exceed the Johnson noise for many configurations. The aforementioned refers to measurement of light fluxes that, while small, nonetheless amount to a continuous stream of multiple photons.
For smaller photon fluxes, the photomultiplier can be operated in photon counting or Geiger mode ("see also: single-photon avalanche diode"). In Geiger mode the photomultiplier gain is set so high (using high voltage) that a single photo-electron resulting from a single photon incident on the primary surface generates a very large current at the output circuit. However, owing to the avalanche of current, a reset of the photomultiplier is required. In either case, the photomultiplier can detect individual photons. The drawback, however, is that not every photon incident on the primary surface is counted either because of less-than-perfect efficiency of the photomultiplier, or because a second photon can arrive at the photomultiplier during the "dead time" associated with a first photon and never be noticed.
Nonetheless, the ability to detect single photons striking the primary photosensitive surface itself reveals the quantization principle that Einstein put forth. Photon-counting (as it is called) reveals that light, not only being a wave, consists of discrete particles (i.e., photons).
* Engstrom, Ralph W., "Photomultiplier Handbook", RCA (1980).
* "Photomultiplier Tubes: Basics and Applications (Second Edition)", Hamamatsu Photonics, Hamamatsu City, Japan, (1999).
* Flyckt, S.O. and Marmonier, C., [http://www.photonis.com/products/photomultiplier-tubes/application_book "Photomultiplier Tubes: Principles and Applications"] , Philips Photonics, Brive, France (2002).
* [http://microscopy.fsu.edu/primer/flash/photomultiplier/ Molecular Expressions] - Java-based simulation and tutorial on photomultiplier tubes
* [http://www.burle.com/cgi-bin/byteserver.pl/pdf/Photo.pdf Photomultiplier Handbook] (4MB PDF) from Burle Industries, essentially the Engstrom-RCA Handbook reprinted
* [http://www.electrontubes.com/info/papers.html Photomultiplier technical papers] from Electron Tubes Ltd.
* [http://sales.hamamatsu.com/assets/applications/ETD/pmt_handbook_complete.pdf Photomultiplier tubes] basics and applications from
* [http://www.vias.org/simulations/simusoft_emultiplier.html Electron Multiplier] - simulation of an electron multiplier tube
Wikimedia Foundation. 2010.
Look at other dictionaries:
photomultiplier — [fōt΄ō mul′tə plī΄ər] n. a photoemissive photoelectric cell which amplifies emitted electrons and converts them into brighter light, an electric signal, etc … English World dictionary
Photomultiplier — Schematische Skizze eines Photomultipliers Photomultiplier, Länge … Deutsch Wikipedia
photomultiplier — fotodaugintuvas statusas T sritis automatika atitikmenys: angl. photomultiplier; photomultiplier tube vok. Photoelektronenvervielfacher, m; Photomultiplikatorröhre, f rus. фотоумножитель, m; фотоэлектронный умножитель, m pranc. tube… … Automatikos terminų žodynas
photomultiplier — fotodaugintuvas statusas T sritis radioelektronika atitikmenys: angl. photomultiplier; photomultiplier tube vok. Photoelektronenvervielfacher, m; Photomultiplikatorröhre, f; Photovervielfacher, m rus. фотоумножитель, m; фотоэлектронный умножитель … Radioelektronikos terminų žodynas
photomultiplier — noun see photomultiplier tube … New Collegiate Dictionary
Photomultiplier — Pho|to|mul|ti|pli|er 〈[ mʌltıplaıə(r)] m. 3〉 = Fotomultiplier * * * Photomultiplier [ mʌltɪpləɪə, englisch], Elektronenvervielfacher … Universal-Lexikon
photomultiplier — fotodaugintuvas statusas T sritis fizika atitikmenys: angl. photomultiplier vok. Photoelektronenvervielfacher, m; Photovervielfacher, m rus. фотоумножитель, m; фотоэлектронный умножитель, m pranc. photomultiplicateur, m … Fizikos terminų žodynas
photomultiplier — /foh teuh mul teuh pluy euhr/, n. an extremely sensitive detector of light and of other radiation, consisting of a tube in which the electrons released by radiation striking a photocathode are accelerated, greatly amplifying the signal obtainable … Universalium
photomultiplier — noun A device, normally in the form of a tube, that uses a photocathode to convert photons into photoelectrons which are then amplified … Wiktionary
photomultiplier — noun an instrument containing a photoelectric cell and a series of electrodes, used to detect and amplify very faint light … English new terms dictionary