- Dose verification system
DVS (Dose Verification System), developed by Sicel Technologies, is an implantable telemetric, radiation sensor that has been cleared for use by the U.S. Food & Drug Administration (FDA) in the treatment of breast cancer and prostate cancer.
The device is used to measure the amount of radiation that is delivered to the tumor tissue, tumor bed and/or healthy tissue. The DVS sensor contains a dosimeter and wireless transmitter inside a sealed, biocompatible glass capsule measuring 0.8 inches (20mm) long and 0.08 inches (2.1mm) across.
The DVS sensors are implanted during a minimally invasive procedure prior to starting radiation therapy. The sensor is placed 3 cm beneath the surface of the skin to ensure adequate build-up, especially at higher energies and to maintain a consistent temperature.
Before and after each treatment, a radiation therapist will use a hand-held reader held above the skin near the dosimeter to transmit and report dosage information immediately to the radiation oncology team. The information alerts the team to random or systematic deviations from the prescribed dose that can have a profound impact on tumor control and surrounding normal tissue.
After Radiation Treatment
At the conclusion of radiation treatment, the DVS sensor remains in the body permanently but has no impact on everyday life. The sensors are compatible with medical procedures such as mammography and MRI therefore it important to tell all healthcare providers about DVS. During a monthly breast self-exam it is possible that a person may be able to feel the dosimeter, so it is also important to note the location when it is implanted.
References and External Links
Wikimedia Foundation. 2010.
Look at other dictionaries:
Verification and Validation — Verification Validation is the process of checking that a product, service, or system meets specifications and that it fulfils its intended purpose. These are critical components of a quality management system such as ISO… … Wikipedia
Verification and validation — IV V redirects here. For NASA s IV V Facility, see Independent Verification and Validation Facility. Verification and validation is the process of checking that a product, service, or system meets specifications and that it fulfills its intended… … Wikipedia
IBM System/36 — The IBM System/36 was a minicomputer marketed by IBM from 1983 to 2000. It was a multi user, multi tasking successor to the System/34. Like the System/34 and the older System/32, the System/36 was primarily programmed in the RPG II language. One… … Wikipedia
Radiation therapy — Radiation (medicine) redirects here. It is not to be confused with Radiation (pain) or Radiology. Radiation therapy Intervention ICD 10 PCS D ICD 9 CM … Wikipedia
Cyberknife — The CyberKnife is a frameless robotic radiosurgery system used for treating benign tumors, malignant tumors and other medical conditions. The system was invented by John R. Adler, a Stanford University Professor of Neurosurgery and… … Wikipedia
Radiosurgery — This article is about the medical procedure. For the album by American rock band New Found Glory, see Radiosurgery (album). Radiosurgery Intervention CLINAC: a model of steerable linear accelerator commonly used in radiosurgery … Wikipedia
Fast neutron therapy — Intervention ICD 10 PCS D?0?5ZZ ICD 9: 92.26 Fast neutron therapy utilizes high energy neutrons typically greater than 20 MeV … Wikipedia
Particle therapy — Intervention ICD 9: 92.26 Particle therapy is a form of external beam radiotherapy using beams of energetic protons, neutrons, or positive ions for cancer treatment. The most common … Wikipedia
Boron neutron capture therapy — Intervention A schematic of therapy facility in Otaniemi, Finland. ICD 10 PCS D?0?6ZZ … Wikipedia
Megavoltage X-rays — Intervention ICD 9: 92.24 Megavoltage X rays are produced by linear accelerators ( linacs ) operating at voltages in excess of 1000 kV (1 MV) range, and therefore have an energy in the MeV ran … Wikipedia