 Modulation transfer function (infrared imaging)

The Modulation Transfer Function (MTF) is used to approximate the position of best focus of an infrared imaging system. In an imaging system, best focus is typically achieved when the MTF is between 0.4 and 0.6; most often at 0.5 (50% cutoff frequency of the MTF) MTF is inversely related to the minimum resolvable temperature difference (MRTD), which is a measure of an infrared sensor's ability to resolve temperature difference. MTF is defined as the discrete fourier transform of the Line Spread Function (LSF). The LSF can be calculated by two different methods. One includes measuring the LSF directly from an idealized line approximation provided by an image of a slit target. The other involves differentiating the Edge Spread Function (ESF).^{[1]}
Contents
ESF evaluation
An operator defines a box area encompassing the edge of a knifeedge test target image backilluminated by a blackbody. The box area is defined to be approximately 10% of the total frame area. The image pixel data is translated into a twodimensional array (pixel intensity and pixel position). The amplitude (pixel intensity) of each line within the array is normalized and averaged. This yields the edge spread function (ESF)^{[2]}
where
 ESF = the output array of normalized pixel intensity data
 = the input array of pixel intensity data
 = the i^{th} element of
 = the average value of the pixel intensity data
 = the standard deviation of the pixel intensity data
 = number of pixels used in average
LSF evaluation
The line spread function (LSF) can be found using two different methods. It can be found directly from an ideal line approximation provided by a slit test target or it can be derived from the ESF. Using the latter method, the LSF is defined as the first derivative of the ESF^{[3]}, which is differentiated using numerical methods.
Since the ESF can not be differentiated analytically, it is numerically approximated using the finite difference:
where
 = the index
 = position of the pixel
 = ESF of the pixel
MTF Evaluation
The Modulation Transfer Function (MTF) is defined as the discrete fourier transform of the Line Spread Function. Thus, given the LSF, the MTF is approximated numerically. This data is graphed against the spatial frequency data. A sixth order polynomial is fitted to the MTF vs. spatial frequency curve to remove any trends. The 50% cutoff frequency is determined to yield the corresponding spatial frequency. Thus, the approximate position of best focus of the Unit Under Test is determined from this data.
MTF calculations
The Fourier transform of the LSF can not be determined analytically by the following equations:
Therefore, the Fourier Transform is numerically approximated using the discrete Fourier transform .^{[4]}
where
 = the k^{th} value of the MTF
 = number of data points
 = index
 = k^{th} term of the LSF data
 = pixel position
Since most computer software is not able to compute complex numbers directly, Euler's identity is implemented to break the transform into separate real and complex terms.
The MTF is then plotted against spatial frequency and all relevant data concerning this test can be determined from that graph.
See also
 Minimum Resolvable Contrast
 Minimum resolvable temperature difference
 Optical transfer function
 Signal transfer function
References
 ^ Holst, G.C. (1998). Testing and Evaluation of Infrared Imaging Systems (2nd ed.). Florida:JCD Publishing, Washington:SPIE.
 ^ Electro Optical Industries, Inc.(2005). EO TestLab Methodology. In Education/Ref. http://www.electrooptical.com/html/toplevel/educationref.asp.
 ^ Mazzetta, J.A.; Scopatz, S.D. (2007). Automated Testing of Ultraviolet, Visible, and Infrared Sensors Using Shared Optics. Infrared Imaging Systems: Design Analysis, Modeling, and Testing XVIII,Vol. 6543, pp. 6543131 65431314
 ^ Chapra, S.C.; Canale, R.P. (2006). Numerical Methods for Engineers (5th ed.). New York, New York: McGrawHill
External links
 SPIE international society advancing light based research.
 Electro Optical Industries infrared and visible light spectrum test and calibration equipment.
 MTF expressing the imaging quality.
 Quick MTF software for measuring resolution, including ESF, LSF and MTF.
 MTF on Optipedia Modulation transfer function written by Glenn D. Boreman on Optipedia.
 How to Measure MTF How to Measure MTF and other Properties of Lenses written by Optikos Corporation.
Categories: Image processing
 Infrared imaging
 Optics
Wikimedia Foundation. 2010.
Look at other dictionaries:
Optical transfer function — The optical transfer function (OTF) of an imaging system (camera, video system, microscope etc) is the true measure of resolution (image sharpness) that the system is capable of. The common practice of defining resolution in terms of pixel count… … Wikipedia
Minimum resolvable temperature difference — (MRTD) is a measure for assessing the performance of infrared cameras, and is inversely proportional to the modulation transfer function. Typically, an operator is asked to assess the minimum temperature difference at which a 4 bar target can be… … Wikipedia
MTF — may refer to: * Make Trade Fair * Male to Female, a transwoman * Manual Transmission Fluid, see hydraulic fluid * Modulation Transfer Function, see optical transfer function * Modulation Transfer Function (Infrared Imaging) * Move to front… … Wikipedia
Optical resolution — This article is about optical resolution in optics. For the method of separating enantiomers in chemistry, see Chiral resolution. Optical resolution describes the ability of an imaging system to resolve detail in the object that is being imaged.… … Wikipedia
Mohammad Ataul Karim — Born May 4, 1953 (1953 05 04) (age 58) Sylhet, Bangladesh Citizenship Bangladesh / USA … Wikipedia
Depth of field — The area within the depth of field appears sharp, while the areas in front of and beyond the depth of field appear blurry … Wikipedia
Superlens — A superlens, super lens or perfect lens is a lens which uses metamaterials to go beyond the diffraction limit. The diffraction limit is an inherent limitation in conventional optical devices or lenses.[1] In 2000, a type of lens was proposed,… … Wikipedia
spectroscopy — spectroscopist /spek tros keuh pist/, n. /spek tros keuh pee, spek treuh skoh pee/, n. the science that deals with the use of the spectroscope and with spectrum analysis. [1865 70; SPECTRO + SCOPY] * * * Branch of analysis devoted to identifying… … Universalium
electromagnetic radiation — Physics. radiation consisting of electromagnetic waves, including radio waves, infrared, visible light, ultraviolet, x rays, and gamma rays. [1950 55] * * * Energy propagated through free space or through a material medium in the form of… … Universalium
Active pixel sensor — An active pixel sensor (APS) is an image sensor consisting of an integrated circuit containing an array of pixel sensors, each pixel containing a photodetector and an active amplifier. There are many types of active pixel sensors including the… … Wikipedia