- Fourier transform ion cyclotron resonance
Infobox chemical analysis

name = Fourier transform ion cyclotron resonance

caption =A FTMS instrument at the Pacific Northwest National Laboratory, USA

acronym = FTMS, FTICR

classification =Mass spectrometry

analytes =

manufacturers =

related =Ion trap Quadrupole ion trap Penning trap Orbitrap

hyphenated =**Fourier transform ion cyclotron resonance mass spectrometry**, also known as**Fourier transform mass spectrometry**, is a type of mass analyzer (ormass spectrometer ) for determining themass-to-charge ratio (m/z) ofions based on the cyclotron frequency of the ions in a fixed magnetic field. [*[*] The ions are trapped in a*http://www.ncbi.nlm.nih.gov/entrez/query.fcgi?cmd=Retrieve&db=pubmed&dopt=Abstract&list_uids=9768511 Marshall, A. G.; Hendrickson, C. L.; Jackson, G. S., Fourier transform ion cyclotron resonance mass spectrometry: a primer. "Mass Spectrom Rev"*]**17**, 1-35.Penning trap (a magnetic field with electric trapping plates) where they are excited to a larger cyclotron radius by an oscillating electric field perpendicular to the magnetic field. The excitation also results in the ions moving in phase (in a packet). The signal is detected as an image current on a pair of plates which the packet of ions passes close to as they cyclotron. The resulting signal is called afree induction decay (FID), transient or interferogram that consists of a superposition ofsine waves . The useful signal is extracted from this data by performing aFourier transform to give amass spectrum .Fourier transform ion cyclotron resonance (FTICR) mass spectrometry is a very high resolution technique in that masses can be determined with very high accuracy. Many applications of FTICR-MS use this mass accuracy to help determine the composition of molecules based on accurate mass. This is possible due to the

mass defect of the elements. Another place that FTICR-MS is useful is in dealing with complex mixtures since the resolution (narrow peak width) allows the signals of two ions of similar mass to charge (m/z) to be detected as distinct ions. This high resolution is also useful in studying large macromolecules such as proteins with multiple charges which can be produced byelectrospray ionization . These large molecules contain a distribution ofisotopes that produce a series of isotopic peaks. Because the isotopic peaks are close to each other on the "m/z" axis, due to the multiple charges, the high resolving power of the FTICR is extremely useful.FTICR-MS differs significantly from other

mass spectrometry techniques in that the ions are not detected by hitting a detector such as anelectron multiplier but only by passing near detection plates. Additionally the masses are not resolved in space or time as with other techniques but only in frequency. Thus, the different ions are not detected in different places as withsector instrument s or at different times as withtime-of-flight instruments but all ions are detected simultaneously over some given period of time.FT-ICR was invented by

Alan G. Marshall and [*http://www.chem.ubc.ca/personnel/faculty/comisarow/ Melvin B. Comisarow*] at the University of British Columbia. The first paper appeared inChemical Physics Letters in 1974. [*[*] The inspiration was earlier developments in conventional ICR and Fourier Transform Nuclear Magnetic Resonance (FT-NMR) spectroscopy. Marshall has continued to develop the technique at The Ohio State University and Florida State University.*http://dx.doi.org/10.1016/0009-2614(74)89137-2 M.B. Comisarow and A.G. Marshall, "Chem. Phys. Lett."*]**25**, 282 (1974)**Theory**The physics of FTICR is similar to that of a

cyclotron at least in the first approximation.In the simplest form (idealized) the relationship between the cyclotron frequency and the mass to charge ratio is given by:

:$f\; =\; frac\{qB\}\{2pi\; m\}$

where "f" = cyclotron frequency, "q" = ion charge, "B" =

magnetic field strength and "m" = ion mass.This is more often represented in

angular frequency ::$omega\; \_c\; =\; frac\{qB\}\{m\}$

where $omega\; \_c$ is the angular cyclotron frequency which is related to frequency by the definition $f\; =\; frac\{omega\}\{2pi\}$.

Because of the quadrupolar electrical field used to trap the ions in the axial direction this relationship is only approximate. The axial electrical trapping results in axial oscillations within the trap with the (angular) frequency:

:$omega\; \_t\; =\; sqrt$frac{qalpha}{m}

Where $alpha$ is a constant similar to the spring constant of a

harmonic oscillator and is dependent on voltage and the trap dimensions and geometry.The electric field and the resulting axial harmonic motion reduces the cyclotron frequency and introduces a second radial motion called magnetron motion that occurs at the magnetron frequency. The cyclotron motion is still the frequency being used but the relationship above is not exact due to this phenomenon. The natural angular frequencies of motion are:

:$omega\; \_pm\; =\; frac\{omega\; \_c\}\{2\}\; pm\; sqrt\{left(\{frac\{omega\; \_c\}\{2\; ight)^2-left(\{frac\{omega\; \_t\}\{2\; ight)^2\}$

where $omega\; \_t$ is the axial trapping frequency due the axial electrical trapping and $omega\; \_+$ is the reduced cyclotron (angular) frequency and $omega\; \_-$ is the magnetron (angular) frequency. Again $omega\; \_+$ is what is typically measured in FTICR. The meaning of this equation can be understood qualitatively by considering the case where $omega\; \_t$ is small, which is generally true. In that case the value of the radical is just slightly less than $omega\_c/2$ and the value of $omega\; \_+$ is just slightly less than $omega\_c$ (the cyclotron frequency has been slightly reduced). For $omega\_-$ the value of the radical is the same (slightly less than $omega\_c/2$) but it is being subtracted from $omega\; \_c/2$ resulting in a small number equal to $omega\; \_c\; -omega\; \_+$ (i.e. the exact amount that the cyclotron frequency was reduced by).

**ICR cell types**A review of different cell geometries with their specific electric configurations is available in the literature [

*[*] . However, ICR cells can belong to one of the following two categories.*http://dx.doi.org/10.1016/0168-1176(95)04190-V S. Guan, A. G. Marshall, Int. J. Mass Spectrom., 146/147 (1995) 261*]**1. Closed cells**Several closed ICR cells with different geometries were fabricated and their performance has been characterized. Grids were used as end caps to apply an axial electric field for trapping ions axially (parallel to the magnetic field lines). Ions can be either generated inside the cell (by Electron impact ionization) or can be enjected to the cell from an external ionization source (such as Electrospray or MALDI). Nested ICR cells with double pair of grids were also fabricated to trap both positive and negative ions simultaneously.

**2. Open cells**The most common geometry is a cylinder, which is axially segmented into different parts to produce different ring electrodes. The central ring electrode is generally used for applying radial excitation electric field and detection. DC electric voltage is applied on the terminal ring electrodes to trap ions along the magnetic field lines. Open cylindrical cells with ring electrodes of different diameters have also been designed [

*[*] . They proved not only capable in trapping and detecting both ion polarities simultaneously, but also they succeeded to separate positive from negative ions radially. This presented a large discrimination in kinetic ion acceleration between positive and negative ions trapped simultaneously inside the new cell. Several ion axial acceleration schemes were recently written for ion-ion collision studies [*http://dx.doi.org/10.1063/1.2751100 B. Kanawati, K. P. Wanczek, Characterisation of a New Open Cylindrical ICR Cell with unusual geometry. Rev. Sci. Instrum. (2007), 78, 7, 074102*]*[*] .*http://dx.doi.org/10.1016/j.ijms.2007.09.007 B. Kanawati, K. P. Wanczek, Characterisation of a New Open Cylindrical ICR Cell for Ion-Ion Collision Studies, Int. J. Mass Spectrom. (2007), in press. DOI:10.1016/j.ijms.2007.09.007*]**ee also***

Ion cyclotron resonance **External links*** [

*http://www.magnet.fsu.edu/science/cimar/icr/ National High Field Fourier Transform Ion Cyclotron Resonance (FT-ICR) mass spectrometry facility, Tallahassee, Florida, USA*]

* [*http://www.magnet.fsu.edu/education/tutorials/magnetacademy/fticr/ What's in an Oil Drop? An Introduction to Fourier Transform Ion Cyclotron Resonance (FT-ICR) for Non-scientists*] National High Magnetic Field Laboratory

* [*http://www.chm.bris.ac.uk/ms/theory/fticr-massspec.html Fourier-transform Ion Cyclotron Resonance (FT-ICR)*] FT-ICR Introduction University of Bristol**References**

*Wikimedia Foundation.
2010.*

### Look at other dictionaries:

**Ion cyclotron resonance**— is a phenomenon related to the movement of ions in a magnetic field. It is used for accelerating ions in a cyclotron, and for measuring the masses of an ionized analyte in mass spectrometry, particularly with Fourier transform ion cyclotron… … Wikipedia**Fourier analysis**— In mathematics, Fourier analysis is a subject area which grew out of the study of Fourier series. The subject began with trying to understand when it was possible to represent general functions by sums of simpler trigonometric functions. The… … Wikipedia**Ion trap**— ). An ion trap mass spectrometer may incorporate a Penning trap (Fourier transform ion cyclotron resonance) [cite journal | title = High accuracy mass spectrometry with stored ions | author = Blaum, Klaus | journal = Physics Reports | volume =… … Wikipedia**FTICR**— Fourier Transform Ion Cyclotron Resonance (Academic & Science » Physics) … Abbreviations dictionary**Mass spectrometry**— (MS) is an analytical technique that measures the mass to charge ratio of charged particles.[1] It is used for determining masses of particles, for determining the elemental composition of a sample or molecule, and for elucidating the chemical… … Wikipedia**Penning trap**— Penning traps are devices for the storage of charged particles using a constant static magnetic field and a spatially inhomogeneous static electric field. This kind of trap is particularly well suited to precision measurements of properties of… … Wikipedia**History of mass spectrometry**— The history of mass spectrometry dates back more than one hundred years and has its roots in physical and chemical studies regarding the nature of matter. The study of gas discharges in the mid 19th century led to the discovery of anode and… … Wikipedia**Michael L. Gross (chemist)**— For other people named Michael Gross, see Michael Gross (disambiguation). Michael L. Gross (born 1940) is Professor of Chemistry, Medicine, and Immunology, at Washington University in St. Louis. He was formerly Professor of Chemistry at the… … Wikipedia**List of materials analysis methods**— List of materials analysis methods: Contents: Top · 0–9 · A B C D E F G H I J K L M N O P Q R S T U V W X Y Z μSR see Muon spin spectroscopy … Wikipedia**Mass spectrum**— Electron ionization mass spectrum of toluene [4]. Note parent peak corresponding to molecular mass M = 92 (C7H8+) and highest peak at M 1 = 91 (C7H7+, quasi stable tropylium cation). A mass spectrum is an inten … Wikipedia