- Pressure Regulated Volume Control
Pressure Regulated Volume Control (PRVC) is a
ventilatormode where the breaths are delivered mandatorily to assure preset volumes, with a constant inspiratory pressurecontinuously adapting to the patient's condition. The flow pattern is decelerating. This mode is a form of assist controlmode of ventilation, the breaths can either be ventilator initiated or patient initiated. This mode combines the advantages of volume controlled and pressure controlled Ventilation.
How the ventilator achieves PRVC mode
The first breath delivered to the patient is a volume controlled breath. The measured
plateau pressureis used as the pressure level for the next breath. The pressure is constant during the set inspiratory timeand the flow is decelerating. The set tidal volumeis achieved by automatic, breath-by-breath pressure regulation. The ventilator will adjust the inspiratory pressure control level, according to the mechanical properties of the airways/ lung/ thorax, to the lowest possible level to guarantee the preset tidal volume. If the measured tidal volume increases above the preset, the pressure level decreases in steps of maximum 3 cm H2O between consecutive breaths until the preset tidal volume is delivered. Maximum available pressure level is 5 cm H2O below a preset upper pressure limit.
Reference: [http://www.aic.cuhk.edu.hk/web8/download.htm#Introduction%20to%20pressure%20regulated%20volume%20control%20mode Proceedings of the International Critical Care Conference, Dubai, April 8-10th 2008]
Wikimedia Foundation. 2010.
Look at other dictionaries:
Pressure controlled intermittent mandatory ventilation — (PC IMV) is any mode of mechanical ventilation with preset inspiratory pressure (Pressure Control) and inspiratory time (Ti). Spontaneous breaths (i.e., inspiration is patient triggered and patient cycled) can exist between mandatory breaths… … Wikipedia
Pressure vessel — Vertical pressure vessels installed in a structure A pressure vessel is a closed container designed to hold gases or liquids at a pressure substantially different from the ambient pressure. The pressure differential is dangerous and many fatal… … Wikipedia
Environmental Control System — The Environmental Control System of an airliner provides air supply, thermal control and cabin pressurization for the passengers and crew. Avionics cooling, smoke detection, and fire suppression are also commonly considered part of the… … Wikipedia
International Narcotics Control Board — The International Narcotics Control Board (INCB) is the independent and quasi judicial control organ for the implementation of the United Nations drug conventions. It plays an important role in monitoring enforcement of restrictions on narcotics… … Wikipedia
Traffic Control — ➡ law enforcement * * * Introduction supervision of the movement of people, goods, or vehicles to ensure efficiency and safety. Traffic is the movement of people and goods from one location to another. The movement typically occurs… … Universalium
PRVC — pressure regulated volume control [respiration] … Medical dictionary
PRVC — • pressure regulated volume control [respiration] … Dictionary of medical acronyms & abbreviations
Mechanical ventilation — In architecture and climate control, mechanical or forced ventilation is the use of powered equipment, e.g. fans and blowers, to move air see ventilation (architecture). Mechanical ventilation Intervention … Wikipedia
Modes of mechanical ventilation — Contents 1 Positive and negative pressure ventilation 1.1 Negative pressure ventilation 1.1.1 Types of negat … Wikipedia
Искусственное дыхание — Эту статью следует викифицировать. Пожалуйста, оформите её согласно правилам оформления статей. Искусственное дыхание (искусственная вентиляция легких, ИВЛ) комплекс мер, направленных на поддержание оборота в … Википедия