# Potentiometer (measuring instrument)

﻿
Potentiometer (measuring instrument)

A potentiometer is an instrument for measuring the potential (voltage) in a circuit. Before the introduction of the moving coil and digital volt meters, potentiometers were used in measuring voltage, hence the '-meter' part of their name. The method was described by Johann Christian Poggendorff around 1841 and became a standard laboratory measuring technique.[1]

In this arrangement, a fraction of a known voltage from a resistive slide wire is compared with an unknown voltage by means of a galvanometer. The sliding contact or wiper of the potentiometer is adjusted and the galvanometer briefly connected between the sliding contact and the unknown voltage. The deflection of the galvanometer is observed and the sliding tap adjusted until the galvanometer no longer deflects from zero. At that point the galvanometer draws no current from the unknown source, and the magnitude of voltage can be calculated from the position of the sliding contact.

This null balance measuring method is still important in electrical metrology and standards work and is also used in other areas of electronics.

Measurement potentiometers are divided into four main classes listed below.

## Constant current potentiometer

A potentiometer being calibrated and then measuring an unknown voltage.
R1 is the resistance of the entire resistance wire. The arrow head represents the moving wiper.

In this circuit, the ends of a uniform resistance wire R1 are connected to a regulated DC supply VS for use as a voltage divider. The potentiometer is first calibrated by positioning the wiper (arrow) at the spot on the R1 wire that corresponds to the voltage of a standard cell so that ${R_2 \over R_1} = {cellvoltage \over V_\mathrm{S}}$

A standard electrochemical cell is used whose emf is known (e.g. 1.0183 volts for a Weston standard cell).[2][3]

The supply voltage VS is then adjusted until the galvanometer shows zero, indicating the voltage on R2 is equal to the standard cell voltage.

An unknown DC voltage, in series with the galvanometer, is then connected to the sliding wiper, across a variable-length section R3 of the resistance wire. The wiper is moved until no current flows into or out of the source of unknown voltage, as indicated by the galvanometer in series with the unknown voltage. The voltage across the selected R3 section of wire is then equal to the unknown voltage. All that remains is to calculate the unknown voltage from the fraction of the length of the resistance wire that was connected to the unknown voltage.

The galvanometer does not need to be calibrated, as its only function is to read zero or not zero. When measuring an unknown voltage and the galvanometer reads zero, no current is drawn from the unknown voltage and so the reading is independent of the source's internal resistance, as if by a voltmeter of infinite resistance.

Because the resistance wire can be made very uniform in cross-section and resistivity, and the position of the wiper can be measured easily, this method can be used to measure unknown DC voltages greater than or less than a calibration voltage produced by a standard cell without drawing any current from the standard cell.

If the potentiometer is attached to a constant voltage DC supply such as a lead–acid battery, then a second variable resistor (not shown) can be used to calibrate the potentiometer by varying the current through the R1 resistance wire.

If the length of the R1 resistance wire is AB, where A is the (-) end and B is the (+) end, and the movable wiper is at point X at a distance AX on the R3 portion of the resistance wire when the galvanometer gives a zero reading for an unknown voltage, the distance AX is measured or read from a preprinted scale next to the resistance wire. The unknown voltage can then be calculated: $V_U = V_S {AX \over AB}$

## Constant resistance potentiometer

The constant resistance potentiometer is a variation of the basic idea in which a variable current is fed through a fixed resistor. These are used primarily for measurements in the millivolt and microvolt range.

## Microvolt potentiometer

This is a form of the constant resistance potentiometer described above but designed to minimize the effects of contact resistance and thermal emf. This equipment is satisfactorily used down to readings of 1000 nV or so.

## Thermocouple potentiometer

Another development of the standard types was the 'thermocouple potentiometer' especially adapted for temperature measurement with thermocouples. [4] Potentiometers for use with thermocouples also measure the temperature at which the thermocouple wires are connected, so that cold-junction compensation may be applied to correct the apparent measured EMF to the standard cold-junction temperature of 0 degrees C.

## Analytical chemistry

To make a potentiometric determination of an analyte in a solution, the potential of the cell is measured. This measurement must be corrected for the reference and junction potentials. The concentration of the analyte can then be calculated from the Nernst Equation. Many varieties of this basic principle exist for quantitative measurements.

## References

1. ^ http://physics.kenyon.edu/EarlyApparatus/Electrical_Measurements/Potentiometer/Potentiometer.html Potentiometer, retrieved 2010 Nov 2
2. ^ Kenyon.edu Dept of Physics.
3. ^ scenta.co.uk Scenta.
4. ^ Kenyon.edu Dept of Physics. Thermodynamics: Thermocouple Potentiometer.

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• measuring instrument — noun instrument that shows the extent or amount or quantity or degree of something • Syn: ↑measuring system, ↑measuring device • Hypernyms: ↑instrument • Hyponyms: ↑accelerometer, ↑actinometer, ↑ …   Useful english dictionary

• measuring device — noun instrument that shows the extent or amount or quantity or degree of something • Syn: ↑measuring instrument, ↑measuring system • Hypernyms: ↑instrument • Hyponyms: ↑accelerometer, ↑actinometer, ↑ …   Useful english dictionary

• measuring system — noun instrument that shows the extent or amount or quantity or degree of something • Syn: ↑measuring instrument, ↑measuring device • Hypernyms: ↑instrument • Hyponyms: ↑accelerometer, ↑actinometer, ↑ …   Useful english dictionary

• Potentiometer — Po*ten ti*om e*ter, n. [Potential + meter.] (Elec.) An instrument for measuring or comparing electrial potentials or electro motive forces. [1913 Webster] …   The Collaborative International Dictionary of English

• potentiometer — ► NOUN ▪ an instrument for measuring or adjusting an electromotive force …   English terms dictionary

• potentiometer — instrument for measuring electromotive forces Scientific Instruments …   Phrontistery dictionary

• potentiometer — Instrument for measuring or controlling by sensing small changes in electrical resistance. Also called a rheostat …   Dictionary of automotive terms

• potentiometer — 1. An instrument used for measuring small differences in electrical potential. 2. An electrical resistor of fixed total resistance between two terminals, but with a third terminal attached to a slider that can make contact at any desired point… …   Medical dictionary

• potentiometer — noun Etymology: International Scientific Vocabulary potential + o + meter Date: 1881 1. an instrument for measuring electromotive forces 2. voltage divider • potentiometric adjective …   New Collegiate Dictionary