- Resolution (mass spectrometry)
Resolution and Resolving Power
There are two different definitions of resolution and resolving power in mass spectrometry.
The IUPAC definition for resolution in mass spectrometry is
- = resolution
- ΔM = resolving power
where a larger resolution indicates a better separation of peaks. This definition is used in a number of mass spectrometry texts. This use is also implied by the term "high-resolution mass spectrometry."
A high value for resolution corresponding to good separation of peaks is similar to the convention used with chromatography separations, although it is important to note that the definitions are not the same. High resolution indicating better peak separation is also used in ion mobility spectrometry
IUPAC defines Δm as the resolving power.
Resolving power definition
Some mass spectrometrists use the definition that is similar to definitions used in some other fields of physics and chemistry. In this case, resolving power is defined as:
- = resolving power.
The minimum peak separation ΔM which allows to distinguish two ion species is then called:
- ΔM = resolution.
Resolution and resolving power, when defined in this way, are consistent with IUPAC recommendations for microscopy, optical spectroscopy. and ion microscopy (SIMS)  but not gas chromatography. This definition also appears in some mass spectrometry texts.
Measuring peak separation
There are several ways to define the minimum peak separation ΔM in mass spectrometry, therefore it is important to report the method used to determine mass resolution when reporting its value. The two most widely used are the peak width definition and the valley definition.
Peak width definition
In the peak width definition, the value of Δm is the width of the peak measured at a specified fraction of the peak height, for example 0.5%, 5%, 10% or 50%. The latter is called the full width at half maximum (FWHM).
The valley definition defines ΔM as the closest spacing of two peaks of equal intensity with the valley (lowest value of signal) between them less than a specified fraction of the peak height. Typical values are 10% or 50%. The value obtained from a 5% peak width is roughly equivalent to a 10% valley.
- ^ a b c IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "resolution in mass spectroscopy".
- ^ Biemann, Klaus (1962). Mass Spectrometry: Organic Chemical Applications. New York: McGraw-Hill. p. 13. ISBN 0070052352.
- ^ Tureček, František; McLafferty, Fred W. (1993). Interpretation of mass spectra. Sausalito, Calif: University Science Books. ISBN 0-935702-25-3.
- ^ Watson, J. S. (1997). Introduction to mass spectrometry. Philadelphia: Lippincott-Raven. ISBN 0-397-51688-6.
- ^ Ashcroft, Alison E. (1997). Ionization methods in organic mass spectrometry. Cambridge, Eng: Royal Society of Chemistry. ISBN 0-85404-570-8.
- ^ JURGEN H. GROSS; Jnrgen H. Gross (2004). Mass Spectrometry: A Textbook. Berlin: Springer-Verlag. ISBN 3-540-40739-1.
- ^ Todd, John F. J.; March, Raymond E. (2005). Quadrupole ion trap mass spectrometry. New York: Wiley-Interscience. ISBN 0-471-48888-7.
- ^ Siuzdak, Gary (2006). The Expanding Role of Mass Spectrometry in Biotechnology, Second Edition. MCC Press. ISBN 0-9742451-2-7.
- ^ Stroobant, Vincent; Hoffmann, Edmond de (2007). Mass spectrometry: principles and applications. London: J. Wiley. ISBN 0-470-03310-X.
- ^ Ingvar Eidhammer (2007). Computational methods for mass spectrometry proteomics. Chichester: John Wiley & Sons. ISBN 0-470-51297-0.
- ^ VanLear GE, McLafferty FW (1969). "Biochemical aspects of high-resolution mass spectrometry". Annu. Rev. Biochem. 38: 289–322. doi:10.1146/annurev.bi.38.070169.001445. PMID 4896241. http://arjournals.annualreviews.org/doi/abs/10.1146/annurev.bi.38.070169.001445?url_ver=Z39.88-2003&rfr_id=ori:rid:crossref.org&rfr_dat=cr_pub%3dncbi.nlm.nih.gov.
- ^ IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "resolution in gas chromatography]]".
- ^ Blumberg LM, Kle MS (November 2001). "Metrics of separation in chromatography". J Chromatogr A 933 (1–2): 1–11. doi:10.1016/S0021-9673(01)01256-0. PMID 11758739.
- ^ Karpas, Zeev; Eiceman, Gary Alan (2005). Ion mobility spectrometry. Boca Raton: CRC Press. ISBN 0-8493-2247-2.
- ^ IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "resolving power in mass spectrometry".
- ^ IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "resolution in optical spectroscopy".
- ^ IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "resolving power in optical spectroscopy".
- ^ IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "resolving power in ion microscopy".
- ^ IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "resolution in gas chromatography".
- ^ David O. Sparkman (2006). Mass Spectrometry Desk Reference. Pittsburgh: Global View Pub. ISBN 0-9660813-9-0.
- ^ Sparkman, O. David (2007). Introduction to mass spectrometry: instrumentation, applications and strategies for data interpretation. Chichester: John Wiley & Sons. ISBN 0-470-51634-8.
- ^ Dass, Chhabil (2007). Fundamentals of contemporary mass spectrometry. Chichester: John Wiley & Sons. ISBN 0-471-68229-2.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Mass (mass spectrometry) — This article is about different concepts of mass used in mass spectrometry. For other uses, see mass. The mass recorded by a mass spectrometer can refer to different physical quantities depending on the characteristics of the instrument and the… … Wikipedia
Mass spectrometry software — is software used for data acquisition, analysis, or representation in mass spectrometry. Contents 1 MS/MS peptide identification 1.1 Database search algorithms 1.1.1 SEQUEST 1.1.2 … Wikipedia
Mass spectrometry — (MS) is an analytical technique that measures the mass to charge ratio of charged particles. It is used for determining masses of particles, for determining the elemental composition of a sample or molecule, and for elucidating the chemical… … Wikipedia
mass spectrometry — or mass spectroscopy Analytic technique by which chemical substances are identified by sorting gaseous ions by mass using electric and magnetic fields. A mass spectrometer uses electrical means to detect the sorted ions, while a mass spectrograph … Universalium
Mass Spectrometry Instruments — Infobox Company name = Mass Spectrometry Instruments Ltd. location = Dewberry, W. Yorkshire, U.K. industry = Mass Spectrometry products = High Resolution Mass Spectrometers homepage = http://www.massint.co.uk Mass Spectrometry Instruments Ltd. or … Wikipedia
Time-of-flight mass spectrometry — (TOFMS) is a method of mass spectrometry in which ions are accelerated by an electric field of known strength. [Stephens, W. E., [http://link.aps.org/abstract/PR/v69/p674/s2 A Pulsed Mass Spectrometer with Time Dispersion] Phys. Rev. , 1946, 69 … Wikipedia
Static secondary ion mass spectrometry — Static secondary ion mass spectrometry, or static SIMS is a technique for chemical analysis including elemental composition and chemical structure of the uppermost atomic or molecular layer of a solid which may be a metal, semiconductor or… … Wikipedia
Resolution — may refer to: Resolution (audio), a measure of digital audio quality Resolution (logic), a rule of inference used for automated theorem proving Resolution (law), a written motion adopted by a deliberative body Resolution (debate), the statement… … Wikipedia
Mass spectrum — Electron ionization mass spectrum of toluene . Note parent peak corresponding to molecular mass M = 92 (C7H8+) and highest peak at M 1 = 91 (C7H7+, quasi stable tropylium cation). A mass spectrum is an inten … Wikipedia
Mass chromatogram — A mass chromatogram is a representation of mass spectrometry data as a chromatogram, where the x axis represents time and the y axis represents signal intensity. The source data contains mass information; however, it is not graphically… … Wikipedia