- Collision/reaction cell
Dynamic reaction cell
The collision reaction cell known by the trade name dynamic reaction cell was introduced by Perkin-Elmer on their Elan DRC (followed by Elan DRC II and Elan DRC-e) instrument. The dynamic reaction cell is a chamber placed before the traditional quadrupole chamber of an ICP-MS device, for eliminating isobaric interferences. The chamber has a quadrupole and can be filled-up with reaction (or collision) gases (ammonia, methane, oxygen or hydrogen), with one gas type at a time or a mixture of two of them, which reacts with the introduced sample, eliminating some of the interference.
The DRC is characterized by the following paramters, that can be modified: RPq (the corresponding q parameter from the Mathieu equation), RPa (the corresponding a parameter from the Mathieu equation), which refer to the voltage applied to the quadrupole rods and the gas flow of the reaction gas.
Ammonia gas is the best solution for the majority of interferences, but it is far for being the perfect gas. Sometimes, for specific isotopes, other gas must be used for better results or even mathematical correction, if no gas offers a satisfactory advantage.
Collisional reaction interface (CRI)
The proprietary collisional reaction interface (CRI) used in the Bruker ICP-MS destroying interfering ions. These ions are removed by injecting a collisional gas (He), or a reactive gas (H2), or a mixture of the two, directly into the plasma as it flows through the skimmer cone and/or the sampler cone. Supplying the reactive/collisional gas into the tip of the skimmer cone and/or into the tip of the sampler cone induces extra collisions and reactions that destroy polyatomic ions in the passing plasma.
Axial field technology
Axial field technology (AFT) is a patented improvement of DRC made by Perkin-Elmer, which consists in two supplementary rods placed in the DRC cell, smaller than normal quadrupole's rods, with the purpose of "pushing" the ions faster to the exit by generating a supplementary electric potential, minimizing the time needed for the gas to be in the DRC and improving analysis speed. The suplimetary potential of the AFT rods does not contribute significantly to the global energy, but drastically improve ion passage time.
Collision cell technology with kinetic energy discrimination
Thermo Scientific's XSeries2 instrument utilizes a collision/reaction cell for interference removal, consisting of a non-consumable hexapole and chicane ion deflector, which takes the ion beam off-axis and leads to low instrument backgrounds of <0.5 integrated counts per second (icps) at vacant masses such as 5 and 220. This hexapole is inherently part of the Thermo lens system and is present in the ion path, regardless of the use of the collision cell. The collision/reaction gas mixtures can be 1% NH3 in He, 7% H2 in He and 100% H2, where the NH3 and H2 are reactive gasses and the He is a collisional gas. The 3rd generation cell utilizes kinetic energy discrimination, which employes running the quadrupole bias slightly less negative (more positive) than the hexopole bias. Polyatomic ions generated within the plasma can have larger atomic radii than analyte ions of similar mass, i.e. the interferent NaAr+ (mass 63) is larger than the analyte Cu+ (mass 63). Thus, when using a collisional/reactive gas mixture, these larger species undergo more collisions/reactions in the cell, in which they lose increasingly more energy, and are then excluded from the quadrupole mass filter by the kinetic energy barrier.
Octopole reaction system
Another implementation of this type of interference removal is an octopole (instead of a quadrupole) collision cell, implemented by Agilent's 7500 series. The octopole reaction system (ORS)) uses only helium or hydrogen and the volume of the cell is smaller than that of a DRC. The small molecules of helium and hydrogen collide with the large, unwanted polyatomic ions formed in the plasma and break them up into other ions that can be separated in the quadrupole mass analyser. However, unlike the DRC the OCR system is based only on collision reactions and not on chemical reactions.
- ^ Yip, Y.; Sham, W (2007). "Applications of collision/reaction-cell technology in isotope dilution mass spectrometry". TrAC Trends in Analytical Chemistry 26: 727. doi:10.1016/j.trac.2007.03.007
- ^ V. Baranov, S. Tanner (1999). "A dynamic reaction cell for ICP-MS. Part 1: The rf-field energy contribution in thermodynamics of ion-molecule reactions". J. Anal. At. Spectrom. 14: 1133–1142. doi:10.1039/a809889a.
- ^ S. Tanner, V. Baranov (1999). "A dynamic reaction cell for ICP-MS. Part 2: Reduction of interferences produced within the cell". J. Am. Soc. Mass Spectrom. 10: 1083–1094. doi:10.1016/S1044-0305(99)00081-1.
- ^ A beginner's Guide to ICP-MS R. Thomas
- ^ S. Tanner, V. Baranov, D. Bandura (2002). "Reaction cells and collision cells for ICP-MS: a tutorial review". Spectrochimica Acta B 57: 1361–1452. Bibcode 2002AcSpe..57.1361T. doi:10.1016/S0584-8547(02)00069-1.
- ^ I. Kalinitchenko, Patent Application under the Patents Cooperation Treaty WO 2004/012223 A1
- ^ Wang, XueDong; Iouri Kalinitchenko. "Principles and performance of the Collision Reaction Interface for the" (PDF). Varian. http://www.varianinc.com/image/vimage/docs/products/spectr/icpms/articles/adv_note_1.pdf. Retrieved 2009-01-20.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Chemical reaction — Chemical reactions redirects here. For the 2007 television episode, see Chemical Reactions (Men in Trees). A thermite reaction using iron(III) oxide. The sparks … Wikipedia
Selected reaction monitoring — (SRM) is a method used in tandem mass spectrometry in which an ion of a particular mass is selected in the first stage of a tandem mass spectrometer and an ion product of a fragmentation reaction of the precursor ion is selected in the second… … Wikipedia
Multiphase particle-in-cell method — The multiphase particle in cell method (MP PIC) is a numerical method for modeling particle fluid and particle particle interactions in a computational fluid dynamics (CFD) calculation. The MP PIC method achieves greater stability than its… … Wikipedia
Inductively coupled plasma mass spectrometry — ICP MS Instrument Acronym ICP MS Classification Mass spectrometry Analytes atomic and polyatomic species in plasma, with exceptions; usually inte … Wikipedia
Индуктивно-связанная плазма в масс-спектрометрии — Эту статью следует викифицировать. Пожалуйста, оформите её согласно правилам оформления статей. Масс спектрометрия с индуктивно связанной плазмой (ИСП МС) это разновидность масс спектрометрии, отличающаяся … Википедия
ИСП-МС — Масс спектрометрия с индуктивно связанной плазмой (ИСП МС) это разновидность масс спектрометрии, отличающаяся высокой чувствительностью и способностью определять ряд металлов и несколько неметаллов в концентрациях, не превышающих 10 10%, т.e.… … Википедия
contact inhibition of movement — Reaction in which the direction of motion of a cell is altered following collision with another cell. In heterologous contacts both cell may respond (mutual inhibition), or only one (non reciprocal). Type I contact inhibition involves paralysis… … Dictionary of molecular biology
radiation — radiational, adj. /ray dee ay sheuhn/, n. 1. Physics. a. the process in which energy is emitted as particles or waves. b. the complete process in which energy is emitted by one body, transmitted through an intervening medium or space, and… … Universalium
Physical Sciences — ▪ 2009 Introduction Scientists discovered a new family of superconducting materials and obtained unique images of individual hydrogen atoms and of a multiple exoplanet system. Europe completed the Large Hadron Collider, and China and India took… … Universalium
Mathematics and Physical Sciences — ▪ 2003 Introduction Mathematics Mathematics in 2002 was marked by two discoveries in number theory. The first may have practical implications; the second satisfied a 150 year old curiosity. Computer scientist Manindra Agrawal of the… … Universalium